-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 02:16:23 2024 UTC
-- File 20230724_193650_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177023
-each separately from 0s to 14s and from 0us to 841250us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 3, 5, 8, 14, 15, 17
      0.833944 in 6 packets: 4, 6, 9, 10, 12, 16
      0.833939 in 2 packets: 7, 13
      0.833942 in 1 packets: 11
MIU packet time from 2023-07-24T19:36:53.000.000.000 to 2023-07-24T19:37:07.177.023.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 3, 5, 8, 14, 15, 17
      0.833944 in 6 packets: 4, 6, 9, 10, 12, 16
      0.833939 in 2 packets: 7, 13
      0.833942 in 1 packets: 11
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.3674632 to 15.3674185
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999968 in 7 packets: 2, 3, 4, 5, 9, 12, 16
      0.9999969 in 4 packets: 6, 10, 14, 17
      0.9999967 in 3 packets: 8, 11, 15
   XO PPM: 
      -3.2 in 7 packets: 2, 3, 4, 5, 9, 12, 16
      -3.1 in 4 packets: 6, 10, 14, 17
      -3.3 in 3 packets: 8, 11, 15
Temperature FPGA: 
      FE -> 57.6 degC in 6 packets: 1, 3, 4, 5, 12, 17
      FD -> 58.9 degC in 11 packets: 2, 6, 7, 8, 9, 10, 11, 13, 14, 16, 18
      FC -> 60.2 degC in 1 packets: 15
Temperature XO: 
      C7 -> 46.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 5 packets: 1, 2, 3, 4, 5
      1 in 4 packets: 6, 7, 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 5 packets: 1, 2, 3, 4, 5
      1 in 4 packets: 6, 7, 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230724_193650_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.3674633 to 15.5444433
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-24T19:36:53.000.000.000 to 2023-07-24T19:37:07.177.023.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 3, 5, 8, 14, 15, 17
      0.833944 in 6 packets: 4, 6, 9, 10, 12, 16
      0.833939 in 2 packets: 7, 13
      0.833942 in 1 packets: 11
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.367463300 corrected by 40bit 10MHz counter of 1.367463200 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996700 - 0.999996900
   Min 1st sample MIU time among all SR packets: 2023-07-24T19:36:52.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-24T19:37:07.177.022.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177025274
   TM values from -151 to 152 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-24T19:36:52.999.999.925 (XO: 0.000000025 - 0.000017025, t=46.8 degC)
   TM values: Max =    57  Min =   -60  Ave =    -4.42  Std =    20.77

 - Packet number: 2 starting at 2023-07-24T19:36:53.833.937.925 (XO: 0.833942694 - 0.833959694, t=46.8 degC)
   TM values: Max =    48  Min =   -61  Ave =    -4.70  Std =    18.77

 - Packet number: 3 starting at 2023-07-24T19:36:54.667.880.925 (XO: 1.667885362 - 1.667902362, t=46.8 degC)
   TM values: Max =    60  Min =   -69  Ave =    -4.61  Std =    24.55

 - Packet number: 4 starting at 2023-07-24T19:36:55.501.823.925 (XO: 2.501828031 - 2.501845031, t=46.8 degC)
   TM values: Max =   152  Min =  -151  Ave =    -4.38  Std =    41.29

 - Packet number: 5 starting at 2023-07-24T19:36:56.335.767.925 (XO: 3.335770699 - 3.335787700, t=46.8 degC)
   TM values: Max =    55  Min =   -68  Ave =    -4.61  Std =    22.68

 - Packet number: 6 starting at 2023-07-24T19:36:57.169.710.925 (XO: 4.169713351 - 4.169730351, t=46.8 degC)
   TM values: Max =    41  Min =   -59  Ave =    -4.43  Std =    16.73

 - Packet number: 7 starting at 2023-07-24T19:36:58.003.654.925 (XO: 5.003655937 - 5.003672937, t=46.8 degC)
   TM values: Max =    80  Min =   -72  Ave =    -4.67  Std =    20.62

 - Packet number: 8 starting at 2023-07-24T19:36:58.837.593.925 (XO: 5.837598689 - 5.837615689, t=46.8 degC)
   TM values: Max =    52  Min =   -62  Ave =    -4.42  Std =    19.31

 - Packet number: 9 starting at 2023-07-24T19:36:59.671.536.925 (XO: 6.671541374 - 6.671558374, t=46.8 degC)
   TM values: Max =    53  Min =   -67  Ave =    -4.59  Std =    18.98

 - Packet number: 10 starting at 2023-07-24T19:37:00.505.480.925 (XO: 7.505483992 - 7.505500992, t=46.8 degC)
   TM values: Max =    50  Min =   -68  Ave =    -4.57  Std =    20.52

 - Packet number: 11 starting at 2023-07-24T19:37:01.339.424.925 (XO: 8.339426645 - 8.339443645, t=46.8 degC)
   TM values: Max =    45  Min =   -61  Ave =    -4.75  Std =    19.04

 - Packet number: 12 starting at 2023-07-24T19:37:02.173.366.925 (XO: 9.173369380 - 9.173386380, t=46.8 degC)
   TM values: Max =    63  Min =   -65  Ave =    -4.41  Std =    19.72

 - Packet number: 13 starting at 2023-07-24T19:37:03.007.310.925 (XO: 10.007312048 - 10.007329048, t=46.8 degC)
   TM values: Max =    49  Min =   -61  Ave =    -4.75  Std =    17.79

 - Packet number: 14 starting at 2023-07-24T19:37:03.841.249.925 (XO: 10.841254633 - 10.841271633, t=46.8 degC)
   TM values: Max =    43  Min =   -52  Ave =    -4.56  Std =    15.90

 - Packet number: 15 starting at 2023-07-24T19:37:04.675.192.925 (XO: 11.675197353 - 11.675214353, t=46.8 degC)
   TM values: Max =    49  Min =   -66  Ave =    -4.55  Std =    17.75

 - Packet number: 16 starting at 2023-07-24T19:37:05.509.135.925 (XO: 12.509140054 - 12.509157054, t=46.8 degC)
   TM values: Max =    49  Min =   -54  Ave =    -4.48  Std =    17.97

 - Packet number: 17 starting at 2023-07-24T19:37:06.343.079.925 (XO: 13.343082689 - 13.343099689, t=46.8 degC)
   TM values: Max =    63  Min =   -74  Ave =    -4.84  Std =    19.80

 - Packet number: 18 starting at 2023-07-24T19:37:07.177.022.925 (XO: 14.177025274 - 14.177042274, t=46.8 degC)
   TM values: Max =    69  Min =   -67  Ave =    -4.38  Std =    22.48
