-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Sep 11 17:09:40 2024 UTC
-- File 20240805_205525_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.891493 to 21.231573
-each separately from 12s to 21s and from 231284us to 891782us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.339502 in 1 packets: 18
MIU packet time from 2024-08-05T20:55:40.891.493.000 to 2024-08-05T20:55:49.231.573.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.339502 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.6032537 to 22.6032224
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999687 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240805_205525_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4947474 to 14.4950364
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3394710  -- OK for SL --
   MIU time from 2024-08-05T20:55:40.891.493.000 to 2024-08-05T20:55:40.891.782.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.339502

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.8345074 to 22.8347964
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-08-05T20:55:49.231.284.000 to 2024-08-05T20:55:49.231.573.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.494747400 corrected by 40bit 10MHz counter of 13.603253700 obtained at 1Hz counter of 12. Result: 12.891496820
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-08-05T20:55:40.891.352.725
   Max 1st sample MIU time among all SL packets: 2024-08-05T20:55:49.231.432.725
   Min 1st sample XO time among all SL packets: 12.891356545
   Max 1st sample XO time among all SL packets: 21.231434535
   TM values from -286 to 244 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-08-05T20:55:40.891.352.725 (XO: 12.891356545 - 12.891662546, t=38.8 degC)  Rank: 00081
   TM values: Max =   244  Min =  -270  Ave =    -4.43  Std =    22.15
   Min TM value:-270 at an offset of 90.250 usec -> 2024-08-05T20:55:40.891.442.975 (XO: 12.891446795)
   Max TM value:244 at an offset of 88.100 usec -> 2024-08-05T20:55:40.891.440.825 (XO: 12.891444645)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 92.575 usec -> 2024-08-05T20:55:40.891.445.300 (XO: 12.891449120)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 90.050 usec -> 2024-08-05T20:55:40.891.442.775 (XO: 12.891446595)

 - Block number: 2 starting at 2024-08-05T20:55:49.231.143.725 (XO: 21.231145534 - 21.231451535, t=38.8 degC)  Rank: 00084
   TM values: Max =   233  Min =  -286  Ave =    -4.50  Std =    24.27
   Min TM value:-286 at an offset of 111.738 usec -> 2024-08-05T20:55:49.231.255.462 (XO: 21.231257272)
   Max TM value:233 at an offset of 111.438 usec -> 2024-08-05T20:55:49.231.255.162 (XO: 21.231256972)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 109.463 usec -> 2024-08-05T20:55:49.231.253.187 (XO: 21.231254997)
   Min dTM/dt [1/us]:-2.80e+03 at an offset of 111.575 usec -> 2024-08-05T20:55:49.231.255.300 (XO: 21.231257109)
