-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Oct  3 03:03:51 2024 UTC
-- File 20240827_045656_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.500064 to 26.240485
-each separately from 11s to 26s and from 240196us to 500353us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.739843 in 1 packets: 18
MIU packet time from 2024-08-27T04:57:10.500.064.000 to 2024-08-27T04:57:25.240.485.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.739843 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.6838353 to 27.6837838
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999485 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240827_045656_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1839006 to 13.1841896
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7397910  -- OK for SL --
   MIU time from 2024-08-27T04:57:10.500.064.000 to 2024-08-27T04:57:10.500.353.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.739843

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.9239806 to 27.9242696
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-27T04:57:25.240.196.000 to 2024-08-27T04:57:25.240.485.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.183900600 corrected by 40bit 10MHz counter of 12.683835300 obtained at 1Hz counter of 11. Result: 11.500067050
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-08-27T04:57:10.499.923.725
   Max 1st sample MIU time among all SL packets: 2024-08-27T04:57:25.240.344.725
   Min 1st sample XO time among all SL packets: 11.499926775
   Max 1st sample XO time among all SL packets: 26.240346366
   TM values from -252 to 228 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-27T04:57:10.499.923.725 (XO: 11.499926775 - 11.500232776, t=38.8 degC)  Rank: 00080
   TM values: Max =   228  Min =  -252  Ave =    -4.37  Std =    23.37
   Min TM value:-252 at an offset of 130.338 usec -> 2024-08-27T04:57:10.500.054.062 (XO: 11.500057113)
   Max TM value:228 at an offset of 125.938 usec -> 2024-08-27T04:57:10.500.049.662 (XO: 11.500052713)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 130.500 usec -> 2024-08-27T04:57:10.500.054.225 (XO: 11.500057275)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 127.888 usec -> 2024-08-27T04:57:10.500.051.612 (XO: 11.500054663)

 - Block number: 2 starting at 2024-08-27T04:57:25.240.055.725 (XO: 26.240057365 - 26.240363366, t=38.8 degC)  Rank: 00077
   TM values: Max =   170  Min =  -243  Ave =    -4.32  Std =    19.73
   Min TM value:-243 at an offset of 126.075 usec -> 2024-08-27T04:57:25.240.181.800 (XO: 26.240183441)
   Max TM value:170 at an offset of 125.713 usec -> 2024-08-27T04:57:25.240.181.437 (XO: 26.240183078)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 128.575 usec -> 2024-08-27T04:57:25.240.184.300 (XO: 26.240185941)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 125.888 usec -> 2024-08-27T04:57:25.240.181.612 (XO: 26.240183253)
