-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 30 00:18:02 2024 UTC
-- File 20231022_110806_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.364924 to 28.005645
-each separately from 9s to 28s and from 5356us to 365213us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.640143 in 1 packets: 18
MIU packet time from 2023-10-22T11:08:18.364.924.000 to 2023-10-22T11:08:37.005.645.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.640143 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8449143 to 29.8448499
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999356 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231022_110806_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.2098393 to 11.2101283
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6400790  -- OK for SL --
   MIU time from 2023-10-22T11:08:18.364.924.000 to 2023-10-22T11:08:18.365.213.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.640143

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8502073 to 29.8504963
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-22T11:08:37.005.356.000 to 2023-10-22T11:08:37.005.645.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.209839300 corrected by 40bit 10MHz counter of 10.844914300 obtained at 1Hz counter of 9. Result: 9.364926241
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-22T11:08:18.364.783.725
   Max 1st sample MIU time among all SL packets: 2023-10-22T11:08:37.005.504.725
   Min 1st sample XO time among all SL packets: 9.364785965
   Max 1st sample XO time among all SL packets: 28.005506144
   TM values from -246 to 236 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-22T11:08:18.364.783.725 (XO: 9.364785965 - 9.365091966, t=37.2 degC)  Rank: 00083
   TM values: Max =   236  Min =  -246  Ave =    -4.61  Std =    19.39
   Min TM value:-246 at an offset of 133.925 usec -> 2023-10-22T11:08:18.364.917.650 (XO: 9.364919891)
   Max TM value:236 at an offset of 133.663 usec -> 2023-10-22T11:08:18.364.917.387 (XO: 9.364919628)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 136.500 usec -> 2023-10-22T11:08:18.364.920.225 (XO: 9.364922466)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 133.775 usec -> 2023-10-22T11:08:18.364.917.500 (XO: 9.364919741)

 - Block number: 2 starting at 2023-10-22T11:08:37.005.215.725 (XO: 28.005217143 - 28.005523144, t=37.2 degC)  Rank: 00076
   TM values: Max =   229  Min =  -232  Ave =    -4.50  Std =    23.24
   Min TM value:-232 at an offset of 118.650 usec -> 2023-10-22T11:08:37.005.334.375 (XO: 28.005335793)
   Max TM value:229 at an offset of 115.900 usec -> 2023-10-22T11:08:37.005.331.625 (XO: 28.005333043)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 118.763 usec -> 2023-10-22T11:08:37.005.334.487 (XO: 28.005335906)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 116.025 usec -> 2023-10-22T11:08:37.005.331.750 (XO: 28.005333168)
