-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 28 17:47:17 2024 UTC
-- File 20231018_030212_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.812190 to 18.349928
-each separately from 1s to 18s and from 349639us to 812479us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.537160 in 1 packets: 18
MIU packet time from 2023-10-18T03:02:16.812.190.000 to 2023-10-18T03:02:33.349.928.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.537160 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8774089 to 19.8773515
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999426 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231018_030212_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6896000 to 3.6898890
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5371030  -- OK for SL --
   MIU time from 2023-10-18T03:02:16.812.190.000 to 2023-10-18T03:02:16.812.479.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.537160

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2269920 to 20.2272810
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-18T03:02:33.349.639.000 to 2023-10-18T03:02:33.349.928.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.689600000 corrected by 40bit 10MHz counter of 2.877408900 obtained at 1Hz counter of 1. Result: 1.812193861
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-18T03:02:16.812.049.725
   Max 1st sample MIU time among all SL packets: 2023-10-18T03:02:33.349.787.725
   Min 1st sample XO time among all SL packets: 1.812053586
   Max 1st sample XO time among all SL packets: 18.349790414
   TM values from -246 to 180 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-18T03:02:16.812.049.725 (XO: 1.812053586 - 1.812359587, t=38.8 degC)  Rank: 00084
   TM values: Max =   180  Min =  -244  Ave =    -4.50  Std =    18.58
   Min TM value:-244 at an offset of 111.275 usec -> 2023-10-18T03:02:16.812.161.000 (XO: 1.812164861)
   Max TM value:180 at an offset of 110.975 usec -> 2023-10-18T03:02:16.812.160.700 (XO: 1.812164561)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 113.763 usec -> 2023-10-18T03:02:16.812.163.487 (XO: 1.812167349)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 111.113 usec -> 2023-10-18T03:02:16.812.160.837 (XO: 1.812164699)

 - Block number: 2 starting at 2023-10-18T03:02:33.349.498.725 (XO: 18.349501413 - 18.349807414, t=38.8 degC)  Rank: 00082
   TM values: Max =   177  Min =  -246  Ave =    -4.48  Std =    18.87
   Min TM value:-246 at an offset of 99.838 usec -> 2023-10-18T03:02:33.349.598.562 (XO: 18.349601251)
   Max TM value:177 at an offset of 99.500 usec -> 2023-10-18T03:02:33.349.598.225 (XO: 18.349600914)
   Max dTM/dt [1/us]:1.20e+03 at an offset of 102.388 usec -> 2023-10-18T03:02:33.349.601.112 (XO: 18.349603801)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 87.300 usec -> 2023-10-18T03:02:33.349.586.025 (XO: 18.349588714)
