-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 29 10:48:55 2024 UTC
-- File 20231020_135505_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.998278 to 35.751037
-each separately from 5s to 35s and from 499288us to 998567us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.500721 in 1 packets: 18
      20.251171 in 1 packets: 36
MIU packet time from 2023-10-20T13:55:13.998.278.000 to 2023-10-20T13:55:43.751.037.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.500721 in 1 packets: 18
      20.251171 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8340563 to 36.8339548
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999662 in 1 packets: 18
      19.9999323 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231020_135505_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8323359 to 7.8326249
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5006870  -- OK for SL --
   MIU time from 2023-10-20T13:55:13.998.278.000 to 2023-10-20T13:55:13.998.567.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.500721

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3333119 to 17.3336009
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.2511030  -- OK for SL --
   MIU time from 2023-10-20T13:55:23.499.288.000 to 2023-10-20T13:55:23.499.577.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.251171

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.5847039 to 37.5849929
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-20T13:55:43.750.748.000 to 2023-10-20T13:55:43.751.037.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.832335900 corrected by 40bit 10MHz counter of 6.834056300 obtained at 1Hz counter of 5. Result: 5.998282994
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-20T13:55:13.998.137.725
   Max 1st sample MIU time among all SL packets: 2023-10-20T13:55:43.750.896.725
   Min 1st sample XO time among all SL packets: 5.998142719
   Max 1st sample XO time among all SL packets: 35.750900303
   TM values from -221 to 194 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-20T13:55:13.998.137.725 (XO: 5.998142719 - 5.998448720, t=38.8 degC)  Rank: 00079
   TM values: Max =   183  Min =  -208  Ave =    -4.40  Std =    26.04
   Min TM value:-208 at an offset of 169.826 usec -> 2023-10-20T13:55:13.998.307.550 (XO: 5.998312544)
   Max TM value:183 at an offset of 125.613 usec -> 2023-10-20T13:55:13.998.263.337 (XO: 5.998268332)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 128.325 usec -> 2023-10-20T13:55:13.998.266.050 (XO: 5.998271044)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 182.388 usec -> 2023-10-20T13:55:13.998.320.113 (XO: 5.998325107)

 - Block number: 2 starting at 2023-10-20T13:55:23.499.147.725 (XO: 15.499150822 - 15.499456823, t=38.8 degC)  Rank: 00074
   TM values: Max =   194  Min =  -204  Ave =    -4.59  Std =    20.37
   Min TM value:-204 at an offset of 129.113 usec -> 2023-10-20T13:55:23.499.276.837 (XO: 15.499279935)
   Max TM value:194 at an offset of 128.838 usec -> 2023-10-20T13:55:23.499.276.562 (XO: 15.499279660)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 189.413 usec -> 2023-10-20T13:55:23.499.337.138 (XO: 15.499340235)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 128.950 usec -> 2023-10-20T13:55:23.499.276.675 (XO: 15.499279773)

 - Block number: 3 starting at 2023-10-20T13:55:43.750.607.725 (XO: 35.750611302 - 35.750917303, t=38.8 degC)  Rank: 00084
   TM values: Max =   161  Min =  -221  Ave =    -4.60  Std =    20.11
   Min TM value:-221 at an offset of 122.863 usec -> 2023-10-20T13:55:43.750.730.587 (XO: 35.750734165)
   Max TM value:161 at an offset of 122.475 usec -> 2023-10-20T13:55:43.750.730.200 (XO: 35.750733777)
   Max dTM/dt [1/us]:1.10e+03 at an offset of 125.450 usec -> 2023-10-20T13:55:43.750.733.175 (XO: 35.750736752)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 110.588 usec -> 2023-10-20T13:55:43.750.718.312 (XO: 35.750721890)
