-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 05:49:46 2024 UTC
-- File 20230725_232456_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.482728 to 28.582313
-each separately from 4s to 28s and from 482728us to 582313us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.099007 in 1 packets: 18
MIU packet time from 2023-07-25T23:25:03.482.728.000 to 2023-07-25T23:25:27.582.313.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.099007 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.3127004 to 30.3126207
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999203 in 1 packets: 18
Temperature FPGA: 
      FE -> 57.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      20 in 1 packets: 1
      21 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      97 in 1 packets: 19
      98 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      28 in 1 packets: 1
      29 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      123 in 1 packets: 19
      124 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230725_232456_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      147 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.7954306 to 6.7957196
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0989270  -- OK for SL --
   MIU time from 2023-07-25T23:25:03.482.728.000 to 2023-07-25T23:25:03.483.017.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.099007

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      144 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.8946466 to 30.8949356
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-25T23:25:27.582.024.000 to 2023-07-25T23:25:27.582.313.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.795430600 corrected by 40bit 10MHz counter of 6.312700400 obtained at 1Hz counter of 4. Result: 4.482731793
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-07-25T23:25:03.482.587.725
   Max 1st sample MIU time among all SL packets: 2023-07-25T23:25:27.582.172.725
   Min 1st sample XO time among all SL packets: 4.482591518
   Max 1st sample XO time among all SL packets: 28.582176604
   TM values from -791 to 1204 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-25T23:25:03.482.587.725 (XO: 4.482591518 - 4.482897519, t=45.2 degC)  Rank: 00147
   TM values: Max =  1204  Min =  -786  Ave =    -4.36  Std =    37.78
   Min TM value:-786 at an offset of 120.688 usec -> 2023-07-25T23:25:03.482.708.412 (XO: 4.482712205)
   Max TM value:1204 at an offset of 120.663 usec -> 2023-07-25T23:25:03.482.708.387 (XO: 4.482712180)
   Max dTM/dt [1/us]:1.53e+04 at an offset of 120.800 usec -> 2023-07-25T23:25:03.482.708.525 (XO: 4.482712318)
   Min dTM/dt [1/us]:-1.39e+04 at an offset of 120.700 usec -> 2023-07-25T23:25:03.482.708.425 (XO: 4.482712218)

 - Block number: 2 starting at 2023-07-25T23:25:27.581.883.725 (XO: 28.581887603 - 28.582193604, t=45.2 degC)  Rank: 00144
   TM values: Max =  1142  Min =  -791  Ave =    -4.36  Std =    38.37
   Min TM value:-791 at an offset of 122.288 usec -> 2023-07-25T23:25:27.582.006.012 (XO: 28.582009891)
   Max TM value:1142 at an offset of 122.175 usec -> 2023-07-25T23:25:27.582.005.900 (XO: 28.582009779)
   Max dTM/dt [1/us]:1.48e+04 at an offset of 122.325 usec -> 2023-07-25T23:25:27.582.006.050 (XO: 28.582009929)
   Min dTM/dt [1/us]:-1.29e+04 at an offset of 122.225 usec -> 2023-07-25T23:25:27.582.005.950 (XO: 28.582009829)
