-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 29 11:53:24 2024 UTC
-- File 20231020_154645_1226.dat, 28  packets found.
APID:
      1226 (SR) in 28 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 28 packets (all packets)
Total number of packets: 
      18 in 28 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 18
Hz and MHz counters from 0.000000 to 22.516454
-each separately from 0s to 22s and from 0us to 848567us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 13 packets: 2, 4, 5, 8, 10, 12, 14, 16, 18, 20, 22, 24, 26
      0.833944 in 9 packets: 3, 6, 9, 11, 15, 17, 21, 23, 27
      0.833940 in 4 packets: 7, 13, 19, 25
MIU packet time from 2023-10-20T15:46:48.000.000.000 to 2023-10-20T15:47:10.516.454.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 13 packets: 2, 4, 5, 8, 10, 12, 14, 16, 18, 20, 22, 24, 26
      0.833944 in 9 packets: 3, 6, 9, 11, 15, 17, 21, 23, 27
      0.833940 in 4 packets: 7, 13, 19, 25
MIU orbit number: 
      60000 in 28 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 28 packets (all packets)
MIU orbit information / number of events: 
      0 in 28 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 28 packets (all packets)
Instrument anomalies: 
      0 (OK) in 28 packets (all packets)
FPGA configuration version: 
      211 in 28 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8512876 to 23.8512128
   Differences in neigboring packets: 
      0.0000000 in 5 packets: 1, 7, 13, 19, 25
      0.9999966 in 18 packets: 2, 3, 5, 6, 8, 9, 10, 11, 12, 14, 15, 17, 18, 20, 22, 24, 26, 27
      0.9999967 in 2 packets: 4, 21
      0.9999965 in 2 packets: 16, 23
   XO PPM: 
      -3.4 in 18 packets: 2, 3, 5, 6, 8, 9, 10, 11, 12, 14, 15, 17, 18, 20, 22, 24, 26, 27
      -3.3 in 2 packets: 4, 21
      -3.5 in 2 packets: 16, 23
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 4, 5, 6, 9, 11, 12, 13, 14, 17, 18, 21, 22, 23, 25, 26, 27, 28
      04 -> 49.8 degC in 9 packets: 2, 3, 7, 8, 10, 15, 16, 19, 20
      02 -> 52.4 degC in 1 packets: 24
Temperature XO: 
      C2 -> 38.8 degC in 28 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 28 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 28 packets (all packets)
Number of external events: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 22 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21, 22, 23, 24, ...
Number of generated alerts: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 22 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21, 22, 23, 24, ...
Survey memory error: 
      0 in 28 packets (all packets)
Event memory error: 
      0 in 28 packets (all packets)
Configuration error: 
      0 in 28 packets (all packets)
SDRAM CRC error: 
      0 in 28 packets (all packets)
Radio gain: 
      0 in 28 packets (all packets)
Radio bandwidth: 
      0 in 28 packets (all packets)
Digital gain: 
      0 in 28 packets (all packets)
Analog gain: 
      0 in 28 packets (all packets)
 
SURVEY DATA READER
-- File 20231020_154645_1226.dat, 1 SR data blocks found. Extra packets: 10.
   ERROR -  Extra packets
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8512877 to 16.0282677
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-20T15:46:48.000.000.000 to 2023-10-20T15:47:02.177.027.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 5, 8, 10, 12, 14, 16
      0.833944 in 6 packets: 3, 6, 9, 11, 15, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 22 XO calibration points at 1Hz counter from 0 to 21
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.851287700 corrected by 40bit 10MHz counter of 1.851287600 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SR packets: 2023-10-20T15:46:47.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-10-20T15:47:02.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028227
   TM values from -80 to 58 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-10-20T15:46:47.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.44  Std =     6.07

 - Packet number: 2 starting at 2023-10-20T15:46:48.833.938.925 (XO: 0.833942860 - 0.833959860, t=38.8 degC)
   TM values: Max =    17  Min =   -24  Ave =    -4.45  Std =     6.01

 - Packet number: 3 starting at 2023-10-20T15:46:49.667.881.925 (XO: 1.667885696 - 1.667902696, t=38.8 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.46  Std =     5.87

 - Packet number: 4 starting at 2023-10-20T15:46:50.501.825.925 (XO: 2.501828481 - 2.501845481, t=38.8 degC)
   TM values: Max =    17  Min =   -30  Ave =    -4.74  Std =     7.85

 - Packet number: 5 starting at 2023-10-20T15:46:51.335.768.925 (XO: 3.335771267 - 3.335788267, t=38.8 degC)
   TM values: Max =    23  Min =   -35  Ave =    -4.29  Std =     8.82

 - Packet number: 6 starting at 2023-10-20T15:46:52.169.711.925 (XO: 4.169714102 - 4.169731102, t=38.8 degC)
   TM values: Max =    16  Min =   -29  Ave =    -4.54  Std =     7.76

 - Packet number: 7 starting at 2023-10-20T15:46:53.003.655.925 (XO: 5.003656937 - 5.003673937, t=38.8 degC)
   TM values: Max =    20  Min =   -27  Ave =    -4.38  Std =     6.85

 - Packet number: 8 starting at 2023-10-20T15:46:53.837.595.925 (XO: 5.837599773 - 5.837616773, t=38.8 degC)
   TM values: Max =    58  Min =   -80  Ave =    -4.39  Std =    17.54

 - Packet number: 9 starting at 2023-10-20T15:46:54.671.538.925 (XO: 6.671542608 - 6.671559608, t=38.8 degC)
   TM values: Max =    14  Min =   -27  Ave =    -4.35  Std =     6.81

 - Packet number: 10 starting at 2023-10-20T15:46:55.505.482.925 (XO: 7.505485444 - 7.505502444, t=38.8 degC)
   TM values: Max =    16  Min =   -28  Ave =    -4.54  Std =     7.05

 - Packet number: 11 starting at 2023-10-20T15:46:56.339.425.925 (XO: 8.339428279 - 8.339445279, t=38.8 degC)
   TM values: Max =    21  Min =   -28  Ave =    -4.16  Std =     7.28

 - Packet number: 12 starting at 2023-10-20T15:46:57.173.369.925 (XO: 9.173371114 - 9.173388115, t=38.8 degC)
   TM values: Max =    21  Min =   -31  Ave =    -4.38  Std =     7.59

 - Packet number: 13 starting at 2023-10-20T15:46:58.007.312.925 (XO: 10.007313950 - 10.007330950, t=38.8 degC)
   TM values: Max =    19  Min =   -30  Ave =    -4.58  Std =     7.85

 - Packet number: 14 starting at 2023-10-20T15:46:58.841.252.925 (XO: 10.841256785 - 10.841273785, t=38.8 degC)
   TM values: Max =    19  Min =   -27  Ave =    -4.28  Std =     7.45

 - Packet number: 15 starting at 2023-10-20T15:46:59.675.195.925 (XO: 11.675199621 - 11.675216621, t=38.8 degC)
   TM values: Max =    21  Min =   -26  Ave =    -4.56  Std =     7.55

 - Packet number: 16 starting at 2023-10-20T15:47:00.509.139.925 (XO: 12.509142507 - 12.509159507, t=38.8 degC)
   TM values: Max =    24  Min =   -27  Ave =    -4.52  Std =     8.29

 - Packet number: 17 starting at 2023-10-20T15:47:01.343.082.925 (XO: 13.343085391 - 13.343102392, t=38.8 degC)
   TM values: Max =    17  Min =   -26  Ave =    -4.43  Std =     7.36

 - Packet number: 18 starting at 2023-10-20T15:47:02.177.026.925 (XO: 14.177028227 - 14.177045227, t=38.8 degC)
   TM values: Max =    18  Min =   -32  Ave =    -4.35  Std =     7.41
