-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 29 18:19:21 2024 UTC
-- File 20231021_161402_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.243866 to 49.589491
-each separately from 9s to 49s and from 243866us to 805591us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.561147 in 1 packets: 18
      12.671852 in 1 packets: 36
      14.111470 in 1 packets: 54
MIU packet time from 2023-10-21T16:14:14.243.866.000 to 2023-10-21T16:14:54.589.491.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.561147 in 1 packets: 18
      12.671852 in 1 packets: 36
      14.111470 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8332037 to 50.8330673
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999557 in 1 packets: 18
      12.9999556 in 1 packets: 36
      13.9999523 in 1 packets: 54
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C2 -> 38.8 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20231021_161402_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.0770708 to 11.0773598
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.5611030  -- OK for SL --
   MIU time from 2023-10-21T16:14:14.243.866.000 to 2023-10-21T16:14:14.244.155.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.561147

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6384628 to 24.6387518
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6718070  -- OK for SL --
   MIU time from 2023-10-21T16:14:27.805.302.000 to 2023-10-21T16:14:27.805.591.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.671852

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.3105588 to 37.3108478
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.1114230  -- OK for SL --
   MIU time from 2023-10-21T16:14:40.477.443.000 to 2023-10-21T16:14:40.477.732.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.111470

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.4222708 to 51.4225598
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-21T16:14:54.589.202.000 to 2023-10-21T16:14:54.589.491.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.077070800 corrected by 40bit 10MHz counter of 10.833203700 obtained at 1Hz counter of 9. Result: 9.243867929
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-21T16:14:14.243.725.725
   Max 1st sample MIU time among all SL packets: 2023-10-21T16:14:54.589.350.725
   Min 1st sample XO time among all SL packets: 9.243727654
   Max 1st sample XO time among all SL packets: 49.589354288
   TM values from -275 to 218 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-21T16:14:14.243.725.725 (XO: 9.243727654 - 9.244033655, t=38.8 degC)  Rank: 00079
   TM values: Max =   206  Min =  -258  Ave =    -4.36  Std =    19.17
   Min TM value:-258 at an offset of 104.000 usec -> 2023-10-21T16:14:14.243.829.725 (XO: 9.243831654)
   Max TM value:206 at an offset of 103.688 usec -> 2023-10-21T16:14:14.243.829.412 (XO: 9.243831342)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 106.488 usec -> 2023-10-21T16:14:14.243.832.212 (XO: 9.243834142)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 103.813 usec -> 2023-10-21T16:14:14.243.829.537 (XO: 9.243831467)

 - Block number: 2 starting at 2023-10-21T16:14:27.805.161.725 (XO: 22.805165863 - 22.805471864, t=38.8 degC)  Rank: 00084
   TM values: Max =   218  Min =  -275  Ave =    -4.47  Std =    21.03
   Min TM value:-275 at an offset of 128.950 usec -> 2023-10-21T16:14:27.805.290.675 (XO: 22.805294813)
   Max TM value:218 at an offset of 128.638 usec -> 2023-10-21T16:14:27.805.290.362 (XO: 22.805294501)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 126.650 usec -> 2023-10-21T16:14:27.805.288.375 (XO: 22.805292513)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 128.750 usec -> 2023-10-21T16:14:27.805.290.475 (XO: 22.805294613)

 - Block number: 3 starting at 2023-10-21T16:14:40.477.302.725 (XO: 35.477305148 - 35.477611149, t=38.8 degC)  Rank: 00079
   TM values: Max =   201  Min =  -234  Ave =    -4.47  Std =    23.13
   Min TM value:-234 at an offset of 104.638 usec -> 2023-10-21T16:14:40.477.407.362 (XO: 35.477409786)
   Max TM value:201 at an offset of 106.738 usec -> 2023-10-21T16:14:40.477.409.462 (XO: 35.477411886)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 104.750 usec -> 2023-10-21T16:14:40.477.407.475 (XO: 35.477409898)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 106.850 usec -> 2023-10-21T16:14:40.477.409.575 (XO: 35.477411998)

 - Block number: 4 starting at 2023-10-21T16:14:54.589.061.725 (XO: 49.589065287 - 49.589371288, t=38.8 degC)  Rank: 00082
   TM values: Max =   191  Min =  -242  Ave =    -4.48  Std =    21.78
   Min TM value:-242 at an offset of 128.425 usec -> 2023-10-21T16:14:54.589.190.150 (XO: 49.589193712)
   Max TM value:191 at an offset of 128.088 usec -> 2023-10-21T16:14:54.589.189.812 (XO: 49.589193375)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 131.013 usec -> 2023-10-21T16:14:54.589.192.737 (XO: 49.589196300)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 171.413 usec -> 2023-10-21T16:14:54.589.233.138 (XO: 49.589236700)
