-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 29 11:33:07 2024 UTC
-- File 20231020_151113_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.336945 to 43.538163
-each separately from 5s to 43s and from 217675us to 538163us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.880441 in 1 packets: 18
      16.319910 in 1 packets: 36
MIU packet time from 2023-10-20T15:11:21.336.945.000 to 2023-10-20T15:11:59.538.163.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.880441 in 1 packets: 18
      16.319910 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8277319 to 44.8276029
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999253 in 1 packets: 18
      15.9999457 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231020_151113_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.1646776 to 7.1649666
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8803670  -- OK for SL --
   MIU time from 2023-10-20T15:11:21.336.945.000 to 2023-10-20T15:11:21.337.234.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.880441

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0453336 to 29.0456226
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3198550  -- OK for SL --
   MIU time from 2023-10-20T15:11:43.217.675.000 to 2023-10-20T15:11:43.217.964.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.319910

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.3654776 to 45.3657666
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-20T15:11:59.537.874.000 to 2023-10-20T15:11:59.538.163.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.164677600 corrected by 40bit 10MHz counter of 6.827731900 obtained at 1Hz counter of 5. Result: 5.336946846
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-20T15:11:21.336.804.725
   Max 1st sample MIU time among all SL packets: 2023-10-20T15:11:59.538.022.725
   Min 1st sample XO time among all SL packets: 5.336806570
   Max 1st sample XO time among all SL packets: 43.538025254
   TM values from -270 to 207 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-20T15:11:21.336.804.725 (XO: 5.336806570 - 5.337112571, t=38.8 degC)  Rank: 00081
   TM values: Max =   207  Min =  -270  Ave =    -4.56  Std =    24.28
   Min TM value:-270 at an offset of 128.575 usec -> 2023-10-20T15:11:21.336.933.300 (XO: 5.336935146)
   Max TM value:207 at an offset of 128.275 usec -> 2023-10-20T15:11:21.336.933.000 (XO: 5.336934846)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 131.013 usec -> 2023-10-20T15:11:21.336.935.737 (XO: 5.336937583)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 128.413 usec -> 2023-10-20T15:11:21.336.933.137 (XO: 5.336934983)

 - Block number: 2 starting at 2023-10-20T15:11:43.217.534.725 (XO: 27.217536865 - 27.217842866, t=38.8 degC)  Rank: 00085
   TM values: Max =   193  Min =  -262  Ave =    -4.43  Std =    19.58
   Min TM value:-262 at an offset of 126.288 usec -> 2023-10-20T15:11:43.217.661.012 (XO: 27.217663153)
   Max TM value:193 at an offset of 125.988 usec -> 2023-10-20T15:11:43.217.660.712 (XO: 27.217662853)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 128.750 usec -> 2023-10-20T15:11:43.217.663.475 (XO: 27.217665615)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 126.125 usec -> 2023-10-20T15:11:43.217.660.850 (XO: 27.217662990)

 - Block number: 3 starting at 2023-10-20T15:11:59.537.733.725 (XO: 43.537736253 - 43.538042254, t=38.8 degC)  Rank: 00079
   TM values: Max =   189  Min =  -245  Ave =    -4.58  Std =    18.12
   Min TM value:-245 at an offset of 124.350 usec -> 2023-10-20T15:11:59.537.858.075 (XO: 43.537860604)
   Max TM value:189 at an offset of 124.063 usec -> 2023-10-20T15:11:59.537.857.787 (XO: 43.537860316)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 126.900 usec -> 2023-10-20T15:11:59.537.860.625 (XO: 43.537863154)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 124.200 usec -> 2023-10-20T15:11:59.537.857.925 (XO: 43.537860454)
