-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151115_105944_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.685582 to 42.219196
-each separately from 4s to 42s and from 218907us to 685871us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.814081 in 1 packets - #/No: 18/18
      25.718666 in 1 packets - #/No: 36/18
MIU packet time from 2015-11-15T10:59:51.685.582.000 to 2015-11-15T11:00:29.219.196.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.814081 in 1 packets - #/No: 18/18
      25.718666 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4282998 to 43.4281893
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999651 in 1 packets - #/No: 18/18
      25.9999244 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      08 -> 44.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151115_105944_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.1138808 to 6.1141698
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8140470  -- OK for SL --
   MIU time from 2015-11-15T10:59:51.685.582.000 to 2015-11-15T10:59:51.685.871.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.814081

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9282168 to 17.9285058
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.7185910  -- OK for SL --
   MIU time from 2015-11-15T11:00:03.499.952.000 to 2015-11-15T11:00:03.500.241.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.718666

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6470968 to 43.6473858
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-11-15T11:00:29.218.907.000 to 2015-11-15T11:00:29.219.196.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-11-15T10:59:51.685.441.750
   Max SL corrected packet time: 2015-11-15T11:00:29.219.055.750
   Min SL corrected XO rel time: 4.6855831
   Max SL corrected XO rel time: 42.2191971
   TM values from -29 to 20 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-11-15T10:59:51.685.441.750 (XO: 4.6855831, t=34.0°C)  Rank: 00007
   TM values: Max =    20  Min =   -29  Ave =    -4.62  Std =     8.81
   Min TM value:-29 at an offset of 168.988 usec -> 2015-11-15T10:59:51.685.610.738 (XO: 4.6857520)
   Max TM value:20 at an offset of 110.550 usec -> 2015-11-15T10:59:51.685.552.300 (XO: 4.6856936)
   Max dTM/dt [1/us]::2.21e+02 at an offset of 91.688 usec -> 2015-11-15T10:59:51.685.533.437 (XO: 4.6856747)
   Min dTM/dt [1/us]::-2.19e+02 at an offset of 112.450 usec -> 2015-11-15T10:59:51.685.554.200 (XO: 4.6856955)

 - Block number: 2 starting at 2015-11-15T11:00:03.499.811.750 (XO: 16.4999533, t=34.0°C)  Rank: 00005
   TM values: Max =    16  Min =   -26  Ave =    -4.78  Std =     6.95
   Min TM value:-26 at an offset of 18.063 usec -> 2015-11-15T11:00:03.499.829.812 (XO: 16.4999714)
   Max TM value:16 at an offset of 18.625 usec -> 2015-11-15T11:00:03.499.830.375 (XO: 16.4999720)
   Max dTM/dt [1/us]::1.94e+02 at an offset of 18.500 usec -> 2015-11-15T11:00:03.499.830.250 (XO: 16.4999718)
   Min dTM/dt [1/us]::-2.02e+02 at an offset of 107.100 usec -> 2015-11-15T11:00:03.499.918.850 (XO: 16.5000605)

 - Block number: 3 starting at 2015-11-15T11:00:29.218.766.750 (XO: 42.2189081, t=34.0°C)  Rank: 00003
   TM values: Max =    14  Min =   -24  Ave =    -4.80  Std =     5.32
   Min TM value:-24 at an offset of 82.925 usec -> 2015-11-15T11:00:29.218.849.675 (XO: 42.2189911)
   Max TM value:14 at an offset of 89.625 usec -> 2015-11-15T11:00:29.218.856.375 (XO: 42.2189978)
   Max dTM/dt [1/us]::1.43e+02 at an offset of 0.000 usec -> 2015-11-15T11:00:29.218.766.750 (XO: 42.2189081)
   Min dTM/dt [1/us]::-1.19e+02 at an offset of 174.051 usec -> 2015-11-15T11:00:29.218.940.800 (XO: 42.2190822)
