-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151203_100422_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.105071 to 38.395410
-each separately from 7s to 38s and from 10984us to 395410us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.905624 in 1 packets - #/No: 18/18
      23.383848 in 1 packets - #/No: 36/18
MIU packet time from 2015-12-03T10:04:32.105.071.000 to 2015-12-03T10:05:03.395.410.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.905624 in 1 packets - #/No: 18/18
      23.383848 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4790305 to 39.4789335
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999750 in 1 packets - #/No: 18/18
      22.9999280 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BE -> 32.4°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151203_100422_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5841028 to 8.5843918
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9055990  -- OK for SL --
   MIU time from 2015-12-03T10:04:32.105.071.000 to 2015-12-03T10:04:32.105.360.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.905624

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4899908 to 16.4902798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3837750  -- OK for SL --
   MIU time from 2015-12-03T10:04:40.010.984.000 to 2015-12-03T10:04:40.011.273.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.383848

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.8740548 to 39.8743438
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-12-03T10:05:03.395.121.000 to 2015-12-03T10:05:03.395.410.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-12-03T10:04:32.104.930.750
   Max SL corrected packet time: 2015-12-03T10:05:03.395.269.750
   Min SL corrected XO rel time: 7.1050726
   Max SL corrected XO rel time: 38.3954116
   TM values from -19 to 10 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-12-03T10:04:32.104.930.750 (XO: 7.1050726, t=34.0°C)  Rank: 00003
   TM values: Max =     9  Min =   -19  Ave =    -4.69  Std =     3.99
   Min TM value:-19 at an offset of 110.950 usec -> 2015-12-03T10:04:32.105.041.700 (XO: 7.1051836)
   Max TM value:9 at an offset of 88.763 usec -> 2015-12-03T10:04:32.105.019.512 (XO: 7.1051614)
   Max dTM/dt [1/us]::8.40e+01 at an offset of 216.363 usec -> 2015-12-03T10:04:32.105.147.113 (XO: 7.1052890)
   Min dTM/dt [1/us]::-8.80e+01 at an offset of 275.013 usec -> 2015-12-03T10:04:32.105.205.763 (XO: 7.1053476)

 - Block number: 2 starting at 2015-12-03T10:04:40.010.843.750 (XO: 15.0109853, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -17  Ave =    -4.74  Std =     3.46
   Min TM value:-17 at an offset of 17.138 usec -> 2015-12-03T10:04:40.010.860.887 (XO: 15.0110025)
   Max TM value:7 at an offset of 30.250 usec -> 2015-12-03T10:04:40.010.874.000 (XO: 15.0110156)
   Max dTM/dt [1/us]::7.50e+01 at an offset of 0.025 usec -> 2015-12-03T10:04:40.010.843.775 (XO: 15.0109854)
   Min dTM/dt [1/us]::-9.40e+01 at an offset of 123.238 usec -> 2015-12-03T10:04:40.010.966.987 (XO: 15.0111086)

 - Block number: 3 starting at 2015-12-03T10:05:03.394.980.750 (XO: 38.3951226, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -19  Ave =    -4.68  Std =     4.01
   Min TM value:-19 at an offset of 189.026 usec -> 2015-12-03T10:05:03.395.169.775 (XO: 38.3953116)
   Max TM value:10 at an offset of 36.063 usec -> 2015-12-03T10:05:03.395.016.812 (XO: 38.3951586)
   Max dTM/dt [1/us]::1.03e+02 at an offset of 164.813 usec -> 2015-12-03T10:05:03.395.145.563 (XO: 38.3952874)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 229.201 usec -> 2015-12-03T10:05:03.395.209.950 (XO: 38.3953518)
