-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20150925_113341_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.975241 to 30.778548
-each separately from 2s to 30s and from 367546us to 975530us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.392016 in 1 packets - #/No: 18/18
      9.410424 in 1 packets - #/No: 36/18
MIU packet time from 2015-09-25T11:33:46.975.241.000 to 2015-09-25T11:34:14.778.548.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.392016 in 1 packets - #/No: 18/18
      9.410424 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.3887381 to 31.3886634
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.9999493 in 1 packets - #/No: 18/18
      8.9999760 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20150925_113341_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3639775 to 4.3642665
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3919670  -- OK for SL --
   MIU time from 2015-09-25T11:33:46.975.241.000 to 2015-09-25T11:33:46.975.530.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.392016

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7562335 to 22.7565225
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4103990  -- OK for SL --
   MIU time from 2015-09-25T11:34:05.367.546.000 to 2015-09-25T11:34:05.367.835.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.410424

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.1669215 to 32.1672105
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2015-09-25T11:34:14.778.259.000 to 2015-09-25T11:34:14.778.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2015-09-25T11:33:46.975.100.750
   Max SL corrected packet time: 2015-09-25T11:34:14.778.407.750
   Min SL corrected XO rel time: 2.9752419
   Max SL corrected XO rel time: 30.7785492
   TM values from -18 to 13 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2015-09-25T11:33:46.975.100.750 (XO: 2.9752419, t=35.6°C)  Rank: 00004
   TM values: Max =    11  Min =   -16  Ave =    -4.79  Std =     3.38
   Min TM value:-16 at an offset of 58.125 usec -> 2015-09-25T11:33:46.975.158.875 (XO: 2.9753001)
   Max TM value:11 at an offset of 166.938 usec -> 2015-09-25T11:33:46.975.267.687 (XO: 2.9754089)
   Max dTM/dt [1/us]::1.40e+02 at an offset of 97.438 usec -> 2015-09-25T11:33:46.975.198.187 (XO: 2.9753394)
   Min dTM/dt [1/us]::-1.34e+02 at an offset of 166.975 usec -> 2015-09-25T11:33:46.975.267.725 (XO: 2.9754089)

 - Block number: 2 starting at 2015-09-25T11:34:05.367.405.750 (XO: 21.3675471, t=35.6°C)  Rank: 00004
   TM values: Max =    12  Min =   -18  Ave =    -4.73  Std =     3.83
   Min TM value:-18 at an offset of 209.251 usec -> 2015-09-25T11:34:05.367.615.000 (XO: 21.3677563)
   Max TM value:12 at an offset of 272.801 usec -> 2015-09-25T11:34:05.367.678.550 (XO: 21.3678199)
   Max dTM/dt [1/us]::1.51e+02 at an offset of 59.663 usec -> 2015-09-25T11:34:05.367.465.412 (XO: 21.3676068)
   Min dTM/dt [1/us]::-1.52e+02 at an offset of 272.838 usec -> 2015-09-25T11:34:05.367.678.588 (XO: 21.3678199)

 - Block number: 3 starting at 2015-09-25T11:34:14.778.118.750 (XO: 30.7782602, t=35.6°C)  Rank: 00004
   TM values: Max =    13  Min =   -18  Ave =    -4.80  Std =     3.73
   Min TM value:-18 at an offset of 115.750 usec -> 2015-09-25T11:34:14.778.234.500 (XO: 30.7783760)
   Max TM value:13 at an offset of 92.100 usec -> 2015-09-25T11:34:14.778.210.850 (XO: 30.7783523)
   Max dTM/dt [1/us]::1.37e+02 at an offset of 89.950 usec -> 2015-09-25T11:34:14.778.208.700 (XO: 30.7783502)
   Min dTM/dt [1/us]::-1.49e+02 at an offset of 90.038 usec -> 2015-09-25T11:34:14.778.208.787 (XO: 30.7783502)
