-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 03:51:16 2024 UTC
-- File 20230725_082841_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.551527 to 46.979420
-each separately from 10s to 46s and from 209736us to 979420us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.058982 in 1 packets: 18
      12.598649 in 1 packets: 36
      11.769106 in 1 packets: 54
MIU packet time from 2023-07-25T08:28:54.551.527.000 to 2023-07-25T08:29:30.979.420.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.058982 in 1 packets: 18
      12.598649 in 1 packets: 36
      11.769106 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.5707552 to 47.5706388
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999613 in 1 packets: 18
      12.9999579 in 1 packets: 36
      10.9999644 in 1 packets: 54
Temperature FPGA: 
      FD -> 58.9 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C6 -> 45.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230725_082841_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1222842 to 12.1225732
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0589430  -- OK for SL --
   MIU time from 2023-07-25T08:28:54.551.527.000 to 2023-07-25T08:28:54.551.816.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.058982

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.1815162 to 24.1818052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5986070  -- OK for SL --
   MIU time from 2023-07-25T08:29:06.610.798.000 to 2023-07-25T08:29:06.611.087.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.598649

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.7804122 to 36.7807012
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7690710  -- OK for SL --
   MIU time from 2023-07-25T08:29:19.209.736.000 to 2023-07-25T08:29:19.210.025.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.769106

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.5497722 to 48.5500612
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-25T08:29:30.979.131.000 to 2023-07-25T08:29:30.979.420.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.122284200 corrected by 40bit 10MHz counter of 11.570755200 obtained at 1Hz counter of 10. Result: 10.551530820
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2023-07-25T08:28:54.551.386.725
   Max 1st sample MIU time among all SL packets: 2023-07-25T08:29:30.979.279.725
   Min 1st sample XO time among all SL packets: 10.551390545
   Max 1st sample XO time among all SL packets: 46.979285259
   TM values from -236 to 180 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-25T08:28:54.551.386.725 (XO: 10.551390545 - 10.551696546, t=45.2 degC)  Rank: 00081
   TM values: Max =   177  Min =  -236  Ave =    -4.39  Std =    18.46
   Min TM value:-236 at an offset of 129.225 usec -> 2023-07-25T08:28:54.551.515.950 (XO: 10.551519770)
   Max TM value:177 at an offset of 131.575 usec -> 2023-07-25T08:28:54.551.518.300 (XO: 10.551522120)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 131.425 usec -> 2023-07-25T08:28:54.551.518.150 (XO: 10.551521970)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 129.075 usec -> 2023-07-25T08:28:54.551.515.800 (XO: 10.551519620)

 - Block number: 2 starting at 2023-07-25T08:29:06.610.657.725 (XO: 22.610661379 - 22.610967380, t=45.2 degC)  Rank: 00078
   TM values: Max =   146  Min =  -177  Ave =    -4.31  Std =    17.11
   Min TM value:-177 at an offset of 126.350 usec -> 2023-07-25T08:29:06.610.784.075 (XO: 22.610787730)
   Max TM value:146 at an offset of 131.200 usec -> 2023-07-25T08:29:06.610.788.925 (XO: 22.610792580)
   Max dTM/dt [1/us]:1.15e+03 at an offset of 116.013 usec -> 2023-07-25T08:29:06.610.773.737 (XO: 22.610777392)
   Min dTM/dt [1/us]:-1.17e+03 at an offset of 126.138 usec -> 2023-07-25T08:29:06.610.783.862 (XO: 22.610787517)

 - Block number: 3 starting at 2023-07-25T08:29:19.209.595.725 (XO: 35.209598217 - 35.209904218, t=45.2 degC)  Rank: 00076
   TM values: Max =   173  Min =  -206  Ave =    -4.51  Std =    28.20
   Min TM value:-206 at an offset of 126.475 usec -> 2023-07-25T08:29:19.209.722.200 (XO: 35.209724692)
   Max TM value:173 at an offset of 186.713 usec -> 2023-07-25T08:29:19.209.782.438 (XO: 35.209784930)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 189.226 usec -> 2023-07-25T08:29:19.209.784.950 (XO: 35.209787442)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 186.863 usec -> 2023-07-25T08:29:19.209.782.588 (XO: 35.209785080)

 - Block number: 4 starting at 2023-07-25T08:29:30.978.990.725 (XO: 46.978996258 - 46.979302259, t=45.2 degC)  Rank: 00084
   TM values: Max =   180  Min =  -209  Ave =    -4.38  Std =    15.66
   Min TM value:-209 at an offset of 129.013 usec -> 2023-07-25T08:29:30.979.119.737 (XO: 46.979125271)
   Max TM value:180 at an offset of 131.375 usec -> 2023-07-25T08:29:30.979.122.100 (XO: 46.979127633)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 131.263 usec -> 2023-07-25T08:29:30.979.121.987 (XO: 46.979127521)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 128.850 usec -> 2023-07-25T08:29:30.979.119.575 (XO: 46.979125108)
