-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170829_215555_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.584406 to 32.165399
-each separately from 7s to 32s and from 165110us to 831799us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.246815 in 1 packets - #/No: 18/18
      16.333311 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-29T21:56:05.584.406.000 to 2017-08-29T21:56:30.165.399.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.246815 in 1 packets - #/No: 18/18
      16.333311 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4411512 to 33.4410537
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999688 in 1 packets - #/No: 18/18
      16.9999337 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170829_215555_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.0255561 to 9.0258451
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2467830  -- OK for SL --
   MIU time from 2017-08-29T21:56:05.584.406.000 to 2017-08-29T21:56:05.584.695.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.246815

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.2726281 to 17.2729171
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3332470  -- OK for SL --
   MIU time from 2017-08-29T21:56:13.831.510.000 to 2017-08-29T21:56:13.831.799.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.333311

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.6061641 to 33.6064531
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-29T21:56:30.165.110.000 to 2017-08-29T21:56:30.165.399.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-29T21:56:05.584.265.750
   Max SL corrected packet time: 2017-08-29T21:56:30.165.258.750
   Min SL corrected XO rel time: 7.5844072
   Max SL corrected XO rel time: 32.1654000
   TM values from -69 to 58 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-29T21:56:05.584.265.750 (XO: 7.5844072, t=34.0°C)  Rank: 00017
   TM values: Max =    51  Min =   -60  Ave =    -4.54  Std =    18.29
   Min TM value:-60 at an offset of 133.038 usec -> 2017-08-29T21:56:05.584.398.788 (XO: 7.5845402)
   Max TM value:51 at an offset of 156.588 usec -> 2017-08-29T21:56:05.584.422.338 (XO: 7.5845638)
   Max dTM/dt [1/us]::9.07e+02 at an offset of 169.738 usec -> 2017-08-29T21:56:05.584.435.488 (XO: 7.5845769)
   Min dTM/dt [1/us]::-8.79e+02 at an offset of 169.388 usec -> 2017-08-29T21:56:05.584.435.138 (XO: 7.5845766)

 - Block number: 2 starting at 2017-08-29T21:56:13.831.369.750 (XO: 15.8315113, t=34.0°C)  Rank: 00017
   TM values: Max =    58  Min =   -69  Ave =    -4.53  Std =    17.95
   Min TM value:-69 at an offset of 89.313 usec -> 2017-08-29T21:56:13.831.459.062 (XO: 15.8316007)
   Max TM value:58 at an offset of 125.625 usec -> 2017-08-29T21:56:13.831.495.375 (XO: 15.8316370)
   Max dTM/dt [1/us]::9.23e+02 at an offset of 126.538 usec -> 2017-08-29T21:56:13.831.496.287 (XO: 15.8316379)
   Min dTM/dt [1/us]::-9.75e+02 at an offset of 125.650 usec -> 2017-08-29T21:56:13.831.495.400 (XO: 15.8316370)

 - Block number: 3 starting at 2017-08-29T21:56:30.164.969.750 (XO: 32.1651110, t=34.0°C)  Rank: 00011
   TM values: Max =    53  Min =   -57  Ave =    -4.60  Std =    14.97
   Min TM value:-57 at an offset of 106.813 usec -> 2017-08-29T21:56:30.165.076.562 (XO: 32.1652179)
   Max TM value:53 at an offset of 227.526 usec -> 2017-08-29T21:56:30.165.197.275 (XO: 32.1653386)
   Max dTM/dt [1/us]::8.50e+02 at an offset of 205.813 usec -> 2017-08-29T21:56:30.165.175.563 (XO: 32.1653169)
   Min dTM/dt [1/us]::-8.04e+02 at an offset of 106.775 usec -> 2017-08-29T21:56:30.165.076.525 (XO: 32.1652178)
