-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Apr 15 23:31:51 2024 UTC
-- File 20240226_022537_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.010848 to 57.310368
-each separately from 11s to 57s and from 10848us to 431012us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.419586 in 1 packets: 18
      10.699572 in 1 packets: 36
      25.179206 in 1 packets: 54
MIU packet time from 2024-02-26T02:25:51.010.848.000 to 2024-02-26T02:26:37.310.368.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.419586 in 1 packets: 18
      10.699572 in 1 packets: 36
      25.179206 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8455257 to 58.8453669
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999654 in 1 packets: 18
      10.9999621 in 1 packets: 36
      24.9999137 in 1 packets: 54
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C0 -> 35.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20240226_022537_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.8563745 to 12.8566635
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4195510  -- OK for SL --
   MIU time from 2024-02-26T02:25:51.010.848.000 to 2024-02-26T02:25:51.011.137.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.419586

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2762145 to 23.2765035
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6995350  -- OK for SL --
   MIU time from 2024-02-26T02:26:01.430.723.000 to 2024-02-26T02:26:01.431.012.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.699572

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.9760385 to 33.9763275
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.1791190  -- OK for SL --
   MIU time from 2024-02-26T02:26:12.130.584.000 to 2024-02-26T02:26:12.130.873.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.179206

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1554465 to 59.1557355
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-26T02:26:37.310.079.000 to 2024-02-26T02:26:37.310.368.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.856374500 corrected by 40bit 10MHz counter of 12.845525700 obtained at 1Hz counter of 11. Result: 11.010848838
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-26T02:25:51.010.707.725
   Max 1st sample MIU time among all SL packets: 2024-02-26T02:26:37.310.227.725
   Min 1st sample XO time among all SL packets: 11.010708562
   Max 1st sample XO time among all SL packets: 57.310229411
   TM values from -271 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-26T02:25:51.010.707.725 (XO: 11.010708562 - 11.011014564, t=35.6 degC)  Rank: 00076
   TM values: Max =   191  Min =  -252  Ave =    -4.36  Std =    19.69
   Min TM value:-252 at an offset of 127.388 usec -> 2024-02-26T02:25:51.010.835.112 (XO: 11.010835950)
   Max TM value:191 at an offset of 127.088 usec -> 2024-02-26T02:25:51.010.834.812 (XO: 11.010835650)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 129.938 usec -> 2024-02-26T02:25:51.010.837.662 (XO: 11.010838500)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 127.225 usec -> 2024-02-26T02:25:51.010.834.950 (XO: 11.010835788)

 - Block number: 2 starting at 2024-02-26T02:26:01.430.582.725 (XO: 21.430584589 - 21.430890590, t=35.6 degC)  Rank: 00078
   TM values: Max =   219  Min =  -245  Ave =    -4.47  Std =    22.42
   Min TM value:-245 at an offset of 124.200 usec -> 2024-02-26T02:26:01.430.706.925 (XO: 21.430708789)
   Max TM value:219 at an offset of 111.900 usec -> 2024-02-26T02:26:01.430.694.625 (XO: 21.430696489)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 111.838 usec -> 2024-02-26T02:26:01.430.694.562 (XO: 21.430696427)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 112.038 usec -> 2024-02-26T02:26:01.430.694.762 (XO: 21.430696627)

 - Block number: 3 starting at 2024-02-26T02:26:12.130.443.725 (XO: 32.130445469 - 32.130751470, t=35.6 degC)  Rank: 00081
   TM values: Max =   240  Min =  -267  Ave =    -4.46  Std =    21.67
   Min TM value:-267 at an offset of 127.113 usec -> 2024-02-26T02:26:12.130.570.837 (XO: 32.130572581)
   Max TM value:240 at an offset of 126.800 usec -> 2024-02-26T02:26:12.130.570.525 (XO: 32.130572269)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 126.738 usec -> 2024-02-26T02:26:12.130.570.462 (XO: 32.130572206)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 126.950 usec -> 2024-02-26T02:26:12.130.570.675 (XO: 32.130572419)

 - Block number: 4 starting at 2024-02-26T02:26:37.309.938.725 (XO: 57.309940410 - 57.310246411, t=35.6 degC)  Rank: 00075
   TM values: Max =   213  Min =  -271  Ave =    -4.47  Std =    22.00
   Min TM value:-271 at an offset of 127.750 usec -> 2024-02-26T02:26:37.310.066.475 (XO: 57.310068160)
   Max TM value:213 at an offset of 127.500 usec -> 2024-02-26T02:26:37.310.066.225 (XO: 57.310067910)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 125.313 usec -> 2024-02-26T02:26:37.310.064.037 (XO: 57.310065723)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 127.575 usec -> 2024-02-26T02:26:37.310.066.300 (XO: 57.310067985)
