-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 01:18:02 2024 UTC
-- File 20230724_085431_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177024
-each separately from 0s to 14s and from 0us to 841250us.
   Differences in neigboring packets: 
      0.833938 in 2 packets: 1, 13
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 15, 17
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 14, 16
      0.833939 in 1 packets: 7
MIU packet time from 2023-07-24T08:54:34.000.000.000 to 2023-07-24T08:54:48.177.024.000
   Differences in neigboring packets: 
      0.833938 in 2 packets: 1, 13
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 15, 17
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 14, 16
      0.833939 in 1 packets: 7
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.6472890 to 15.6472433
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999968 in 5 packets: 2, 4, 8, 12, 16
      0.9999967 in 9 packets: 3, 5, 6, 9, 10, 11, 14, 15, 17
   XO PPM: 
      -3.2 in 5 packets: 2, 4, 8, 12, 16
      -3.3 in 9 packets: 3, 5, 6, 9, 10, 11, 14, 15, 17
Temperature FPGA: 
      FF -> 56.3 degC in 15 packets: 1, 2, 5, 6, 7, 8, 9, 10, 11, 13, 14, 15, 16, 17, 18
      FE -> 57.6 degC in 2 packets: 3, 4
      00 -> 55.0 degC in 1 packets: 12
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      02 -> 100.7 degC in 1 packets: 1
      01 -> 108.0 degC in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 10 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      2 in 2 packets: 17, 18
Number of generated alerts: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 10 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      2 in 2 packets: 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230724_085431_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.6472891 to 15.8242691
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-24T08:54:34.000.000.000 to 2023-07-24T08:54:48.177.024.000
      Difference between neigboring packets: 
      0.833938 in 2 packets: 1, 13
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 15, 17
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 14, 16
      0.833939 in 1 packets: 7
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.647289100 corrected by 40bit 10MHz counter of 1.647289000 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SR packets: 2023-07-24T08:54:33.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-24T08:54:48.177.023.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177026309
   TM values from -33 to 25 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-24T08:54:33.999.999.925 (XO: 0.000000025 - 0.000017025, t=45.2 degC)
   TM values: Max =     9  Min =   -17  Ave =    -4.56  Std =     5.25

 - Packet number: 2 starting at 2023-07-24T08:54:34.833.937.925 (XO: 0.833942694 - 0.833959694, t=45.2 degC)
   TM values: Max =    10  Min =   -17  Ave =    -4.39  Std =     5.30

 - Packet number: 3 starting at 2023-07-24T08:54:35.667.880.925 (XO: 1.667885429 - 1.667902429, t=45.2 degC)
   TM values: Max =    11  Min =   -18  Ave =    -4.41  Std =     5.70

 - Packet number: 4 starting at 2023-07-24T08:54:36.501.824.925 (XO: 2.501828131 - 2.501845131, t=45.2 degC)
   TM values: Max =    21  Min =   -28  Ave =    -4.30  Std =    13.29

 - Packet number: 5 starting at 2023-07-24T08:54:37.335.767.925 (XO: 3.335770833 - 3.335787833, t=45.2 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.23  Std =     6.60

 - Packet number: 6 starting at 2023-07-24T08:54:38.169.711.925 (XO: 4.169713585 - 4.169730585, t=45.2 degC)
   TM values: Max =    15  Min =   -28  Ave =    -4.25  Std =    11.22

 - Packet number: 7 starting at 2023-07-24T08:54:39.003.654.925 (XO: 5.003656337 - 5.003673337, t=45.2 degC)
   TM values: Max =    12  Min =   -21  Ave =    -4.52  Std =     8.09

 - Packet number: 8 starting at 2023-07-24T08:54:39.837.593.925 (XO: 5.837599005 - 5.837616005, t=45.2 degC)
   TM values: Max =     9  Min =   -19  Ave =    -4.22  Std =     7.16

 - Packet number: 9 starting at 2023-07-24T08:54:40.671.537.925 (XO: 6.671541741 - 6.671558741, t=45.2 degC)
   TM values: Max =    10  Min =   -19  Ave =    -4.41  Std =     6.90

 - Packet number: 10 starting at 2023-07-24T08:54:41.505.480.925 (XO: 7.505484493 - 7.505501493, t=45.2 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.55  Std =     7.01

 - Packet number: 11 starting at 2023-07-24T08:54:42.339.424.925 (XO: 8.339427245 - 8.339444245, t=45.2 degC)
   TM values: Max =     8  Min =   -18  Ave =    -4.29  Std =     5.92

 - Packet number: 12 starting at 2023-07-24T08:54:43.173.367.925 (XO: 9.173369980 - 9.173386980, t=45.2 degC)
   TM values: Max =    16  Min =   -24  Ave =    -4.48  Std =     8.07

 - Packet number: 13 starting at 2023-07-24T08:54:44.007.311.925 (XO: 10.007312649 - 10.007329649, t=45.2 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.60  Std =     8.36

 - Packet number: 14 starting at 2023-07-24T08:54:44.841.249.925 (XO: 10.841255401 - 10.841272401, t=45.2 degC)
   TM values: Max =    13  Min =   -23  Ave =    -4.42  Std =     9.28

 - Packet number: 15 starting at 2023-07-24T08:54:45.675.193.925 (XO: 11.675198153 - 11.675215153, t=45.2 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.39  Std =     8.45

 - Packet number: 16 starting at 2023-07-24T08:54:46.509.136.925 (XO: 12.509140854 - 12.509157854, t=45.2 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.33  Std =     7.97

 - Packet number: 17 starting at 2023-07-24T08:54:47.343.080.925 (XO: 13.343083557 - 13.343100557, t=45.2 degC)
   TM values: Max =     6  Min =   -15  Ave =    -4.30  Std =     3.88

 - Packet number: 18 starting at 2023-07-24T08:54:48.177.023.925 (XO: 14.177026309 - 14.177043309, t=45.2 degC)
   TM values: Max =    25  Min =   -33  Ave =    -4.22  Std =    16.53
