-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20151029_164608_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.969700 to 34.826119
-each separately from 0s to 34s and from 825830us to 969989us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.928627 in 1 packets - #/No: 18/18
      15.926925 in 1 packets - #/No: 36/18
MIU packet time from 2015-10-29T16:46:11.969.700.000 to 2015-10-29T16:46:45.826.119.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.928627 in 1 packets - #/No: 18/18
      15.926925 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4964049 to 35.4963065
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999479 in 1 packets - #/No: 18/18
      15.9999537 in 1 packets - #/No: 36/18
Temperature FPGA: 
      08 -> 44.6°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20151029_164608_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4661029 to 2.4663919
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9285750  -- OK for SL --
   MIU time from 2015-10-29T16:46:11.969.700.000 to 2015-10-29T16:46:11.969.989.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.928627

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3949669 to 20.3952559
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9268790  -- OK for SL --
   MIU time from 2015-10-29T16:46:29.898.616.000 to 2015-10-29T16:46:29.898.905.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.926925

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.3221349 to 36.3224239
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-10-29T16:46:45.825.830.000 to 2015-10-29T16:46:45.826.119.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-10-29T16:46:11.969.559.750
   Max SL corrected packet time: 2015-10-29T16:46:45.825.978.750
   Min SL corrected XO rel time: 0.9697008
   Max SL corrected XO rel time: 34.8261198
   TM values from -53 to 72 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-10-29T16:46:11.969.559.750 (XO: 0.9697008, t=34.0°C)  Rank: 00008
   TM values: Max =    34  Min =   -46  Ave =    -4.70  Std =    12.58
   Min TM value:-46 at an offset of 74.813 usec -> 2015-10-29T16:46:11.969.634.562 (XO: 0.9697756)
   Max TM value:34 at an offset of 125.625 usec -> 2015-10-29T16:46:11.969.685.375 (XO: 0.9698264)
   Max dTM/dt [1/us]::3.47e+02 at an offset of 178.513 usec -> 2015-10-29T16:46:11.969.738.263 (XO: 0.9698793)
   Min dTM/dt [1/us]::-3.40e+02 at an offset of 298.201 usec -> 2015-10-29T16:46:11.969.857.950 (XO: 0.9699990)

 - Block number: 2 starting at 2015-10-29T16:46:29.898.475.750 (XO: 18.8986167, t=34.0°C)  Rank: 00017
   TM values: Max =    72  Min =   -44  Ave =    -4.56  Std =     9.04
   Min TM value:-44 at an offset of 43.050 usec -> 2015-10-29T16:46:29.898.518.800 (XO: 18.8986598)
   Max TM value:72 at an offset of 40.550 usec -> 2015-10-29T16:46:29.898.516.300 (XO: 18.8986573)
   Max dTM/dt [1/us]::3.96e+02 at an offset of 40.488 usec -> 2015-10-29T16:46:29.898.516.237 (XO: 18.8986572)
   Min dTM/dt [1/us]::-3.59e+02 at an offset of 41.250 usec -> 2015-10-29T16:46:29.898.517.000 (XO: 18.8986580)

 - Block number: 3 starting at 2015-10-29T16:46:45.825.689.750 (XO: 34.8258308, t=34.0°C)  Rank: 00009
   TM values: Max =    42  Min =   -53  Ave =    -4.64  Std =    17.30
   Min TM value:-53 at an offset of 215.088 usec -> 2015-10-29T16:46:45.825.904.838 (XO: 34.8260459)
   Max TM value:42 at an offset of 227.238 usec -> 2015-10-29T16:46:45.825.916.988 (XO: 34.8260580)
   Max dTM/dt [1/us]::4.05e+02 at an offset of 296.863 usec -> 2015-10-29T16:46:45.825.986.613 (XO: 34.8261277)
   Min dTM/dt [1/us]::-4.18e+02 at an offset of 130.788 usec -> 2015-10-29T16:46:45.825.820.537 (XO: 34.8259616)
