-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 01:01:01 2024 UTC
-- File 20230724_080506_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.969926 to 18.060070
-each separately from 9s to 18s and from 59781us to 970215us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.089566 in 1 packets: 18
MIU packet time from 2023-07-24T08:05:18.969.926.000 to 2023-07-24T08:05:27.060.070.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.089566 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.0709191 to 19.0708897
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999706 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230724_080506_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.0408478 to 11.0411368
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0895350  -- OK for SL --
   MIU time from 2023-07-24T08:05:18.969.926.000 to 2023-07-24T08:05:18.970.215.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.089566

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1306718 to 19.1309608
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-24T08:05:27.059.781.000 to 2023-07-24T08:05:27.060.070.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 30 XO calibration points at 1Hz counter from 0 to 29
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.040847800 corrected by 40bit 10MHz counter of 10.070919100 obtained at 1Hz counter of 9. Result: 9.969931901
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2023-07-24T08:05:18.969.785.725
   Max 1st sample MIU time among all SL packets: 2023-07-24T08:05:27.059.929.725
   Min 1st sample XO time among all SL packets: 9.969791625
   Max 1st sample XO time among all SL packets: 18.059931017
   TM values from -286 to 200 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-24T08:05:18.969.785.725 (XO: 9.969791625 - 9.970097626, t=45.2 degC)  Rank: 00086
   TM values: Max =   200  Min =  -286  Ave =    -4.43  Std =    26.71
   Min TM value:-286 at an offset of 122.500 usec -> 2023-07-24T08:05:18.969.908.225 (XO: 9.969914126)
   Max TM value:200 at an offset of 155.926 usec -> 2023-07-24T08:05:18.969.941.650 (XO: 9.969947551)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 124.913 usec -> 2023-07-24T08:05:18.969.910.637 (XO: 9.969916538)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 122.400 usec -> 2023-07-24T08:05:18.969.908.125 (XO: 9.969914026)

 - Block number: 2 starting at 2023-07-24T08:05:27.059.640.725 (XO: 18.059642016 - 18.059948017, t=45.2 degC)  Rank: 00079
   TM values: Max =   181  Min =  -213  Ave =    -4.48  Std =    20.63
   Min TM value:-213 at an offset of 127.475 usec -> 2023-07-24T08:05:27.059.768.200 (XO: 18.059769491)
   Max TM value:181 at an offset of 130.113 usec -> 2023-07-24T08:05:27.059.770.837 (XO: 18.059772129)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 127.588 usec -> 2023-07-24T08:05:27.059.768.312 (XO: 18.059769604)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 130.225 usec -> 2023-07-24T08:05:27.059.770.950 (XO: 18.059772241)
