-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 02:11:50 2024 UTC
-- File 20230724_184541_1229.dat, 1251  packets found.
APID:
      1229 (EW) in 1251 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 1251 packets (all packets)
Total number of packets: 
      36516 in 1251 packets (all packets)
Packet No from 1 to 1251
!!!ERROR: Maximum packet number must be = 36516
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36516
Hz and MHz counters from 19.969148 to 19.969148
-each separately from 19s to 19s and from 969148us to 969148us.
   Differences in neigboring packets: 
      0.000000 in 1250 packets (all packets)
MIU packet time from 2023-07-24T18:46:03.969.148.000 to 2023-07-24T18:46:03.969.148.000
   Differences in neigboring packets: 
      0.000000 in 1250 packets (all packets)
MIU orbit number: 
      60000 in 1251 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 1251 packets (all packets)
MIU orbit information / number of events: 
      0 in 1251 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 1251 packets (all packets)
Instrument anomalies: 
      0 (OK) in 1251 packets (all packets)
FPGA configuration version: 
      211 in 1251 packets (all packets)
10 MHz counter at the last 1Hz pulse from 20.6011465 to 20.6011465
   Differences in neigboring packets: 
      0.0000000 in 1250 packets (all packets)
Temperature FPGA: 
      FD -> 58.9 degC in 1251 packets (all packets)
Temperature XO: 
      C7 -> 46.8 degC in 1251 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 1251 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 1251 packets (all packets)
Number of external events: 
      2 in 1251 packets (all packets)
Number of generated alerts: 
      2 in 1251 packets (all packets)
Survey memory error: 
      0 in 1251 packets (all packets)
Event memory error: 
      0 in 1251 packets (all packets)
Configuration error: 
      0 in 1251 packets (all packets)
SDRAM CRC error: 
      0 in 1251 packets (all packets)
Radio gain: 
      0 in 1251 packets (all packets)
Radio bandwidth: 
      0 in 1251 packets (all packets)
Digital gain: 
      0 in 1251 packets (all packets)
Analog gain: 
      0 in 1251 packets (all packets)

EVENT DATA READER
-- File 20230724_184541_1229.dat, 6 EW pages found. Extra packets: 27.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 2
      2 in 1224 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 21.5425530 to 21.5633440
      Difference between the neighboring packets: 
      0.0000170 in 1223 packets (all packets)
   Memory page number from 0 to 5
      Difference between the neighboring packets: 
      0 in 1218 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 5 packets: 204, 408, 612, 816, 1020
   Event page number from 9 to 14
      Difference between the neighboring packets: 
      0 in 1218 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 5 packets: 204, 408, 612, 816, 1020
   MIU event flag 
      1 in 1224 packets (all packets)
   Time triggered event flag 
      0 in 1224 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-24T18:46:03.913.659.648
   Min EW header abs. time found among all events            : 2023-07-24T18:46:03.969.148.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-24T18:46:03.913.659.648
   Max EW header abs. time found among all events            : 2023-07-24T18:46:03.969.148.000
   TM values from -384 to 359 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 1224
    Page no. 9:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 6  ordered packets: 1224
   TM values from -384 to 359 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=19 C1MHz=0.969148
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1223 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 21.5425530
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T18:46:03.969.148.000
   Event 1st sample abs. time : 2023-07-24T18:46:03.913.659.648)
   --- Rel. time correction for 1 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 19, 32bit 10MHz counter = 21.542553000 corrected by 40bit 10MHz counter of 14.601165800 obtained at 1Hz counter of 13. Result: 19.941409413
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              19.941409238
   Event data descriptor XO time (reconstructed seconds from XP start)    :              19.941409413
   Sampling frequency correction factor for this event:     0.999996800
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2023-07-24T18:46:03.913.659.648 to 2023-07-24T18:46:03.934.467.714 (XO: 19.941409238 - 19.962217304)
   Min TM value:-384 at an offset of 4986.578 usec -> 2023-07-24T18:46:03.918.646.226 (XO: 19.946395816)
   Max TM value:359 at an offset of 4898.553 usec -> 2023-07-24T18:46:03.918.558.201 (XO: 19.946307791)
   Max dTM/dt [1/us]:3.30e+03 at an offset of 4943.216 usec -> 2023-07-24T18:46:03.918.602.863 (XO: 19.946352453)
   Min dTM/dt [1/us]:-3.11e+03 at an offset of 4984.278 usec -> 2023-07-24T18:46:03.918.643.926 (XO: 19.946393516)
   Med: -4.00000   Ave: -4.4955438   Std: 94.829636
