-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20170227_013238_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.649327 to 32.008958
-each separately from 5s to 32s and from 8669us to 934539us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.284634 in 1 packets - #/No: 18/18
      14.074130 in 1 packets - #/No: 36/18
MIU packet time from 2017-02-27T01:32:46.649.327.000 to 2017-02-27T01:33:13.008.958.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.284634 in 1 packets - #/No: 18/18
      14.074130 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4215662 to 33.4214704
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999574 in 1 packets - #/No: 18/18
      14.9999468 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20170227_013238_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.0708922 to 7.0711812
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2845910  -- OK for SL --
   MIU time from 2017-02-27T01:32:46.649.327.000 to 2017-02-27T01:32:46.649.616.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.284634

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3557722 to 19.3560612
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0740790  -- OK for SL --
   MIU time from 2017-02-27T01:32:58.934.250.000 to 2017-02-27T01:32:58.934.539.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.074130

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.4301402 to 33.4304292
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-02-27T01:33:13.008.669.000 to 2017-02-27T01:33:13.008.958.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-02-27T01:32:46.649.186.750
   Max SL corrected packet time: 2017-02-27T01:33:13.008.817.750
   Min SL corrected XO rel time: 5.6493283
   Max SL corrected XO rel time: 32.0089588
   TM values from -37 to 30 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-02-27T01:32:46.649.186.750 (XO: 5.6493283, t=35.6°C)  Rank: 00010
   TM values: Max =    30  Min =   -37  Ave =    -4.42  Std =     9.29
   Min TM value:-37 at an offset of 98.925 usec -> 2017-02-27T01:32:46.649.285.675 (XO: 5.6494273)
   Max TM value:30 at an offset of 99.400 usec -> 2017-02-27T01:32:46.649.286.150 (XO: 5.6494277)
   Max dTM/dt [1/us]::4.32e+02 at an offset of 89.163 usec -> 2017-02-27T01:32:46.649.275.912 (XO: 5.6494175)
   Min dTM/dt [1/us]::-4.46e+02 at an offset of 173.151 usec -> 2017-02-27T01:32:46.649.359.900 (XO: 5.6495015)

 - Block number: 2 starting at 2017-02-27T01:32:58.934.109.750 (XO: 17.9342520, t=35.6°C)  Rank: 00010
   TM values: Max =    27  Min =   -36  Ave =    -4.49  Std =     8.67
   Min TM value:-36 at an offset of 105.000 usec -> 2017-02-27T01:32:58.934.214.750 (XO: 17.9343570)
   Max TM value:27 at an offset of 100.313 usec -> 2017-02-27T01:32:58.934.210.062 (XO: 17.9343523)
   Max dTM/dt [1/us]::4.55e+02 at an offset of 106.263 usec -> 2017-02-27T01:32:58.934.216.012 (XO: 17.9343582)
   Min dTM/dt [1/us]::-4.80e+02 at an offset of 193.126 usec -> 2017-02-27T01:32:58.934.302.875 (XO: 17.9344451)

 - Block number: 3 starting at 2017-02-27T01:33:13.008.528.750 (XO: 32.0086698, t=35.6°C)  Rank: 00011
   TM values: Max =    26  Min =   -35  Ave =    -4.43  Std =     7.93
   Min TM value:-35 at an offset of 107.188 usec -> 2017-02-27T01:33:13.008.635.937 (XO: 32.0087770)
   Max TM value:26 at an offset of 108.888 usec -> 2017-02-27T01:33:13.008.637.637 (XO: 32.0087787)
   Max dTM/dt [1/us]::4.59e+02 at an offset of 113.488 usec -> 2017-02-27T01:33:13.008.642.237 (XO: 32.0087833)
   Min dTM/dt [1/us]::-4.73e+02 at an offset of 107.150 usec -> 2017-02-27T01:33:13.008.635.900 (XO: 32.0087770)
