-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 03:11:43 2024 UTC
-- File 20230725_012254_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.866818 to 46.592347
-each separately from 4s to 46s and from 410589us to 867107us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.681226 in 1 packets: 18
      9.861967 in 1 packets: 36
      9.181180 in 1 packets: 54
MIU packet time from 2023-07-25T01:23:01.866.818.000 to 2023-07-25T01:23:43.592.347.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.681226 in 1 packets: 18
      9.861967 in 1 packets: 36
      9.181180 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.5080301 to 47.5078950
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999259 in 1 packets: 18
      9.9999679 in 1 packets: 36
      8.9999711 in 1 packets: 54
Temperature FPGA: 
      FD -> 58.9 degC in 72 packets (all packets)
Temperature XO: 
      C7 -> 46.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C6 -> 45.2 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230725_012254_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.3748511 to 6.3751401
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6811510  -- OK for SL --
   MIU time from 2023-07-25T01:23:01.866.818.000 to 2023-07-25T01:23:01.867.107.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.681226

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0562911 to 29.0565801
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.8619350  -- OK for SL --
   MIU time from 2023-07-25T01:23:24.548.333.000 to 2023-07-25T01:23:24.548.622.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.861967

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.9185151 to 38.9188041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1811510  -- OK for SL --
   MIU time from 2023-07-25T01:23:34.410.589.000 to 2023-07-25T01:23:34.410.878.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.181180

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.0999551 to 48.1002441
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-25T01:23:43.592.058.000 to 2023-07-25T01:23:43.592.347.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.374851100 corrected by 40bit 10MHz counter of 5.508030100 obtained at 1Hz counter of 4. Result: 4.866823861
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2023-07-25T01:23:01.866.677.725
   Max 1st sample MIU time among all SL packets: 2023-07-25T01:23:43.592.206.725
   Min 1st sample XO time among all SL packets: 4.866683585
   Max 1st sample XO time among all SL packets: 46.592210720
   TM values from -264 to 199 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-25T01:23:01.866.677.725 (XO: 4.866683585 - 4.866989586, t=46.8 degC)  Rank: 00076
   TM values: Max =   199  Min =  -248  Ave =    -4.37  Std =    26.53
   Min TM value:-248 at an offset of 115.013 usec -> 2023-07-25T01:23:01.866.792.737 (XO: 4.866798598)
   Max TM value:199 at an offset of 157.401 usec -> 2023-07-25T01:23:01.866.835.125 (XO: 4.866840986)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 117.288 usec -> 2023-07-25T01:23:01.866.795.012 (XO: 4.866800873)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 157.538 usec -> 2023-07-25T01:23:01.866.835.263 (XO: 4.866841123)

 - Block number: 2 starting at 2023-07-25T01:23:24.548.192.725 (XO: 27.548196579 - 27.548502580, t=45.2 degC)  Rank: 00083
   TM values: Max =   197  Min =  -264  Ave =    -4.50  Std =    28.96
   Min TM value:-264 at an offset of 123.725 usec -> 2023-07-25T01:23:24.548.316.450 (XO: 27.548320305)
   Max TM value:197 at an offset of 128.400 usec -> 2023-07-25T01:23:24.548.321.125 (XO: 27.548324980)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 155.088 usec -> 2023-07-25T01:23:24.548.347.812 (XO: 27.548351667)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 140.063 usec -> 2023-07-25T01:23:24.548.332.787 (XO: 27.548336642)

 - Block number: 3 starting at 2023-07-25T01:23:34.410.448.725 (XO: 37.410452279 - 37.410758281, t=45.2 degC)  Rank: 00090
   TM values: Max =   198  Min =  -246  Ave =    -4.47  Std =    26.60
   Min TM value:-246 at an offset of 124.763 usec -> 2023-07-25T01:23:34.410.573.487 (XO: 37.410577042)
   Max TM value:198 at an offset of 129.500 usec -> 2023-07-25T01:23:34.410.578.225 (XO: 37.410581780)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 169.013 usec -> 2023-07-25T01:23:34.410.617.738 (XO: 37.410621293)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 166.526 usec -> 2023-07-25T01:23:34.410.615.250 (XO: 37.410618805)

 - Block number: 4 starting at 2023-07-25T01:23:43.591.917.725 (XO: 46.591921719 - 46.592227720, t=46.8 degC)  Rank: 00079
   TM values: Max =   191  Min =  -197  Ave =    -4.51  Std =    23.26
   Min TM value:-197 at an offset of 113.550 usec -> 2023-07-25T01:23:43.592.031.275 (XO: 46.592035270)
   Max TM value:191 at an offset of 157.126 usec -> 2023-07-25T01:23:43.592.074.850 (XO: 46.592078845)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 156.951 usec -> 2023-07-25T01:23:43.592.074.675 (XO: 46.592078670)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 129.738 usec -> 2023-07-25T01:23:43.592.047.462 (XO: 46.592051457)
