-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Tue Feb  6 05:02:08 2024 UTC
-- File 20240129_062653_1229.dat, 11344  packets found.
APID:
      1229 (EW) in 11344 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 11344 packets (all packets)
Total number of packets: 
      36720 in 11344 packets (all packets)
Packet No from 1 to 11344
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 123.259949 to 266.176724
-each separately from 123s to 266s and from 176724us to 259949us.
   Differences in neigboring packets: 
      0.000000 in 11342 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -142.916775 in 1 packets: 6528
MIU packet time from 2024-01-29T06:28:59.259.949.000 to 2024-01-29T06:31:22.176.724.000
   Differences in neigboring packets: 
      0.000000 in 11342 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -142.916775 in 1 packets: 6528
MIU orbit number: 
      60000 in 11344 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 11344 packets (all packets)
MIU orbit information / number of events: 
      0 in 11344 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 11344 packets (all packets)
Instrument anomalies: 
      0 (OK) in 11344 packets (all packets)
FPGA configuration version: 
      211 in 11344 packets (all packets)
10 MHz counter at the last 1Hz pulse from 124.6939041 to 267.6934094
   Differences in neigboring packets: 
      0.0000000 in 11342 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -142.9995053 in 1 packets: 6528
Temperature FPGA: 
      07 -> 45.9 degC in 6528 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      08 -> 44.6 degC in 4816 packets: 6529, 6530, 6531, 6532, 6533, 6534, 6535, 6536, 6537, 6538, 6539, 6540, 6541, 6542, 6543, 6544, 6545, 6546, ...
Temperature XO: 
      BF -> 34.0 degC in 11344 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 11344 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 11344 packets (all packets)
Number of external events: 
      3 in 6528 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 4816 packets: 6529, 6530, 6531, 6532, 6533, 6534, 6535, 6536, 6537, 6538, 6539, 6540, 6541, 6542, 6543, 6544, 6545, 6546, ...
Number of generated alerts: 
      3 in 6528 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 4816 packets: 6529, 6530, 6531, 6532, 6533, 6534, 6535, 6536, 6537, 6538, 6539, 6540, 6541, 6542, 6543, 6544, 6545, 6546, ...
Survey memory error: 
      0 in 11344 packets (all packets)
Event memory error: 
      0 in 11344 packets (all packets)
Configuration error: 
      0 in 11344 packets (all packets)
SDRAM CRC error: 
      0 in 11344 packets (all packets)
Radio gain: 
      0 in 11344 packets (all packets)
Radio bandwidth: 
      0 in 11344 packets (all packets)
Digital gain: 
      0 in 11344 packets (all packets)
Analog gain: 
      0 in 11344 packets (all packets)

EVENT DATA READER
-- File 20240129_062653_1229.dat, 55 EW pages found. Extra packets: 124.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 3
      3 in 6528 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 4692 packets: 6529, 6530, 6531, 6532, 6533, 6534, 6535, 6536, 6537, 6538, 6539, 6540, 6541, 6542, 6543, 6544, 6545, 6546, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 124.8983663 to 267.9256053
      Difference between the neighboring packets: 
      0.0000170 in 11218 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -143.0272390 in 1 packets: 6528
   Memory page number from 47 to 101
      Difference between the neighboring packets: 
      0 in 11165 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 32
      Difference between the neighboring packets: 
      0 in 11165 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -31 in 1 packets: 6528
   MIU event flag 
      1 in 11220 packets (all packets)
   Time triggered event flag 
      0 in 11220 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-01-29T06:28:59.204.460.637
   Min EW header abs. time found among all events            : 2024-01-29T06:28:59.259.949.000
   Max EW event (1st sample) abs. time found among all events: 2024-01-29T06:31:22.121.235.637
   Max EW header abs. time found among all events            : 2024-01-29T06:31:22.176.724.000
   TM values from -260 to 230 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 4692
    Page no. 1:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 23  ordered packets: 4692
   TM values from -246 to 230 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=123 C1MHz=0.259949
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 4691 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 124.8983663
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-01-29T06:28:59.259.949.000
   Event 1st sample abs. time : 2024-01-29T06:28:59.204.460.637)
   --- Rel. time correction for 1 points based on 254 XO calibration points at 1Hz counter from 0 to 253
   --- First point: 1Hz counter = 123, 32bit 10MHz counter = 124.898366300 corrected by 40bit 10MHz counter of 124.693904100 obtained at 1Hz counter of 123. Result: 123.204462895
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             123.204462720
   Event data descriptor XO time (reconstructed seconds from XP start)    :             123.204462895
   Sampling frequency correction factor for this event:     0.999996600

 Event no. 3 total number of packets: 6528
    Page no. 1:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 32  ordered packets: 6528
   TM values from -260 to 228 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=266 C1MHz=0.176724
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 6527 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 267.8146463
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-01-29T06:31:22.176.724.000
   Event 1st sample abs. time : 2024-01-29T06:31:22.121.235.637)
   --- Rel. time correction for 1 points based on 254 XO calibration points at 1Hz counter from 0 to 253
   --- First point: 1Hz counter = 266, 32bit 10MHz counter = 267.814646300 corrected by 40bit 10MHz counter of 254.693454300 obtained at 1Hz counter of 253. Result: 266.121236612
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             266.121236437
   Event data descriptor XO time (reconstructed seconds from XP start)    :             266.121236612
   Sampling frequency correction factor for this event:     0.999996600
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2024-01-29T06:28:59.204.460.637 to 2024-01-29T06:28:59.284.224.908 (XO: 123.204462720 - 123.284226991)
   Min TM value:-246 at an offset of 33760.615 usec -> 2024-01-29T06:28:59.238.221.251 (XO: 123.238223335)
   Max TM value:230 at an offset of 53751.070 usec -> 2024-01-29T06:28:59.258.211.707 (XO: 123.258213790)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 33763.165 usec -> 2024-01-29T06:28:59.238.223.801 (XO: 123.238225885)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 33760.452 usec -> 2024-01-29T06:28:59.238.221.089 (XO: 123.238223172)
   Med: -5.00000   Ave: -4.5022672   Std: 6.3226608

 - Event number: 3 from 2024-01-29T06:31:22.121.235.637 to 2024-01-29T06:31:22.232.212.014 (XO: 266.121236437 - 266.232212815)
   Min TM value:-260 at an offset of 54015.834 usec -> 2024-01-29T06:31:22.175.251.470 (XO: 266.175252271)
   Max TM value:228 at an offset of 54015.559 usec -> 2024-01-29T06:31:22.175.251.195 (XO: 266.175251996)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 4050.826 usec -> 2024-01-29T06:31:22.125.286.463 (XO: 266.125287263)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 54015.671 usec -> 2024-01-29T06:31:22.175.251.308 (XO: 266.175252108)
   Med: -5.00000   Ave: -4.5608892   Std: 6.4356731
