-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 20 19:31:27 2024 UTC
-- File 20230920_061342_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841253us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2023-09-20T06:13:45.000.000.000 to 2023-09-20T06:13:59.177.027.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7044204 to 15.7043728
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999967 in 1 packets: 2
      0.9999965 in 1 packets: 3
      0.9999966 in 12 packets: 4, 5, 6, 8, 9, 10, 11, 12, 14, 15, 16, 17
   XO PPM: 
      -3.3 in 1 packets: 2
      -3.5 in 1 packets: 3
      -3.4 in 12 packets: 4, 5, 6, 8, 9, 10, 11, 12, 14, 15, 16, 17
Temperature FPGA: 
      03 -> 51.1 degC in 4 packets: 1, 3, 4, 17
      04 -> 49.8 degC in 12 packets: 2, 5, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      05 -> 48.5 degC in 2 packets: 6, 18
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      1 in 5 packets: 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 13 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
      1 in 5 packets: 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230920_061342_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7044205 to 15.8814005
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-20T06:13:45.000.000.000 to 2023-09-20T06:13:59.177.027.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 9, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 8, 11, 14, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.704420500 corrected by 40bit 10MHz counter of 1.704420400 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SR packets: 2023-09-20T06:13:44.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-09-20T06:13:59.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028227
   TM values from -19 to 10 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-09-20T06:13:44.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =     7  Min =   -18  Ave =    -4.69  Std =     4.73

 - Packet number: 2 starting at 2023-09-20T06:13:45.833.938.925 (XO: 0.833942777 - 0.833959777, t=38.8 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.31  Std =     4.17

 - Packet number: 3 starting at 2023-09-20T06:13:46.667.881.925 (XO: 1.667885663 - 1.667902663, t=38.8 degC)
   TM values: Max =     7  Min =   -19  Ave =    -4.44  Std =     4.59

 - Packet number: 4 starting at 2023-09-20T06:13:47.501.825.925 (XO: 2.501828531 - 2.501845531, t=38.8 degC)
   TM values: Max =     7  Min =   -14  Ave =    -4.36  Std =     2.71

 - Packet number: 5 starting at 2023-09-20T06:13:48.335.768.925 (XO: 3.335771367 - 3.335788367, t=38.8 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.64  Std =     4.61

 - Packet number: 6 starting at 2023-09-20T06:13:49.169.712.925 (XO: 4.169714202 - 4.169731202, t=38.8 degC)
   TM values: Max =     8  Min =   -14  Ave =    -4.44  Std =     3.26

 - Packet number: 7 starting at 2023-09-20T06:13:50.003.655.925 (XO: 5.003657037 - 5.003674037, t=38.8 degC)
   TM values: Max =     7  Min =   -15  Ave =    -4.68  Std =     4.28

 - Packet number: 8 starting at 2023-09-20T06:13:50.837.595.925 (XO: 5.837599873 - 5.837616873, t=38.8 degC)
   TM values: Max =     5  Min =   -14  Ave =    -4.30  Std =     3.56

 - Packet number: 9 starting at 2023-09-20T06:13:51.671.539.925 (XO: 6.671542708 - 6.671559708, t=38.8 degC)
   TM values: Max =     4  Min =   -14  Ave =    -4.72  Std =     3.36

 - Packet number: 10 starting at 2023-09-20T06:13:52.505.482.925 (XO: 7.505485544 - 7.505502544, t=38.8 degC)
   TM values: Max =     4  Min =   -13  Ave =    -4.54  Std =     2.39

 - Packet number: 11 starting at 2023-09-20T06:13:53.339.425.925 (XO: 8.339428379 - 8.339445379, t=38.8 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.34  Std =     3.75

 - Packet number: 12 starting at 2023-09-20T06:13:54.173.369.925 (XO: 9.173371214 - 9.173388215, t=38.8 degC)
   TM values: Max =     6  Min =   -16  Ave =    -4.52  Std =     4.42

 - Packet number: 13 starting at 2023-09-20T06:13:55.007.312.925 (XO: 10.007314050 - 10.007331050, t=38.8 degC)
   TM values: Max =     3  Min =   -12  Ave =    -4.51  Std =     2.55

 - Packet number: 14 starting at 2023-09-20T06:13:55.841.252.925 (XO: 10.841256885 - 10.841273885, t=38.8 degC)
   TM values: Max =     6  Min =   -13  Ave =    -4.38  Std =     3.19

 - Packet number: 15 starting at 2023-09-20T06:13:56.675.196.925 (XO: 11.675199721 - 11.675216721, t=38.8 degC)
   TM values: Max =    10  Min =   -17  Ave =    -4.44  Std =     5.80

 - Packet number: 16 starting at 2023-09-20T06:13:57.509.139.925 (XO: 12.509142556 - 12.509159556, t=38.8 degC)
   TM values: Max =     6  Min =   -16  Ave =    -4.66  Std =     4.37

 - Packet number: 17 starting at 2023-09-20T06:13:58.343.082.925 (XO: 13.343085391 - 13.343102392, t=38.8 degC)
   TM values: Max =     8  Min =   -14  Ave =    -4.43  Std =     3.39

 - Packet number: 18 starting at 2023-09-20T06:13:59.177.026.925 (XO: 14.177028227 - 14.177045227, t=38.8 degC)
   TM values: Max =    10  Min =   -18  Ave =    -4.68  Std =     5.19
