-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 23:07:47 2024 UTC
-- File 20240128_095441_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.147447 to 40.937946
-each separately from 12s to 40s and from 147447us to 937946us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.457509 in 1 packets: 18
      22.332123 in 1 packets: 36
MIU packet time from 2024-01-28T09:54:56.147.447.000 to 2024-01-28T09:55:24.937.946.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.457509 in 1 packets: 18
      22.332123 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8139459 to 41.8138497
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999794 in 1 packets: 18
      21.9999244 in 1 packets: 36
Temperature FPGA: 
      07 -> 45.9 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240128_095441_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      106 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9613942 to 13.9616832
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.4574870  -- OK for SL --
   MIU time from 2024-01-28T09:54:56.147.447.000 to 2024-01-28T09:54:56.147.736.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.457509

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      113 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.4191702 to 20.4194592
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.3320470  -- OK for SL --
   MIU time from 2024-01-28T09:55:02.605.245.000 to 2024-01-28T09:55:02.605.534.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.332123

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.7515062 to 42.7517952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-28T09:55:24.937.657.000 to 2024-01-28T09:55:24.937.946.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.961394200 corrected by 40bit 10MHz counter of 13.813945900 obtained at 1Hz counter of 12. Result: 12.147448816
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-28T09:54:56.147.306.725
   Max 1st sample MIU time among all SL packets: 2024-01-28T09:55:24.937.805.725
   Min 1st sample XO time among all SL packets: 12.147308541
   Max 1st sample XO time among all SL packets: 40.937808414
   TM values from -1019 to 1059 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-28T09:54:56.147.306.725 (XO: 12.147308541 - 12.147614542, t=34.0 degC)  Rank: 00106
   TM values: Max =  1032  Min = -1008  Ave =    -4.55  Std =    34.24
   Min TM value:-1008 at an offset of 107.775 usec -> 2024-01-28T09:54:56.147.414.500 (XO: 12.147416316)
   Max TM value:1032 at an offset of 107.575 usec -> 2024-01-28T09:54:56.147.414.300 (XO: 12.147416116)
   Max dTM/dt [1/us]:1.53e+04 at an offset of 107.550 usec -> 2024-01-28T09:54:56.147.414.275 (XO: 12.147416091)
   Min dTM/dt [1/us]:-1.05e+04 at an offset of 107.738 usec -> 2024-01-28T09:54:56.147.414.462 (XO: 12.147416278)

 - Block number: 2 starting at 2024-01-28T09:55:02.605.104.725 (XO: 18.605106743 - 18.605412744, t=34.0 degC)  Rank: 00113
   TM values: Max =  1059  Min = -1019  Ave =    -4.54  Std =    33.92
   Min TM value:-1019 at an offset of 93.338 usec -> 2024-01-28T09:55:02.605.198.062 (XO: 18.605200081)
   Max TM value:1059 at an offset of 93.138 usec -> 2024-01-28T09:55:02.605.197.862 (XO: 18.605199881)
   Max dTM/dt [1/us]:1.46e+04 at an offset of 93.125 usec -> 2024-01-28T09:55:02.605.197.850 (XO: 18.605199868)
   Min dTM/dt [1/us]:-1.07e+04 at an offset of 93.313 usec -> 2024-01-28T09:55:02.605.198.037 (XO: 18.605200056)

 - Block number: 3 starting at 2024-01-28T09:55:24.937.516.725 (XO: 40.937519413 - 40.937825414, t=34.0 degC)  Rank: 00079
   TM values: Max =   239  Min =  -229  Ave =    -4.46  Std =    19.78
   Min TM value:-229 at an offset of 127.300 usec -> 2024-01-28T09:55:24.937.644.025 (XO: 40.937646713)
   Max TM value:239 at an offset of 129.700 usec -> 2024-01-28T09:55:24.937.646.425 (XO: 40.937649113)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 127.400 usec -> 2024-01-28T09:55:24.937.644.125 (XO: 40.937646813)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 129.788 usec -> 2024-01-28T09:55:24.937.646.512 (XO: 40.937649201)
