-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 02:37:22 2024 UTC
-- File 20230724_222710_1229.dat, 5998  packets found.
APID:
      1229 (EW) in 5998 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 5998 packets (all packets)
Total number of packets: 
      36720 in 5998 packets (all packets)
Packet No from 1 to 5998
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 5.187961 to 34.842921
-each separately from 5s to 34s and from 187961us to 842921us.
   Differences in neigboring packets: 
      0.000000 in 5995 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      28.475837 in 1 packets: 3876
      -29.654960 in 1 packets: 5304
MIU packet time from 2023-07-24T22:27:18.187.961.000 to 2023-07-24T22:27:47.842.921.000
   Differences in neigboring packets: 
      0.000000 in 5995 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      28.475837 in 1 packets: 3876
      -29.654960 in 1 packets: 5304
MIU orbit number: 
      60000 in 5998 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 5998 packets (all packets)
MIU orbit information / number of events: 
      0 in 5998 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 5998 packets (all packets)
Instrument anomalies: 
      0 (OK) in 5998 packets (all packets)
FPGA configuration version: 
      211 in 5998 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8750794 to 35.8749868
   Differences in neigboring packets: 
      0.0000000 in 5995 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      27.9999105 in 1 packets: 3876
      -28.9999074 in 1 packets: 5304
Temperature FPGA: 
      FD -> 58.9 degC in 5304 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 694 packets: 5305, 5306, 5307, 5308, 5309, 5310, 5311, 5312, 5313, 5314, 5315, 5316, 5317, 5318, 5319, 5320, 5321, 5322, ...
Temperature XO: 
      C7 -> 46.8 degC in 5998 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 5998 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 5998 packets (all packets)
Number of external events: 
      2 in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 1428 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
      1 in 694 packets: 5305, 5306, 5307, 5308, 5309, 5310, 5311, 5312, 5313, 5314, 5315, 5316, 5317, 5318, 5319, 5320, 5321, 5322, ...
Number of generated alerts: 
      2 in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 1428 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
      1 in 694 packets: 5305, 5306, 5307, 5308, 5309, 5310, 5311, 5312, 5313, 5314, 5315, 5316, 5317, 5318, 5319, 5320, 5321, 5322, ...
Survey memory error: 
      0 in 5998 packets (all packets)
Event memory error: 
      0 in 5998 packets (all packets)
Configuration error: 
      0 in 5998 packets (all packets)
SDRAM CRC error: 
      0 in 5998 packets (all packets)
Radio gain: 
      0 in 5998 packets (all packets)
Radio bandwidth: 
      0 in 5998 packets (all packets)
Digital gain: 
      0 in 5998 packets (all packets)
Analog gain: 
      0 in 5998 packets (all packets)

EVENT DATA READER
-- File 20230724_222710_1229.dat, 29 EW pages found. Extra packets: 82.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      2 in 3876 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 1428 packets: 3877, 3878, 3879, 3880, 3881, 3882, 3883, 3884, 3885, 3886, 3887, 3888, 3889, 3890, 3891, 3892, 3893, 3894, ...
      1 in 612 packets: 5305, 5306, 5307, 5308, 5309, 5310, 5311, 5312, 5313, 5314, 5315, 5316, 5317, 5318, 5319, 5320, 5321, 5322, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 7.0075542 to 36.6866812
      Difference between the neighboring packets: 
      0.0000170 in 5913 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      28.2676850 in 1 packets: 3876
      -29.6791270 in 1 packets: 5304
   Memory page number from 0 to 37
      Difference between the neighboring packets: 
      0 in 5887 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 27 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      10 in 1 packets: 3876
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 5887 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 26 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -59 in 1 packets: 3876
      -6 in 1 packets: 5304
   MIU event flag 
      1 in 5916 packets (all packets)
   Time triggered event flag 
      0 in 5916 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-24T22:27:18.132.472.653
   Min EW header abs. time found among all events            : 2023-07-24T22:27:18.187.961.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-24T22:27:47.787.432.648
   Max EW header abs. time found among all events            : 2023-07-24T22:27:47.842.921.000
   TM values from -251 to 224 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 612
    Page no. 1:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 3  ordered packets: 612
   TM values from -235 to 198 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=5 C1MHz=0.187961
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 611 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 7.0075542
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T22:27:18.187.961.000
   Event 1st sample abs. time : 2023-07-24T22:27:18.132.472.653)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.007554200 corrected by 40bit 10MHz counter of 6.875079400 obtained at 1Hz counter of 5. Result: 5.132475211
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):               5.132475036
   Event data descriptor XO time (reconstructed seconds from XP start)    :               5.132475211
   Sampling frequency correction factor for this event:     0.999996900

 Event no. 2 total number of packets: 3876
    Page no. 42:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 19  ordered packets: 3876
   TM values from -251 to 215 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=6 C1MHz=0.367084
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 3875 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 8.3288622
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T22:27:19.367.084.000
   Event 1st sample abs. time : 2023-07-24T22:27:19.311.595.642)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.328862200 corrected by 40bit 10MHz counter of 7.875076300 obtained at 1Hz counter of 6. Result: 6.453787397
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):               6.453787222
   Event data descriptor XO time (reconstructed seconds from XP start)    :               6.453787397
   Sampling frequency correction factor for this event:     0.999996700

 Event no. 3 total number of packets: 1428
    Page no. 1:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 7  ordered packets: 1428
   TM values from -236 to 224 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=34 C1MHz=0.842921
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1427 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 36.6624222
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T22:27:47.842.921.000
   Event 1st sample abs. time : 2023-07-24T22:27:47.787.432.648)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 34, 32bit 10MHz counter = 36.662422200 corrected by 40bit 10MHz counter of 29.875005900 obtained at 1Hz counter of 28. Result: 34.787438020
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              34.787437845
   Event data descriptor XO time (reconstructed seconds from XP start)    :              34.787438020
   Sampling frequency correction factor for this event:     0.999996800
--
Extremes in separate events:

 - Event number: 1 from 2023-07-24T22:27:18.132.472.653 to 2023-07-24T22:27:18.142.876.685 (XO: 5.132475036 - 5.142879068)
   Min TM value:-235 at an offset of 2033.456 usec -> 2023-07-24T22:27:18.134.506.109 (XO: 5.134508492)
   Max TM value:198 at an offset of 1912.293 usec -> 2023-07-24T22:27:18.134.384.946 (XO: 5.134387329)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 2033.581 usec -> 2023-07-24T22:27:18.134.506.234 (XO: 5.134508617)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 1991.456 usec -> 2023-07-24T22:27:18.134.464.109 (XO: 5.134466492)
   Med: -5.00000   Ave: -4.5769115   Std: 9.3303648

 - Event number: 2 from 2023-07-24T22:27:19.311.595.642 to 2023-07-24T22:27:19.377.487.859 (XO: 6.453787222 - 6.519679440)
   Min TM value:-251 at an offset of 50566.792 usec -> 2023-07-24T22:27:19.362.162.433 (XO: 6.504354014)
   Max TM value:215 at an offset of 30591.076 usec -> 2023-07-24T22:27:19.342.186.717 (XO: 6.484378298)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 30590.926 usec -> 2023-07-24T22:27:19.342.186.567 (XO: 6.484378148)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 50566.629 usec -> 2023-07-24T22:27:19.362.162.271 (XO: 6.504353852)
   Med: -5.00000   Ave: -4.5360178   Std: 10.133100

 - Event number: 3 from 2023-07-24T22:27:47.787.432.648 to 2023-07-24T22:27:47.811.708.725 (XO: 34.787437845 - 34.811713922)
   Min TM value:-236 at an offset of 3643.762 usec -> 2023-07-24T22:27:47.791.076.409 (XO: 34.791081606)
   Max TM value:224 at an offset of 13622.719 usec -> 2023-07-24T22:27:47.801.055.366 (XO: 34.801060563)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 3643.862 usec -> 2023-07-24T22:27:47.791.076.509 (XO: 34.791081706)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 13622.794 usec -> 2023-07-24T22:27:47.801.055.441 (XO: 34.801060638)
   Med: -4.00000   Ave: -4.4303299   Std: 10.921287
