-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 02:11:09 2024 UTC
-- File 20230724_173709_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.957112 to 51.313838
-each separately from 0s to 51s and from 313549us to 957401us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.916518 in 1 packets: 18
      7.659337 in 1 packets: 36
      15.779715 in 1 packets: 54
MIU packet time from 2023-07-24T17:37:12.957.112.000 to 2023-07-24T17:38:03.313.838.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.916518 in 1 packets: 18
      7.659337 in 1 packets: 36
      15.779715 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2686384 to 53.2684744
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999132 in 1 packets: 18
      7.9999742 in 1 packets: 36
      15.9999486 in 1 packets: 54
Temperature FPGA: 
      FD -> 58.9 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C7 -> 46.8 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230724_173709_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2257532 to 3.2260422
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.9164310  -- OK for SL --
   MIU time from 2023-07-24T17:37:12.957.112.000 to 2023-07-24T17:37:12.957.401.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.916518

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.1424732 to 30.1427622
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.6593110  -- OK for SL --
   MIU time from 2023-07-24T17:37:39.873.919.000 to 2023-07-24T17:37:39.874.208.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.659337

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.8020732 to 37.8023622
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7796630  -- OK for SL --
   MIU time from 2023-07-24T17:37:47.533.545.000 to 2023-07-24T17:37:47.533.834.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.779715

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.5820252 to 53.5823142
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-24T17:38:03.313.549.000 to 2023-07-24T17:38:03.313.838.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 3.225753200 corrected by 40bit 10MHz counter of 2.268638400 obtained at 1Hz counter of 0. Result: 0.957117863
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996800 - 0.999996900
   Min 1st sample MIU time among all SL packets: 2023-07-24T17:37:12.956.971.725
   Max 1st sample MIU time among all SL packets: 2023-07-24T17:38:03.313.697.725
   Min 1st sample XO time among all SL packets: 0.956977587
   Max 1st sample XO time among all SL packets: 51.313700529
   TM values from -336 to 250 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-24T17:37:12.956.971.725 (XO: 0.956977587 - 0.957283588, t=46.8 degC)  Rank: 00077
   TM values: Max =   206  Min =  -296  Ave =    -4.55  Std =    27.71
   Min TM value:-296 at an offset of 111.925 usec -> 2023-07-24T17:37:12.957.083.650 (XO: 0.957089513)
   Max TM value:206 at an offset of 114.375 usec -> 2023-07-24T17:37:12.957.086.100 (XO: 0.957091963)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 114.200 usec -> 2023-07-24T17:37:12.957.085.925 (XO: 0.957091788)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 111.788 usec -> 2023-07-24T17:37:12.957.083.512 (XO: 0.957089375)

 - Block number: 2 starting at 2023-07-24T17:37:39.873.778.725 (XO: 27.873784121 - 27.874090122, t=46.8 degC)  Rank: 00074
   TM values: Max =   245  Min =  -306  Ave =    -4.50  Std =    32.29
   Min TM value:-306 at an offset of 122.038 usec -> 2023-07-24T17:37:39.873.900.762 (XO: 27.873906159)
   Max TM value:245 at an offset of 120.075 usec -> 2023-07-24T17:37:39.873.898.800 (XO: 27.873904196)
   Max dTM/dt [1/us]:2.38e+03 at an offset of 119.925 usec -> 2023-07-24T17:37:39.873.898.650 (XO: 27.873904046)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 121.925 usec -> 2023-07-24T17:37:39.873.900.650 (XO: 27.873906046)

 - Block number: 3 starting at 2023-07-24T17:37:47.533.404.725 (XO: 35.533408779 - 35.533714780, t=46.8 degC)  Rank: 00078
   TM values: Max =   250  Min =  -336  Ave =    -4.54  Std =    27.41
   Min TM value:-336 at an offset of 109.900 usec -> 2023-07-24T17:37:47.533.514.625 (XO: 35.533518679)
   Max TM value:250 at an offset of 112.250 usec -> 2023-07-24T17:37:47.533.516.975 (XO: 35.533521029)
   Max dTM/dt [1/us]:2.68e+03 at an offset of 112.113 usec -> 2023-07-24T17:37:47.533.516.837 (XO: 35.533520891)
   Min dTM/dt [1/us]:-2.77e+03 at an offset of 109.763 usec -> 2023-07-24T17:37:47.533.514.487 (XO: 35.533518541)

 - Block number: 4 starting at 2023-07-24T17:38:03.313.408.725 (XO: 51.313411528 - 51.313717529, t=46.8 degC)  Rank: 00082
   TM values: Max =   236  Min =  -312  Ave =    -4.44  Std =    31.81
   Min TM value:-312 at an offset of 123.638 usec -> 2023-07-24T17:38:03.313.532.362 (XO: 51.313535166)
   Max TM value:236 at an offset of 121.175 usec -> 2023-07-24T17:38:03.313.529.900 (XO: 51.313532703)
   Max dTM/dt [1/us]:2.61e+03 at an offset of 123.688 usec -> 2023-07-24T17:38:03.313.532.412 (XO: 51.313535216)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 138.525 usec -> 2023-07-24T17:38:03.313.547.250 (XO: 51.313550053)
