-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 01:46:50 2024 UTC
-- File 20230724_131616_1229.dat, 4622  packets found.
APID:
      1229 (EW) in 4622 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 4622 packets (all packets)
Total number of packets: 
      36720 in 4622 packets (all packets)
Packet No from 1 to 4622
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 4.818197 to 9.139342
-each separately from 4s to 9s and from 139342us to 818197us.
   Differences in neigboring packets: 
      0.000000 in 4620 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -4.321145 in 1 packets: 3672
MIU packet time from 2023-07-24T13:16:23.818.197.000 to 2023-07-24T13:16:28.139.342.000
   Differences in neigboring packets: 
      0.000000 in 4620 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -4.321145 in 1 packets: 3672
MIU orbit number: 
      60000 in 4622 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 4622 packets (all packets)
MIU orbit information / number of events: 
      0 in 4622 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 4622 packets (all packets)
Instrument anomalies: 
      0 (OK) in 4622 packets (all packets)
FPGA configuration version: 
      211 in 4622 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9433910 to 10.9433748
   Differences in neigboring packets: 
      0.0000000 in 4620 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -4.9999838 in 1 packets: 3672
Temperature FPGA: 
      FF -> 56.3 degC in 4622 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 4622 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 4622 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 4622 packets (all packets)
Number of external events: 
      3 in 3672 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 950 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
Number of generated alerts: 
      3 in 3672 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 950 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
Survey memory error: 
      0 in 4622 packets (all packets)
Event memory error: 
      0 in 4622 packets (all packets)
Configuration error: 
      0 in 4622 packets (all packets)
SDRAM CRC error: 
      0 in 4622 packets (all packets)
Radio gain: 
      0 in 4622 packets (all packets)
Radio bandwidth: 
      0 in 4622 packets (all packets)
Digital gain: 
      0 in 4622 packets (all packets)
Analog gain: 
      0 in 4622 packets (all packets)

EVENT DATA READER
-- File 20230724_131616_1229.dat, 22 EW pages found. Extra packets: 134.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 3
      3 in 3672 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 816 packets: 3673, 3674, 3675, 3676, 3677, 3678, 3679, 3680, 3681, 3682, 3683, 3684, 3685, 3686, 3687, 3688, 3689, 3690, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 6.7061022 to 11.2352932
      Difference between the neighboring packets: 
      0.0000170 in 4486 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -4.5291910 in 1 packets: 3672
   Memory page number from 0 to 73
      Difference between the neighboring packets: 
      0 in 4466 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 20 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3876, ...
      53 in 1 packets: 3672
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 4466 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 20 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3876, ...
      -59 in 1 packets: 3672
   MIU event flag 
      1 in 4488 packets (all packets)
   Time triggered event flag 
      0 in 4488 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-24T13:16:23.762.708.825
   Min EW header abs. time found among all events            : 2023-07-24T13:16:23.818.197.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-24T13:16:28.083.853.825
   Max EW header abs. time found among all events            : 2023-07-24T13:16:28.139.342.000
   TM values from -275 to 251 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 816
    Page no. 1:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 4  ordered packets: 816
   TM values from -256 to 200 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=4 C1MHz=0.818197
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 815 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 6.7061022
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T13:16:23.818.197.000
   Event 1st sample abs. time : 2023-07-24T13:16:23.762.708.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000

 Event no. 3 total number of packets: 3672
    Page no. 43:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 18  ordered packets: 3672
   TM values from -275 to 251 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=9 C1MHz=0.139342
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 3671 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 11.1728862
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T13:16:28.139.342.000
   Event 1st sample abs. time : 2023-07-24T13:16:28.083.853.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2023-07-24T13:16:23.762.708.825 to 2023-07-24T13:16:23.776.580.825 (XO: -333.000000175 - -332.986128175)
   Min TM value:-256 at an offset of 5398.225 usec -> 2023-07-24T13:16:23.768.107.050 (XO: -332.994601950)
   Max TM value:200 at an offset of 5397.987 usec -> 2023-07-24T13:16:23.768.106.812 (XO: -332.994602188)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 5421.750 usec -> 2023-07-24T13:16:23.768.130.575 (XO: -332.994578425)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 5398.062 usec -> 2023-07-24T13:16:23.768.106.887 (XO: -332.994602113)
   Med: -4.00000   Ave: -4.4724886   Std: 7.6597015

 - Event number: 3 from 2023-07-24T13:16:28.083.853.825 to 2023-07-24T13:16:28.146.277.825 (XO: -333.000000175 - -332.937576175)
   Min TM value:-275 at an offset of 28846.163 usec -> 2023-07-24T13:16:28.112.699.987 (XO: -332.971154013)
   Max TM value:251 at an offset of 28843.737 usec -> 2023-07-24T13:16:28.112.697.562 (XO: -332.971156438)
   Max dTM/dt [1/us]:2.54e+03 at an offset of 28846.237 usec -> 2023-07-24T13:16:28.112.700.062 (XO: -332.971153938)
   Min dTM/dt [1/us]:-3.00e+03 at an offset of 28843.838 usec -> 2023-07-24T13:16:28.112.697.662 (XO: -332.971156338)
   Med: -5.00000   Ave: -4.5293523   Std: 6.9824363
