-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 11:20:24 2024 UTC
-- File 20231025_231300_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.194490 to 21.183481
-each separately from 12s to 21s and from 183192us to 194779us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.988413 in 1 packets: 18
MIU packet time from 2023-10-25T23:13:15.194.490.000 to 2023-10-25T23:13:24.183.481.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.988413 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8645428 to 22.8645122
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999694 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231025_231300_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.0590337 to 14.0593227
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9883830  -- OK for SL --
   MIU time from 2023-10-25T23:13:15.194.490.000 to 2023-10-25T23:13:15.194.779.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.988413

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      120 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0477057 to 23.0479947
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-25T23:13:24.183.192.000 to 2023-10-25T23:13:24.183.481.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.059033700 corrected by 40bit 10MHz counter of 13.864542800 obtained at 1Hz counter of 12. Result: 12.194491542
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-25T23:13:15.194.349.725
   Max 1st sample MIU time among all SL packets: 2023-10-25T23:13:24.183.340.725
   Min 1st sample XO time among all SL packets: 12.194351266
   Max 1st sample XO time among all SL packets: 21.183342848
   TM values from -903 to 1050 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-25T23:13:15.194.349.725 (XO: 12.194351266 - 12.194657267, t=37.2 degC)  Rank: 00076
   TM values: Max =   203  Min =  -214  Ave =    -4.67  Std =    23.67
   Min TM value:-214 at an offset of 117.525 usec -> 2023-10-25T23:13:15.194.467.250 (XO: 12.194468792)
   Max TM value:203 at an offset of 117.225 usec -> 2023-10-25T23:13:15.194.466.950 (XO: 12.194468492)
   Max dTM/dt [1/us]:1.34e+03 at an offset of 185.013 usec -> 2023-10-25T23:13:15.194.534.738 (XO: 12.194536279)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 174.688 usec -> 2023-10-25T23:13:15.194.524.413 (XO: 12.194525954)

 - Block number: 2 starting at 2023-10-25T23:13:24.183.051.725 (XO: 21.183053847 - 21.183359848, t=37.2 degC)  Rank: 00120
   TM values: Max =  1050  Min =  -903  Ave =    -4.60  Std =    32.53
   Min TM value:-903 at an offset of 105.325 usec -> 2023-10-25T23:13:24.183.157.050 (XO: 21.183159173)
   Max TM value:1050 at an offset of 105.388 usec -> 2023-10-25T23:13:24.183.157.112 (XO: 21.183159235)
   Max dTM/dt [1/us]:1.50e+04 at an offset of 105.363 usec -> 2023-10-25T23:13:24.183.157.087 (XO: 21.183159210)
   Min dTM/dt [1/us]:-9.72e+03 at an offset of 105.550 usec -> 2023-10-25T23:13:24.183.157.275 (XO: 21.183159398)
