-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 05:47:19 2024 UTC
-- File 20230725_214134_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177025
-each separately from 0s to 14s and from 0us to 841251us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU packet time from 2023-07-25T21:41:37.000.000.000 to 2023-07-25T21:41:51.177.025.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.3540849 to 15.3540386
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999968 in 2 packets: 2, 12
      0.9999966 in 3 packets: 3, 11, 14
      0.9999967 in 9 packets: 4, 5, 6, 8, 9, 10, 15, 16, 17
   XO PPM: 
      -3.2 in 2 packets: 2, 12
      -3.4 in 3 packets: 3, 11, 14
      -3.3 in 9 packets: 4, 5, 6, 8, 9, 10, 15, 16, 17
Temperature FPGA: 
      FE -> 57.6 degC in 11 packets: 1, 4, 5, 6, 7, 8, 9, 10, 15, 16, 18
      FF -> 56.3 degC in 4 packets: 2, 3, 11, 12
      FD -> 58.9 degC in 3 packets: 13, 14, 17
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      5 in 1 packets: 2
      12 in 1 packets: 3
      14 in 1 packets: 4
      17 in 1 packets: 5
      24 in 1 packets: 6
      29 in 1 packets: 7
      33 in 1 packets: 8
      38 in 1 packets: 9
      40 in 1 packets: 10
      45 in 1 packets: 11
      50 in 1 packets: 12
      55 in 1 packets: 13
      56 in 1 packets: 14
      59 in 1 packets: 15
      65 in 1 packets: 16
      66 in 1 packets: 17
      69 in 1 packets: 18
Number of generated alerts: 
      0 in 1 packets: 1
      8 in 1 packets: 2
      16 in 1 packets: 3
      18 in 1 packets: 4
      23 in 1 packets: 5
      31 in 1 packets: 6
      38 in 1 packets: 7
      43 in 1 packets: 8
      49 in 1 packets: 9
      52 in 1 packets: 10
      57 in 1 packets: 11
      64 in 1 packets: 12
      71 in 1 packets: 13
      72 in 1 packets: 14
      77 in 1 packets: 15
      86 in 1 packets: 16
      88 in 1 packets: 17
      91 in 1 packets: 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230725_214134_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.3540850 to 15.5310650
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-25T21:41:37.000.000.000 to 2023-07-25T21:41:51.177.025.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.354085000 corrected by 40bit 10MHz counter of 1.354084900 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SR packets: 2023-07-25T21:41:36.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-25T21:41:51.177.024.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177026909
   TM values from -266 to 247 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-25T21:41:36.999.999.925 (XO: 0.000000025 - 0.000017025, t=45.2 degC)
   TM values: Max =    24  Min =   -33  Ave =    -4.34  Std =     9.27

 - Packet number: 2 starting at 2023-07-25T21:41:37.833.937.925 (XO: 0.833942694 - 0.833959694, t=45.2 degC)
   TM values: Max =    24  Min =   -32  Ave =    -4.44  Std =     9.57

 - Packet number: 3 starting at 2023-07-25T21:41:38.667.880.925 (XO: 1.667885496 - 1.667902496, t=45.2 degC)
   TM values: Max =    24  Min =   -34  Ave =    -4.09  Std =    12.00

 - Packet number: 4 starting at 2023-07-25T21:41:39.501.824.925 (XO: 2.501828281 - 2.501845281, t=45.2 degC)
   TM values: Max =    36  Min =   -42  Ave =    -4.57  Std =    12.71

 - Packet number: 5 starting at 2023-07-25T21:41:40.335.767.925 (XO: 3.335771033 - 3.335788033, t=45.2 degC)
   TM values: Max =   247  Min =  -266  Ave =    -5.76  Std =    67.61

 - Packet number: 6 starting at 2023-07-25T21:41:41.169.711.925 (XO: 4.169713785 - 4.169730785, t=45.2 degC)
   TM values: Max =    22  Min =   -33  Ave =    -4.51  Std =    10.11

 - Packet number: 7 starting at 2023-07-25T21:41:42.003.654.925 (XO: 5.003656537 - 5.003673537, t=45.2 degC)
   TM values: Max =    39  Min =   -72  Ave =    -4.11  Std =    12.13

 - Packet number: 8 starting at 2023-07-25T21:41:42.837.593.925 (XO: 5.837599289 - 5.837616289, t=45.2 degC)
   TM values: Max =    29  Min =   -42  Ave =    -4.54  Std =     9.93

 - Packet number: 9 starting at 2023-07-25T21:41:43.671.537.925 (XO: 6.671542041 - 6.671559041, t=45.2 degC)
   TM values: Max =    30  Min =   -35  Ave =    -4.21  Std =     8.67

 - Packet number: 10 starting at 2023-07-25T21:41:44.505.480.925 (XO: 7.505484793 - 7.505501793, t=45.2 degC)
   TM values: Max =    29  Min =   -39  Ave =    -4.64  Std =    10.65

 - Packet number: 11 starting at 2023-07-25T21:41:45.339.424.925 (XO: 8.339427579 - 8.339444579, t=45.2 degC)
   TM values: Max =    33  Min =   -39  Ave =    -4.53  Std =    11.19

 - Packet number: 12 starting at 2023-07-25T21:41:46.173.367.925 (XO: 9.173370380 - 9.173387380, t=45.2 degC)
   TM values: Max =    28  Min =   -36  Ave =    -4.44  Std =    11.31

 - Packet number: 13 starting at 2023-07-25T21:41:47.007.311.925 (XO: 10.007313050 - 10.007330050, t=45.2 degC)
   TM values: Max =    27  Min =   -36  Ave =    -4.42  Std =     9.97

 - Packet number: 14 starting at 2023-07-25T21:41:47.841.250.925 (XO: 10.841255885 - 10.841272885, t=45.2 degC)
   TM values: Max =    27  Min =   -38  Ave =    -4.10  Std =    11.78

 - Packet number: 15 starting at 2023-07-25T21:41:48.675.193.925 (XO: 11.675198653 - 11.675215653, t=45.2 degC)
   TM values: Max =    40  Min =   -42  Ave =    -4.46  Std =    13.53

 - Packet number: 16 starting at 2023-07-25T21:41:49.509.137.925 (XO: 12.509141405 - 12.509158405, t=45.2 degC)
   TM values: Max =    41  Min =   -40  Ave =    -4.26  Std =     9.88

 - Packet number: 17 starting at 2023-07-25T21:41:50.343.080.925 (XO: 13.343084157 - 13.343101157, t=45.2 degC)
   TM values: Max =    29  Min =   -38  Ave =    -4.32  Std =     9.89

 - Packet number: 18 starting at 2023-07-25T21:41:51.177.024.925 (XO: 14.177026909 - 14.177043909, t=45.2 degC)
   TM values: Max =    36  Min =   -35  Ave =    -4.33  Std =    11.56
