-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 03:37:28 2024 UTC
-- File 20230725_053936_1229.dat, 1092  packets found.
APID:
      1229 (EW) in 1092 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 1092 packets (all packets)
Total number of packets: 
      22440 in 1092 packets (all packets)
Packet No from 1 to 1092
!!!ERROR: Maximum packet number must be = 22440
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 22440
Hz and MHz counters from 10.551949 to 10.551949
-each separately from 10s to 10s and from 551949us to 551949us.
   Differences in neigboring packets: 
      0.000000 in 1091 packets (all packets)
MIU packet time from 2023-07-25T05:39:49.551.949.000 to 2023-07-25T05:39:49.551.949.000
   Differences in neigboring packets: 
      0.000000 in 1091 packets (all packets)
MIU orbit number: 
      60000 in 1092 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 1092 packets (all packets)
MIU orbit information / number of events: 
      0 in 1092 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 1092 packets (all packets)
Instrument anomalies: 
      0 (OK) in 1092 packets (all packets)
FPGA configuration version: 
      211 in 1092 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7553590 to 11.7553590
   Differences in neigboring packets: 
      0.0000000 in 1091 packets (all packets)
Temperature FPGA: 
      FD -> 58.9 degC in 1092 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 1092 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 1092 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 1092 packets (all packets)
Number of external events: 
      1 in 1092 packets (all packets)
Number of generated alerts: 
      2 in 1092 packets (all packets)
Survey memory error: 
      0 in 1092 packets (all packets)
Event memory error: 
      0 in 1092 packets (all packets)
Configuration error: 
      0 in 1092 packets (all packets)
SDRAM CRC error: 
      0 in 1092 packets (all packets)
Radio gain: 
      0 in 1092 packets (all packets)
Radio bandwidth: 
      0 in 1092 packets (all packets)
Digital gain: 
      0 in 1092 packets (all packets)
Analog gain: 
      0 in 1092 packets (all packets)

EVENT DATA READER
-- File 20230725_053936_1229.dat, 5 EW pages found. Extra packets: 72.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 1
      1 in 1020 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 12.2518224 to 12.2691454
      Difference between the neighboring packets: 
      0.0000170 in 1019 packets (all packets)
   Memory page number from 118 to 122
      Difference between the neighboring packets: 
      0 in 1015 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 4 packets: 204, 408, 612, 816
   Event page number from 1 to 5
      Difference between the neighboring packets: 
      0 in 1015 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 4 packets: 204, 408, 612, 816
   MIU event flag 
      1 in 1020 packets (all packets)
   Time triggered event flag 
      0 in 1020 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-25T05:39:49.496.460.825
   Min EW header abs. time found among all events            : 2023-07-25T05:39:49.551.949.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-25T05:39:49.496.460.825
   Max EW header abs. time found among all events            : 2023-07-25T05:39:49.551.949.000
   TM values from -170 to 143 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 1020
    Page no. 1:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 5  ordered packets: 1020
   TM values from -170 to 143 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=10 C1MHz=0.551949
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1019 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 12.2518224
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-25T05:39:49.551.949.000
   Event 1st sample abs. time : 2023-07-25T05:39:49.496.460.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 from 2023-07-25T05:39:49.496.460.825 to 2023-07-25T05:39:49.513.800.825 (XO: -333.000000175 - -332.982660175)
   Min TM value:-170 at an offset of 14161.513 usec -> 2023-07-25T05:39:49.510.622.337 (XO: -332.985838663)
   Max TM value:143 at an offset of 14295.888 usec -> 2023-07-25T05:39:49.510.756.712 (XO: -332.985704288)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 14298.487 usec -> 2023-07-25T05:39:49.510.759.312 (XO: -332.985701688)
   Min dTM/dt [1/us]:-1.34e+03 at an offset of 14296.000 usec -> 2023-07-25T05:39:49.510.756.825 (XO: -332.985704175)
   Med: -4.00000   Ave: -4.4557353   Std: 12.552875
