-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 11:56:26 2024 UTC
-- File 20231014_170139_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.841115 to 30.091251
-each separately from 14s to 30s and from 90962us to 971285us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.129592 in 1 packets: 18
      3.119677 in 1 packets: 36
MIU packet time from 2023-10-14T17:01:56.841.115.000 to 2023-10-14T17:02:12.091.251.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.129592 in 1 packets: 18
      3.119677 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8594072 to 31.8593526
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999591 in 1 packets: 18
      3.9999863 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231014_170139_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7005236 to 16.7008126
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1295510  -- OK for SL --
   MIU time from 2023-10-14T17:01:56.841.115.000 to 2023-10-14T17:01:56.841.404.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.129592

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8303636 to 28.8306526
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.1196630  -- OK for SL --
   MIU time from 2023-10-14T17:02:08.970.996.000 to 2023-10-14T17:02:08.971.285.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.119677

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9503156 to 31.9506046
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-14T17:02:12.090.962.000 to 2023-10-14T17:02:12.091.251.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.700523600 corrected by 40bit 10MHz counter of 15.859407200 obtained at 1Hz counter of 14. Result: 14.841119260
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-14T17:01:56.840.974.725
   Max 1st sample MIU time among all SL packets: 2023-10-14T17:02:12.091.110.725
   Min 1st sample XO time among all SL packets: 14.840978984
   Max 1st sample XO time among all SL packets: 30.091112035
   TM values from -275 to 223 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-14T17:01:56.840.974.725 (XO: 14.840978984 - 14.841284985, t=38.8 degC)  Rank: 00075
   TM values: Max =   180  Min =  -219  Ave =    -4.53  Std =    25.07
   Min TM value:-219 at an offset of 118.425 usec -> 2023-10-14T17:01:56.841.093.150 (XO: 14.841097410)
   Max TM value:180 at an offset of 118.725 usec -> 2023-10-14T17:01:56.841.093.450 (XO: 14.841097710)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 118.550 usec -> 2023-10-14T17:01:56.841.093.275 (XO: 14.841097535)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 103.250 usec -> 2023-10-14T17:01:56.841.077.975 (XO: 14.841082235)

 - Block number: 2 starting at 2023-10-14T17:02:08.970.855.725 (XO: 26.970860326 - 26.971166327, t=38.8 degC)  Rank: 00079
   TM values: Max =   202  Min =  -262  Ave =    -4.55  Std =    24.47
   Min TM value:-262 at an offset of 118.263 usec -> 2023-10-14T17:02:08.970.973.987 (XO: 26.970978589)
   Max TM value:202 at an offset of 117.900 usec -> 2023-10-14T17:02:08.970.973.625 (XO: 26.970978226)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 120.763 usec -> 2023-10-14T17:02:08.970.976.487 (XO: 26.970981089)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 118.038 usec -> 2023-10-14T17:02:08.970.973.762 (XO: 26.970978364)

 - Block number: 3 starting at 2023-10-14T17:02:12.090.821.725 (XO: 30.090823034 - 30.091129035, t=38.8 degC)  Rank: 00080
   TM values: Max =   223  Min =  -275  Ave =    -4.50  Std =    26.97
   Min TM value:-275 at an offset of 129.838 usec -> 2023-10-14T17:02:12.090.951.562 (XO: 30.090952872)
   Max TM value:223 at an offset of 129.550 usec -> 2023-10-14T17:02:12.090.951.275 (XO: 30.090952584)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 132.313 usec -> 2023-10-14T17:02:12.090.954.037 (XO: 30.090955347)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 129.700 usec -> 2023-10-14T17:02:12.090.951.425 (XO: 30.090952734)
