-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 08:32:11 2024 UTC
-- File 20231014_084550_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841253us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 17
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 16
      0.833940 in 2 packets: 7, 13
MIU packet time from 2023-10-14T08:45:53.000.000.000 to 2023-10-14T08:46:07.177.027.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 17
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 16
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8264634 to 15.8264155
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999966 in 11 packets: 2, 3, 4, 6, 8, 9, 11, 12, 14, 16, 17
      0.9999965 in 3 packets: 5, 10, 15
   XO PPM: 
      -3.4 in 11 packets: 2, 3, 4, 6, 8, 9, 11, 12, 14, 16, 17
      -3.5 in 3 packets: 5, 10, 15
Temperature FPGA: 
      04 -> 49.8 degC in 7 packets: 1, 5, 11, 12, 13, 14, 18
      03 -> 51.1 degC in 11 packets: 2, 3, 4, 6, 7, 8, 9, 10, 15, 16, 17
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20231014_084550_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8264635 to 16.0034435
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-14T08:45:53.000.000.000 to 2023-10-14T08:46:07.177.027.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 17
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 16
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.826463500 corrected by 40bit 10MHz counter of 1.826463400 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SR packets: 2023-10-14T08:45:52.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-10-14T08:46:07.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028527
   TM values from -52 to 46 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-10-14T08:45:52.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    17  Min =   -28  Ave =    -4.79  Std =     9.18

 - Packet number: 2 starting at 2023-10-14T08:45:53.833.938.925 (XO: 0.833942860 - 0.833959860, t=38.8 degC)
   TM values: Max =    23  Min =   -34  Ave =    -4.56  Std =    11.74

 - Packet number: 3 starting at 2023-10-14T08:45:54.667.881.925 (XO: 1.667885696 - 1.667902696, t=38.8 degC)
   TM values: Max =    34  Min =   -43  Ave =    -4.46  Std =    20.84

 - Packet number: 4 starting at 2023-10-14T08:45:55.501.825.925 (XO: 2.501828531 - 2.501845531, t=38.8 degC)
   TM values: Max =    30  Min =   -42  Ave =    -4.58  Std =    14.08

 - Packet number: 5 starting at 2023-10-14T08:45:56.335.768.925 (XO: 3.335771400 - 3.335788400, t=38.8 degC)
   TM values: Max =    23  Min =   -34  Ave =    -4.56  Std =    11.20

 - Packet number: 6 starting at 2023-10-14T08:45:57.169.712.925 (XO: 4.169714302 - 4.169731302, t=38.8 degC)
   TM values: Max =    20  Min =   -31  Ave =    -4.58  Std =    11.23

 - Packet number: 7 starting at 2023-10-14T08:45:58.003.655.925 (XO: 5.003657137 - 5.003674137, t=38.8 degC)
   TM values: Max =    22  Min =   -31  Ave =    -4.71  Std =    11.35

 - Packet number: 8 starting at 2023-10-14T08:45:58.837.595.925 (XO: 5.837599973 - 5.837616973, t=38.8 degC)
   TM values: Max =    15  Min =   -26  Ave =    -4.67  Std =     7.33

 - Packet number: 9 starting at 2023-10-14T08:45:59.671.539.925 (XO: 6.671542808 - 6.671559808, t=38.8 degC)
   TM values: Max =    29  Min =   -40  Ave =    -4.85  Std =    17.58

 - Packet number: 10 starting at 2023-10-14T08:46:00.505.482.925 (XO: 7.505485694 - 7.505502694, t=38.8 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.76  Std =     7.76

 - Packet number: 11 starting at 2023-10-14T08:46:01.339.425.925 (XO: 8.339428579 - 8.339445579, t=38.8 degC)
   TM values: Max =    19  Min =   -27  Ave =    -4.68  Std =     9.20

 - Packet number: 12 starting at 2023-10-14T08:46:02.173.369.925 (XO: 9.173371414 - 9.173388415, t=38.8 degC)
   TM values: Max =    22  Min =   -37  Ave =    -4.56  Std =    13.97

 - Packet number: 13 starting at 2023-10-14T08:46:03.007.312.925 (XO: 10.007314250 - 10.007331250, t=38.8 degC)
   TM values: Max =    15  Min =   -29  Ave =    -4.78  Std =     7.67

 - Packet number: 14 starting at 2023-10-14T08:46:03.841.252.925 (XO: 10.841257085 - 10.841274085, t=38.8 degC)
   TM values: Max =    14  Min =   -24  Ave =    -4.62  Std =     6.31

 - Packet number: 15 starting at 2023-10-14T08:46:04.675.196.925 (XO: 11.675199988 - 11.675216988, t=38.8 degC)
   TM values: Max =    46  Min =   -52  Ave =    -4.82  Std =    13.12

 - Packet number: 16 starting at 2023-10-14T08:46:05.509.139.925 (XO: 12.509142856 - 12.509159856, t=38.8 degC)
   TM values: Max =    20  Min =   -28  Ave =    -4.76  Std =     9.03

 - Packet number: 17 starting at 2023-10-14T08:46:06.343.083.925 (XO: 13.343085691 - 13.343102692, t=38.8 degC)
   TM values: Max =    19  Min =   -30  Ave =    -4.70  Std =    10.00

 - Packet number: 18 starting at 2023-10-14T08:46:07.177.026.925 (XO: 14.177028527 - 14.177045527, t=38.8 degC)
   TM values: Max =    14  Min =   -20  Ave =    -4.47  Std =     6.16
