-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 00:35:59 2024 UTC
-- File 20230724_063538_1229.dat, 2774  packets found.
APID:
      1229 (EW) in 2774 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 2774 packets (all packets)
Total number of packets: 
      36720 in 2774 packets (all packets)
Packet No from 1 to 2774
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 47.497928 to 47.497928
-each separately from 47s to 47s and from 497928us to 497928us.
   Differences in neigboring packets: 
      0.000000 in 2773 packets (all packets)
MIU packet time from 2023-07-24T06:36:28.497.928.000 to 2023-07-24T06:36:28.497.928.000
   Differences in neigboring packets: 
      0.000000 in 2773 packets (all packets)
MIU orbit number: 
      60000 in 2774 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 2774 packets (all packets)
MIU orbit information / number of events: 
      0 in 2774 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 2774 packets (all packets)
Instrument anomalies: 
      0 (OK) in 2774 packets (all packets)
FPGA configuration version: 
      211 in 2774 packets (all packets)
10 MHz counter at the last 1Hz pulse from 48.3879486 to 48.3879486
   Differences in neigboring packets: 
      0.0000000 in 2773 packets (all packets)
Temperature FPGA: 
      FE -> 57.6 degC in 2774 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 2774 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 2774 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 2774 packets (all packets)
Number of external events: 
      2 in 2774 packets (all packets)
Number of generated alerts: 
      2 in 2774 packets (all packets)
Survey memory error: 
      0 in 2774 packets (all packets)
Event memory error: 
      0 in 2774 packets (all packets)
Configuration error: 
      0 in 2774 packets (all packets)
SDRAM CRC error: 
      0 in 2774 packets (all packets)
Radio gain: 
      0 in 2774 packets (all packets)
Radio bandwidth: 
      0 in 2774 packets (all packets)
Digital gain: 
      0 in 2774 packets (all packets)
Analog gain: 
      0 in 2774 packets (all packets)

EVENT DATA READER
-- File 20230724_063538_1229.dat, 13 EW pages found. Extra packets: 122.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 2
      2 in 2652 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 48.8303909 to 48.8754579
      Difference between the neighboring packets: 
      0.0000170 in 2651 packets (all packets)
   Memory page number from 39 to 51
      Difference between the neighboring packets: 
      0 in 2639 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 12 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448
   Event page number from 1 to 13
      Difference between the neighboring packets: 
      0 in 2639 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 12 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448
   MIU event flag 
      1 in 2652 packets (all packets)
   Time triggered event flag 
      0 in 2652 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-24T06:36:28.442.439.648
   Min EW header abs. time found among all events            : 2023-07-24T06:36:28.497.928.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-24T06:36:28.442.439.648
   Max EW header abs. time found among all events            : 2023-07-24T06:36:28.497.928.000
   TM values from -245 to 197 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 2652
    Page no. 1:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 13  ordered packets: 2652
   TM values from -245 to 197 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=47 C1MHz=0.497928
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 2651 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 48.8303909
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-24T06:36:28.497.928.000
   Event 1st sample abs. time : 2023-07-24T06:36:28.442.439.648)
   --- Rel. time correction for 1 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 47, 32bit 10MHz counter = 48.830390900 corrected by 40bit 10MHz counter of 44.387961500 obtained at 1Hz counter of 43. Result: 47.442443616
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              47.442443441
   Event data descriptor XO time (reconstructed seconds from XP start)    :              47.442443616
   Sampling frequency correction factor for this event:     0.999996800
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2023-07-24T06:36:28.442.439.648 to 2023-07-24T06:36:28.487.523.792 (XO: 47.442443441 - 47.487527585)
   Min TM value:-245 at an offset of 43214.426 usec -> 2023-07-24T06:36:28.485.654.073 (XO: 47.485657867)
   Max TM value:197 at an offset of 13212.342 usec -> 2023-07-24T06:36:28.455.651.990 (XO: 47.455655783)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 43214.551 usec -> 2023-07-24T06:36:28.485.654.198 (XO: 47.485657992)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 13212.455 usec -> 2023-07-24T06:36:28.455.652.102 (XO: 47.455655896)
   Med: -4.00000   Ave: -4.4479386   Std: 7.1455650
