-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 21 02:03:27 2024 UTC
-- File 20231002_224953_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.345577 to 31.451715
-each separately from 5s to 31s and from 267725us to 451715us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.921859 in 1 packets: 18
      11.183412 in 1 packets: 36
MIU packet time from 2023-10-02T22:50:01.345.577.000 to 2023-10-02T22:50:27.451.715.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.921859 in 1 packets: 18
      11.183412 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9093574 to 32.9092684
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999487 in 1 packets: 18
      10.9999623 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231002_224953_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2549358 to 7.2552248
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9218070  -- OK for SL --
   MIU time from 2023-10-02T22:50:01.345.577.000 to 2023-10-02T22:50:01.345.866.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.921859

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1770318 to 22.1773208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1833750  -- OK for SL --
   MIU time from 2023-10-02T22:50:16.267.725.000 to 2023-10-02T22:50:16.268.014.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.183412

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.3606958 to 33.3609848
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-02T22:50:27.451.426.000 to 2023-10-02T22:50:27.451.715.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.254935800 corrected by 40bit 10MHz counter of 6.909357400 obtained at 1Hz counter of 5. Result: 5.345579610
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-02T22:50:01.345.436.725
   Max 1st sample MIU time among all SL packets: 2023-10-02T22:50:27.451.574.725
   Min 1st sample XO time among all SL packets: 5.345439334
   Max 1st sample XO time among all SL packets: 31.451577660
   TM values from -273 to 208 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-02T22:50:01.345.436.725 (XO: 5.345439334 - 5.345745335, t=38.8 degC)  Rank: 00084
   TM values: Max =   163  Min =  -221  Ave =    -4.52  Std =    18.24
   Min TM value:-221 at an offset of 107.938 usec -> 2023-10-02T22:50:01.345.544.662 (XO: 5.345547272)
   Max TM value:163 at an offset of 107.563 usec -> 2023-10-02T22:50:01.345.544.287 (XO: 5.345546897)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 110.550 usec -> 2023-10-02T22:50:01.345.547.275 (XO: 5.345549884)
   Min dTM/dt [1/us]:-1.51e+03 at an offset of 107.725 usec -> 2023-10-02T22:50:01.345.544.450 (XO: 5.345547059)

 - Block number: 2 starting at 2023-10-02T22:50:16.267.584.725 (XO: 20.267586335 - 20.267892336, t=38.8 degC)  Rank: 00079
   TM values: Max =   184  Min =  -273  Ave =    -4.55  Std =    21.14
   Min TM value:-273 at an offset of 82.838 usec -> 2023-10-02T22:50:16.267.667.562 (XO: 20.267669173)
   Max TM value:184 at an offset of 69.988 usec -> 2023-10-02T22:50:16.267.654.712 (XO: 20.267656323)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 85.325 usec -> 2023-10-02T22:50:16.267.670.050 (XO: 20.267671660)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 82.663 usec -> 2023-10-02T22:50:16.267.667.387 (XO: 20.267668998)

 - Block number: 3 starting at 2023-10-02T22:50:27.451.285.725 (XO: 31.451288659 - 31.451594660, t=38.8 degC)  Rank: 00076
   TM values: Max =   208  Min =  -217  Ave =    -4.57  Std =    25.27
   Min TM value:-217 at an offset of 124.575 usec -> 2023-10-02T22:50:27.451.410.300 (XO: 31.451413235)
   Max TM value:208 at an offset of 124.313 usec -> 2023-10-02T22:50:27.451.410.037 (XO: 31.451412972)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 122.188 usec -> 2023-10-02T22:50:27.451.407.912 (XO: 31.451410847)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 124.438 usec -> 2023-10-02T22:50:27.451.410.162 (XO: 31.451413097)
