-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 02:03:43 2024 UTC
-- File 20230724_151642_1229.dat, 4130  packets found.
APID:
      1229 (EW) in 4130 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 4130 packets (all packets)
Total number of packets: 
      36720 in 4130 packets (all packets)
Packet No from 1 to 4130
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 6.655299 to 6.655299
-each separately from 6s to 6s and from 655299us to 655299us.
   Differences in neigboring packets: 
      0.000000 in 4129 packets (all packets)
MIU packet time from 2023-07-24T15:16:51.655.299.000 to 2023-07-24T15:16:51.655.299.000
   Differences in neigboring packets: 
      0.000000 in 4129 packets (all packets)
MIU orbit number: 
      60000 in 4130 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 4130 packets (all packets)
MIU orbit information / number of events: 
      0 in 4130 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 4130 packets (all packets)
Instrument anomalies: 
      0 (OK) in 4130 packets (all packets)
FPGA configuration version: 
      211 in 4130 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9233848 to 7.9233848
   Differences in neigboring packets: 
      0.0000000 in 4129 packets (all packets)
Temperature FPGA: 
      FD -> 58.9 degC in 4130 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 4130 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 4130 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 4130 packets (all packets)
Number of external events: 
      3 in 4130 packets (all packets)
Number of generated alerts: 
      3 in 4130 packets (all packets)
Survey memory error: 
      0 in 4130 packets (all packets)
Event memory error: 
      0 in 4130 packets (all packets)
Configuration error: 
      0 in 4130 packets (all packets)
SDRAM CRC error: 
      0 in 4130 packets (all packets)
Radio gain: 
      0 in 4130 packets (all packets)
Radio bandwidth: 
      0 in 4130 packets (all packets)
Digital gain: 
      0 in 4130 packets (all packets)
Analog gain: 
      0 in 4130 packets (all packets)

EVENT DATA READER
-- File 20230724_151642_1229.dat, 20 EW pages found. Extra packets: 50.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 3 to 3
      3 in 4080 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 8.5231978 to 8.5925408
      Difference between the neighboring packets: 
      0.0000170 in 4079 packets (all packets)
   Memory page number from 8 to 27
      Difference between the neighboring packets: 
      0 in 4060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 19 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 20
      Difference between the neighboring packets: 
      0 in 4060 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 19 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   MIU event flag 
      1 in 4080 packets (all packets)
   Time triggered event flag 
      0 in 4080 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-24T15:16:51.599.810.825
   Min EW header abs. time found among all events            : 2023-07-24T15:16:51.655.299.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-24T15:16:51.599.810.825
   Max EW header abs. time found among all events            : 2023-07-24T15:16:51.655.299.000
   TM values from -307 to 219 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 4080
    Page no. 1:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 20  ordered packets: 4080
   TM values from -307 to 219 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=6 C1MHz=0.655299
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 4079 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 8.5231978
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T15:16:51.655.299.000
   Event 1st sample abs. time : 2023-07-24T15:16:51.599.810.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2023-07-24T15:16:51.599.810.825 to 2023-07-24T15:16:51.669.170.825 (XO: -333.000000175 - -332.930640175)
   Min TM value:-307 at an offset of 53798.662 usec -> 2023-07-24T15:16:51.653.609.487 (XO: -332.946201513)
   Max TM value:219 at an offset of 53781.550 usec -> 2023-07-24T15:16:51.653.592.375 (XO: -332.946218625)
   Max dTM/dt [1/us]:2.70e+03 at an offset of 53784.062 usec -> 2023-07-24T15:16:51.653.594.887 (XO: -332.946216113)
   Min dTM/dt [1/us]:-2.92e+03 at an offset of 53798.550 usec -> 2023-07-24T15:16:51.653.609.375 (XO: -332.946201625)
   Med: -5.00000   Ave: -4.5367283   Std: 5.7705781
