-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170808_162121_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.546565 to 42.475221
-each separately from 5s to 42s and from 20410us to 546854us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.473556 in 1 packets - #/No: 18/18
      27.454233 in 1 packets - #/No: 36/18
MIU packet time from 2017-08-08T16:21:29.546.565.000 to 2017-08-08T16:22:06.475.221.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.473556 in 1 packets - #/No: 18/18
      27.454233 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4364410 to 43.4362976
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999612 in 1 packets - #/No: 18/18
      26.9998954 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170808_162121_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9830054 to 6.9832944
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4735190  -- OK for SL --
   MIU time from 2017-08-08T16:21:29.546.565.000 to 2017-08-08T16:21:29.546.854.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.473556

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4568134 to 16.4571024
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.4541270  -- OK for SL --
   MIU time from 2017-08-08T16:21:39.020.410.000 to 2017-08-08T16:21:39.020.699.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.454233

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.9112294 to 43.9115184
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-08-08T16:22:06.474.932.000 to 2017-08-08T16:22:06.475.221.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-08-08T16:21:29.546.424.750
   Max SL corrected packet time: 2017-08-08T16:22:06.475.080.750
   Min SL corrected XO rel time: 5.5465665
   Max SL corrected XO rel time: 42.4752227
   TM values from -72 to 106 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-08-08T16:21:29.546.424.750 (XO: 5.5465665, t=35.6°C)  Rank: 00009
   TM values: Max =   106  Min =   -49  Ave =    -4.55  Std =     7.93
   Min TM value:-49 at an offset of 119.663 usec -> 2017-08-08T16:21:29.546.544.412 (XO: 5.5466861)
   Max TM value:106 at an offset of 119.450 usec -> 2017-08-08T16:21:29.546.544.200 (XO: 5.5466859)
   Max dTM/dt [1/us]::8.33e+02 at an offset of 119.400 usec -> 2017-08-08T16:21:29.546.544.150 (XO: 5.5466859)
   Min dTM/dt [1/us]::-7.84e+02 at an offset of 119.513 usec -> 2017-08-08T16:21:29.546.544.262 (XO: 5.5466860)

 - Block number: 2 starting at 2017-08-08T16:21:39.020.269.750 (XO: 15.0204113, t=35.6°C)  Rank: 00008
   TM values: Max =    24  Min =   -33  Ave =    -4.48  Std =     8.31
   Min TM value:-33 at an offset of 72.900 usec -> 2017-08-08T16:21:39.020.342.650 (XO: 15.0204842)
   Max TM value:24 at an offset of 146.338 usec -> 2017-08-08T16:21:39.020.416.088 (XO: 15.0205576)
   Max dTM/dt [1/us]::2.56e+02 at an offset of 72.950 usec -> 2017-08-08T16:21:39.020.342.700 (XO: 15.0204842)
   Min dTM/dt [1/us]::-1.58e+02 at an offset of 72.863 usec -> 2017-08-08T16:21:39.020.342.612 (XO: 15.0204841)

 - Block number: 3 starting at 2017-08-08T16:22:06.474.791.750 (XO: 42.4749337, t=35.6°C)  Rank: 00011
   TM values: Max =    62  Min =   -72  Ave =    -4.43  Std =    12.65
   Min TM value:-72 at an offset of 115.200 usec -> 2017-08-08T16:22:06.474.906.950 (XO: 42.4750489)
   Max TM value:62 at an offset of 161.301 usec -> 2017-08-08T16:22:06.474.953.050 (XO: 42.4750950)
   Max dTM/dt [1/us]::8.38e+02 at an offset of 115.263 usec -> 2017-08-08T16:22:06.474.907.012 (XO: 42.4750489)
   Min dTM/dt [1/us]::-6.96e+02 at an offset of 115.438 usec -> 2017-08-08T16:22:06.474.907.187 (XO: 42.4750491)
