-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 02:21:58 2024 UTC
-- File 20230724_201414_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.212556 to 24.682581
-each separately from 6s to 24s and from 212556us to 682581us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.469447 in 1 packets: 18
MIU packet time from 2023-07-24T20:14:23.212.556.000 to 2023-07-24T20:14:41.682.581.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.469447 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9861329 to 25.9860753
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999424 in 1 packets: 18
Temperature FPGA: 
      FD -> 58.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FE -> 57.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C7 -> 46.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230724_201414_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.1986900 to 8.1989790
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4693910  -- OK for SL --
   MIU time from 2023-07-24T20:14:23.212.556.000 to 2023-07-24T20:14:23.212.845.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.469447

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      143 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6683700 to 26.6686590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-24T20:14:41.682.292.000 to 2023-07-24T20:14:41.682.581.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.198690000 corrected by 40bit 10MHz counter of 7.986132900 obtained at 1Hz counter of 6. Result: 6.212557780
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996800 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2023-07-24T20:14:23.212.415.725
   Max 1st sample MIU time among all SL packets: 2023-07-24T20:14:41.682.440.725
   Min 1st sample XO time among all SL packets: 6.212417505
   Max 1st sample XO time among all SL packets: 24.682445609
   TM values from -1109 to 1233 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-24T20:14:23.212.415.725 (XO: 6.212417505 - 6.212723506, t=46.8 degC)  Rank: 00080
   TM values: Max =   292  Min =  -326  Ave =    -4.60  Std =    38.58
   Min TM value:-326 at an offset of 111.150 usec -> 2023-07-24T20:14:23.212.526.875 (XO: 6.212528655)
   Max TM value:292 at an offset of 85.338 usec -> 2023-07-24T20:14:23.212.501.062 (XO: 6.212502843)
   Max dTM/dt [1/us]:2.65e+03 at an offset of 82.938 usec -> 2023-07-24T20:14:23.212.498.662 (XO: 6.212500443)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 108.863 usec -> 2023-07-24T20:14:23.212.524.587 (XO: 6.212526368)

 - Block number: 2 starting at 2023-07-24T20:14:41.682.151.725 (XO: 24.682156608 - 24.682462609, t=46.8 degC)  Rank: 00143
   TM values: Max =  1233  Min = -1109  Ave =    -4.56  Std =    51.64
   Min TM value:-1109 at an offset of 104.850 usec -> 2023-07-24T20:14:41.682.256.575 (XO: 24.682261458)
   Max TM value:1233 at an offset of 104.700 usec -> 2023-07-24T20:14:41.682.256.425 (XO: 24.682261308)
   Max dTM/dt [1/us]:1.38e+04 at an offset of 104.638 usec -> 2023-07-24T20:14:41.682.256.362 (XO: 24.682261246)
   Min dTM/dt [1/us]:-1.53e+04 at an offset of 104.738 usec -> 2023-07-24T20:14:41.682.256.462 (XO: 24.682261346)
