-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 28 17:27:25 2024 UTC
-- File 20231018_023352_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.752990 to 23.339881
-each separately from 6s to 23s and from 339592us to 753279us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.586313 in 1 packets: 18
MIU packet time from 2023-10-18T02:34:01.752.990.000 to 2023-10-18T02:34:18.339.881.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.586313 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8282004 to 24.8281429
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999425 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231018_023352_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5811919 to 8.5814809
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5862550  -- OK for SL --
   MIU time from 2023-10-18T02:34:01.752.990.000 to 2023-10-18T02:34:01.753.279.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.586313

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1677359 to 25.1680249
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-18T02:34:18.339.592.000 to 2023-10-18T02:34:18.339.881.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.581191900 corrected by 40bit 10MHz counter of 7.828200400 obtained at 1Hz counter of 6. Result: 6.752994060
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-18T02:34:01.752.849.725
   Max 1st sample MIU time among all SL packets: 2023-10-18T02:34:18.339.740.725
   Min 1st sample XO time among all SL packets: 6.752853785
   Max 1st sample XO time among all SL packets: 23.339742880
   TM values from -237 to 164 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-18T02:34:01.752.849.725 (XO: 6.752853785 - 6.753159786, t=38.8 degC)  Rank: 00088
   TM values: Max =   164  Min =  -237  Ave =    -4.47  Std =    18.45
   Min TM value:-237 at an offset of 115.013 usec -> 2023-10-18T02:34:01.752.964.737 (XO: 6.752968798)
   Max TM value:164 at an offset of 102.038 usec -> 2023-10-18T02:34:01.752.951.762 (XO: 6.752955823)
   Max dTM/dt [1/us]:1.17e+03 at an offset of 117.588 usec -> 2023-10-18T02:34:01.752.967.312 (XO: 6.752971373)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 102.175 usec -> 2023-10-18T02:34:01.752.951.900 (XO: 6.752955960)

 - Block number: 2 starting at 2023-10-18T02:34:18.339.451.725 (XO: 23.339453879 - 23.339759880, t=38.8 degC)  Rank: 00077
   TM values: Max =   160  Min =  -200  Ave =    -4.47  Std =    19.89
   Min TM value:-200 at an offset of 130.200 usec -> 2023-10-18T02:34:18.339.581.925 (XO: 23.339584080)
   Max TM value:160 at an offset of 129.913 usec -> 2023-10-18T02:34:18.339.581.637 (XO: 23.339583792)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 132.775 usec -> 2023-10-18T02:34:18.339.584.500 (XO: 23.339586655)
   Min dTM/dt [1/us]:-1.40e+03 at an offset of 130.050 usec -> 2023-10-18T02:34:18.339.581.775 (XO: 23.339583930)
