-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri May 31 23:24:20 2024 UTC
-- File 20240531_192158_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000089 to 90.066162
-each separately from 0s to 90s and from 89us to 66162us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
      15.010707 in 3 packets: 36, 72, 108
      15.010659 in 2 packets: 54, 90
MIU packet time from 2024-05-31T19:22:01.000.089.000 to 2024-05-31T19:23:31.066.162.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010611 in 1 packets: 18
      15.010707 in 3 packets: 36, 72, 108
      15.010659 in 2 packets: 54, 90
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4371280 to 91.4368165
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999480 in 2 packets: 18, 54
      14.9999481 in 3 packets: 36, 90, 108
      14.9999482 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 126 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 126 packets (all packets)
Number of generated alerts: 
      0 in 126 packets (all packets)
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20240531_192158_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4372186 to 1.4375076
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2024-05-31T19:22:01.000.089.000 to 2024-05-31T19:22:01.000.378.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010611

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4480666 to 16.4483556
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-05-31T19:22:16.010.989.000 to 2024-05-31T19:22:16.011.278.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4590106 to 31.4592996
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-05-31T19:22:31.021.985.000 to 2024-05-31T19:22:31.022.274.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4699066 to 46.4701956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-05-31T19:22:46.032.933.000 to 2024-05-31T19:22:46.033.222.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.4808506 to 61.4811396
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106070  -- OK for SL --
   MIU time from 2024-05-31T19:23:01.043.929.000 to 2024-05-31T19:23:01.044.218.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010659

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.4917466 to 76.4920356
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2024-05-31T19:23:16.054.877.000 to 2024-05-31T19:23:16.055.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010707

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 91.5026906 to 91.5029796
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-05-31T19:23:31.065.873.000 to 2024-05-31T19:23:31.066.162.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.437218600 corrected by 40bit 10MHz counter of 1.437128000 obtained at 1Hz counter of 0. Result: 0.000090600
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-05-31T19:22:00.999.948.725
   Max 1st sample MIU time among all SL packets: 2024-05-31T19:23:31.066.021.725
   Min 1st sample XO time among all SL packets: -0.000049675
   Max 1st sample XO time among all SL packets: 90.066023056
   TM values from -6 to -3 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-05-31T19:22:00.999.948.725 (XO: -0.000049675 - 0.000256326, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.47  Std =     0.50
   Min TM value:-6 at an offset of 26.325 usec -> 2024-05-31T19:22:00.999.975.050 (XO: -0.000023350)
   Max TM value:-3 at an offset of 3.350 usec -> 2024-05-31T19:22:00.999.952.075 (XO: -0.000046325)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 26.375 usec -> 2024-05-31T19:22:00.999.975.100 (XO: -0.000023300)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 94.975 usec -> 2024-05-31T19:22:01.000.043.700 (XO: 0.000045300)

 - Block number: 2 starting at 2024-05-31T19:22:16.010.848.725 (XO: 15.010850363 - 15.011156364, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.39  Std =     0.49
   Min TM value:-6 at an offset of 2.075 usec -> 2024-05-31T19:22:16.010.850.800 (XO: 15.010852438)
   Max TM value:-3 at an offset of 7.038 usec -> 2024-05-31T19:22:16.010.855.762 (XO: 15.010857401)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 286.039 usec -> 2024-05-31T19:22:16.011.134.763 (XO: 15.011136401)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 188.051 usec -> 2024-05-31T19:22:16.011.036.775 (XO: 15.011038414)

 - Block number: 3 starting at 2024-05-31T19:22:31.021.844.725 (XO: 30.021846301 - 30.022152303, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.35  Std =     0.48
   Min TM value:-6 at an offset of 48.775 usec -> 2024-05-31T19:22:31.021.893.500 (XO: 30.021895077)
   Max TM value:-3 at an offset of 53.975 usec -> 2024-05-31T19:22:31.021.898.700 (XO: 30.021900277)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 228.326 usec -> 2024-05-31T19:22:31.022.073.050 (XO: 30.022074627)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 48.725 usec -> 2024-05-31T19:22:31.021.893.450 (XO: 30.021895027)

 - Block number: 4 starting at 2024-05-31T19:22:46.032.792.725 (XO: 45.032794337 - 45.033100338, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.56  Std =     0.50
   Min TM value:-6 at an offset of 37.775 usec -> 2024-05-31T19:22:46.032.830.500 (XO: 45.032832112)
   Max TM value:-3 at an offset of 4.938 usec -> 2024-05-31T19:22:46.032.797.662 (XO: 45.032799274)
   Max dTM/dt [1/us]:1.10e+01 at an offset of 0.225 usec -> 2024-05-31T19:22:46.032.792.950 (XO: 45.032794562)
   Min dTM/dt [1/us]:-1.30e+01 at an offset of 241.226 usec -> 2024-05-31T19:22:46.033.033.950 (XO: 45.033035562)

 - Block number: 5 starting at 2024-05-31T19:23:01.043.788.725 (XO: 60.043790178 - 60.044096179, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.37  Std =     0.48
   Min TM value:-6 at an offset of 105.888 usec -> 2024-05-31T19:23:01.043.894.612 (XO: 60.043896066)
   Max TM value:-3 at an offset of 11.738 usec -> 2024-05-31T19:23:01.043.800.462 (XO: 60.043801916)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 158.563 usec -> 2024-05-31T19:23:01.043.947.288 (XO: 60.043948741)
   Min dTM/dt [1/us]:-1.50e+01 at an offset of 71.763 usec -> 2024-05-31T19:23:01.043.860.487 (XO: 60.043861941)

 - Block number: 6 starting at 2024-05-31T19:23:16.054.736.725 (XO: 75.054738117 - 75.055044118, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.36  Std =     0.48
   Min TM value:-6 at an offset of 65.288 usec -> 2024-05-31T19:23:16.054.802.012 (XO: 75.054803404)
   Max TM value:-3 at an offset of 13.225 usec -> 2024-05-31T19:23:16.054.749.950 (XO: 75.054751342)
   Max dTM/dt [1/us]:1.60e+01 at an offset of 288.464 usec -> 2024-05-31T19:23:16.055.025.188 (XO: 75.055026580)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 132.875 usec -> 2024-05-31T19:23:16.054.869.600 (XO: 75.054870992)

 - Block number: 7 starting at 2024-05-31T19:23:31.065.732.725 (XO: 90.065734055 - 90.066040056, t=34.0 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -6  Ave =    -4.42  Std =     0.49
   Min TM value:-6 at an offset of 7.400 usec -> 2024-05-31T19:23:31.065.740.125 (XO: 90.065741455)
   Max TM value:-3 at an offset of 76.013 usec -> 2024-05-31T19:23:31.065.808.737 (XO: 90.065810068)
   Max dTM/dt [1/us]:1.10e+01 at an offset of 0.788 usec -> 2024-05-31T19:23:31.065.733.512 (XO: 90.065734843)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 242.313 usec -> 2024-05-31T19:23:31.065.975.038 (XO: 90.065976368)
