-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170707_235703_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.833907 to 30.466533
-each separately from 0s to 30s and from 442930us to 834196us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.608734 in 1 packets - #/No: 18/18
      13.023025 in 1 packets - #/No: 36/18
MIU packet time from 2017-07-07T23:57:06.833.907.000 to 2017-07-07T23:57:36.466.533.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.608734 in 1 packets - #/No: 18/18
      13.023025 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4397837 to 31.4396694
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999353 in 1 packets - #/No: 18/18
      12.9999504 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170707_235703_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2736888 to 2.2739778
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6086710  -- OK for SL --
   MIU time from 2017-07-07T23:57:06.833.907.000 to 2017-07-07T23:57:06.834.196.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.608734

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.8826488 to 18.8829378
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0229750  -- OK for SL --
   MIU time from 2017-07-07T23:57:23.442.930.000 to 2017-07-07T23:57:23.443.219.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.023025

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9059128 to 31.9062018
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-07-07T23:57:36.466.244.000 to 2017-07-07T23:57:36.466.533.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-07-07T23:57:06.833.766.750
   Max SL corrected packet time: 2017-07-07T23:57:36.466.392.750
   Min SL corrected XO rel time: 0.8339083
   Max SL corrected XO rel time: 30.4665342
   TM values from -27 to 18 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-07-07T23:57:06.833.766.750 (XO: 0.8339083, t=34.0°C)  Rank: 00003
   TM values: Max =    16  Min =   -24  Ave =    -4.63  Std =     5.25
   Min TM value:-24 at an offset of 140.488 usec -> 2017-07-07T23:57:06.833.907.238 (XO: 0.8340488)
   Max TM value:16 at an offset of 301.389 usec -> 2017-07-07T23:57:06.834.068.138 (XO: 0.8342097)
   Max dTM/dt [1/us]::2.28e+02 at an offset of 70.750 usec -> 2017-07-07T23:57:06.833.837.500 (XO: 0.8339790)
   Min dTM/dt [1/us]::-2.50e+02 at an offset of 90.250 usec -> 2017-07-07T23:57:06.833.857.000 (XO: 0.8339985)

 - Block number: 2 starting at 2017-07-07T23:57:23.442.789.750 (XO: 17.4429315, t=34.0°C)  Rank: 00003
   TM values: Max =    18  Min =   -26  Ave =    -4.53  Std =     6.96
   Min TM value:-26 at an offset of 53.075 usec -> 2017-07-07T23:57:23.442.842.825 (XO: 17.4429846)
   Max TM value:18 at an offset of 113.625 usec -> 2017-07-07T23:57:23.442.903.375 (XO: 17.4430451)
   Max dTM/dt [1/us]::3.54e+02 at an offset of 96.463 usec -> 2017-07-07T23:57:23.442.886.212 (XO: 17.4430279)
   Min dTM/dt [1/us]::-3.12e+02 at an offset of 71.463 usec -> 2017-07-07T23:57:23.442.861.212 (XO: 17.4430029)

 - Block number: 3 starting at 2017-07-07T23:57:36.466.103.750 (XO: 30.4662452, t=34.0°C)  Rank: 00003
   TM values: Max =    15  Min =   -27  Ave =    -4.59  Std =     5.43
   Min TM value:-27 at an offset of 216.788 usec -> 2017-07-07T23:57:36.466.320.538 (XO: 30.4664620)
   Max TM value:15 at an offset of 5.263 usec -> 2017-07-07T23:57:36.466.109.012 (XO: 30.4662504)
   Max dTM/dt [1/us]::2.62e+02 at an offset of 97.200 usec -> 2017-07-07T23:57:36.466.200.950 (XO: 30.4663424)
   Min dTM/dt [1/us]::-2.37e+02 at an offset of 98.488 usec -> 2017-07-07T23:57:36.466.202.237 (XO: 30.4663437)
