-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 05:32:59 2024 UTC
-- File 20231025_055528_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.462785 to 22.691110
-each separately from 2s to 22s and from 462785us to 691110us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.227747 in 1 packets: 18
MIU packet time from 2023-10-25T05:55:33.462.785.000 to 2023-10-25T05:55:53.691.110.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.227747 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8251451 to 23.8250769
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999318 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231025_055528_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2879308 to 4.2882198
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.2276790  -- OK for SL --
   MIU time from 2023-10-25T05:55:33.462.785.000 to 2023-10-25T05:55:33.463.074.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.227747

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5158988 to 24.5161878
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-25T05:55:53.690.821.000 to 2023-10-25T05:55:53.691.110.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.287930800 corrected by 40bit 10MHz counter of 3.825145100 obtained at 1Hz counter of 2. Result: 2.462787273
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-25T05:55:33.462.644.725
   Max 1st sample MIU time among all SL packets: 2023-10-25T05:55:53.690.969.725
   Min 1st sample XO time among all SL packets: 2.462646998
   Max 1st sample XO time among all SL packets: 22.690972974
   TM values from -248 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-25T05:55:33.462.644.725 (XO: 2.462646998 - 2.462952999, t=37.2 degC)  Rank: 00083
   TM values: Max =   222  Min =  -241  Ave =    -4.50  Std =    27.95
   Min TM value:-241 at an offset of 124.625 usec -> 2023-10-25T05:55:33.462.769.350 (XO: 2.462771623)
   Max TM value:222 at an offset of 207.563 usec -> 2023-10-25T05:55:33.462.852.288 (XO: 2.462854561)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 205.288 usec -> 2023-10-25T05:55:33.462.850.013 (XO: 2.462852286)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 207.688 usec -> 2023-10-25T05:55:33.462.852.413 (XO: 2.462854686)

 - Block number: 2 starting at 2023-10-25T05:55:53.690.680.725 (XO: 22.690683973 - 22.690989974, t=37.2 degC)  Rank: 00082
   TM values: Max =   212  Min =  -248  Ave =    -4.63  Std =    19.16
   Min TM value:-248 at an offset of 90.025 usec -> 2023-10-25T05:55:53.690.770.750 (XO: 22.690773999)
   Max TM value:212 at an offset of 92.125 usec -> 2023-10-25T05:55:53.690.772.850 (XO: 22.690776099)
   Max dTM/dt [1/us]:2.31e+03 at an offset of 90.125 usec -> 2023-10-25T05:55:53.690.770.850 (XO: 22.690774099)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 92.238 usec -> 2023-10-25T05:55:53.690.772.962 (XO: 22.690776211)
