-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  5 17:54:43 2024 UTC
-- File 20240623_105826_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000087 to 90.066158
-each separately from 0s to 90s and from 87us to 66158us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
      15.010706 in 2 packets: 36, 108
      15.010659 in 2 packets: 54, 90
      15.010707 in 1 packets: 72
MIU packet time from 2024-06-23T10:58:29.000.087.000 to 2024-06-23T10:59:59.066.158.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
      15.010706 in 2 packets: 36, 108
      15.010659 in 2 packets: 54, 90
      15.010707 in 1 packets: 72
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4314545 to 91.4311440
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999482 in 3 packets: 18, 36, 90
      14.9999483 in 3 packets: 54, 72, 108
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      BF -> 34.0 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 126 packets (all packets)
Number of generated alerts: 
      0 in 126 packets (all packets)
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20240623_105826_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4315424 to 1.4318314
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-06-23T10:58:29.000.087.000 to 2024-06-23T10:58:29.000.376.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010611

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4423904 to 16.4426794
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-06-23T10:58:44.010.987.000 to 2024-06-23T10:58:44.011.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4533344 to 31.4536234
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-06-23T10:58:59.021.982.000 to 2024-06-23T10:58:59.022.271.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4642304 to 46.4645194
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-06-23T10:59:14.032.930.000 to 2024-06-23T10:59:14.033.219.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.4751744 to 61.4754634
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-06-23T10:59:29.043.926.000 to 2024-06-23T10:59:29.044.215.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.4860704 to 76.4863594
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-06-23T10:59:44.054.874.000 to 2024-06-23T10:59:44.055.163.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010706

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 91.4970144 to 91.4973034
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-06-23T10:59:59.065.869.000 to 2024-06-23T10:59:59.066.158.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.431542400 corrected by 40bit 10MHz counter of 1.431454500 obtained at 1Hz counter of 0. Result: 0.000087900
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-06-23T10:58:28.999.946.725
   Max 1st sample MIU time among all SL packets: 2024-06-23T10:59:59.066.017.725
   Min 1st sample XO time among all SL packets: -0.000052375
   Max 1st sample XO time among all SL packets: 90.066019349
   TM values from -6 to -3 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-06-23T10:58:28.999.946.725 (XO: -0.000052375 - 0.000253626, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.58  Std =     0.50
   Min TM value:-6 at an offset of 5.500 usec -> 2024-06-23T10:58:28.999.952.225 (XO: -0.000046875)
   Max TM value:-3 at an offset of 73.338 usec -> 2024-06-23T10:58:29.000.020.062 (XO: 0.000020963)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 270.201 usec -> 2024-06-23T10:58:29.000.216.925 (XO: 0.000217826)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 172.963 usec -> 2024-06-23T10:58:29.000.119.688 (XO: 0.000120588)

 - Block number: 2 starting at 2024-06-23T10:58:44.010.846.725 (XO: 15.010847463 - 15.011153464, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.62  Std =     0.49
   Min TM value:-6 at an offset of 11.688 usec -> 2024-06-23T10:58:44.010.858.412 (XO: 15.010859151)
   Max TM value:-3 at an offset of 200.288 usec -> 2024-06-23T10:58:44.011.047.013 (XO: 15.011047751)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 175.963 usec -> 2024-06-23T10:58:44.011.022.688 (XO: 15.011023426)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 175.863 usec -> 2024-06-23T10:58:44.011.022.588 (XO: 15.011023326)

 - Block number: 3 starting at 2024-06-23T10:58:59.021.841.725 (XO: 30.021843299 - 30.022149300, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.55  Std =     0.50
   Min TM value:-6 at an offset of 34.263 usec -> 2024-06-23T10:58:59.021.875.987 (XO: 30.021877562)
   Max TM value:-3 at an offset of 162.576 usec -> 2024-06-23T10:58:59.022.004.300 (XO: 30.022005875)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 64.100 usec -> 2024-06-23T10:58:59.021.905.825 (XO: 30.021907399)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 87.238 usec -> 2024-06-23T10:58:59.021.928.962 (XO: 30.021930537)

 - Block number: 4 starting at 2024-06-23T10:59:14.032.789.725 (XO: 45.032791040 - 45.033097041, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.50  Std =     0.50
   Min TM value:-6 at an offset of 41.675 usec -> 2024-06-23T10:59:14.032.831.400 (XO: 45.032832715)
   Max TM value:-3 at an offset of 1.513 usec -> 2024-06-23T10:59:14.032.791.237 (XO: 45.032792552)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 215.426 usec -> 2024-06-23T10:59:14.033.005.150 (XO: 45.033006466)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 154.988 usec -> 2024-06-23T10:59:14.032.944.713 (XO: 45.032946028)

 - Block number: 5 starting at 2024-06-23T10:59:29.043.785.725 (XO: 60.043786774 - 60.044092775, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.65  Std =     0.48
   Min TM value:-6 at an offset of 0.963 usec -> 2024-06-23T10:59:29.043.786.687 (XO: 60.043787736)
   Max TM value:-3 at an offset of 24.875 usec -> 2024-06-23T10:59:29.043.810.600 (XO: 60.043811649)
   Max dTM/dt [1/us]:1.40e+01 at an offset of 72.875 usec -> 2024-06-23T10:59:29.043.858.600 (XO: 60.043859649)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 296.951 usec -> 2024-06-23T10:59:29.044.082.676 (XO: 60.044083725)

 - Block number: 6 starting at 2024-06-23T10:59:44.054.733.725 (XO: 75.054734611 - 75.055040612, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.43  Std =     0.50
   Min TM value:-6 at an offset of 79.738 usec -> 2024-06-23T10:59:44.054.813.462 (XO: 75.054814349)
   Max TM value:-3 at an offset of 34.488 usec -> 2024-06-23T10:59:44.054.768.212 (XO: 75.054769099)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 185.638 usec -> 2024-06-23T10:59:44.054.919.363 (XO: 75.054920249)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 79.688 usec -> 2024-06-23T10:59:44.054.813.412 (XO: 75.054814299)

 - Block number: 7 starting at 2024-06-23T10:59:59.065.728.725 (XO: 90.065730348 - 90.066036350, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.37  Std =     0.48
   Min TM value:-6 at an offset of 199.638 usec -> 2024-06-23T10:59:59.065.928.363 (XO: 90.065929987)
   Max TM value:-3 at an offset of 0.825 usec -> 2024-06-23T10:59:59.065.729.550 (XO: 90.065731173)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 244.063 usec -> 2024-06-23T10:59:59.065.972.788 (XO: 90.065974412)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 205.151 usec -> 2024-06-23T10:59:59.065.933.875 (XO: 90.065935499)
