-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 11 23:28:39 2024 UTC
-- File 20230724_020707_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177024
-each separately from 0s to 14s and from 0us to 841250us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 5, 8, 10, 11, 14, 16
      0.833944 in 6 packets: 3, 6, 9, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU packet time from 2023-07-24T02:07:10.000.000.000 to 2023-07-24T02:07:24.177.024.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 5, 8, 10, 11, 14, 16
      0.833944 in 6 packets: 3, 6, 9, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5197322 to 15.5196874
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999969 in 4 packets: 2, 5, 8, 10
      0.9999967 in 4 packets: 3, 6, 9, 11
      0.9999968 in 6 packets: 4, 12, 14, 15, 16, 17
   XO PPM: 
      -3.1 in 4 packets: 2, 5, 8, 10
      -3.3 in 4 packets: 3, 6, 9, 11
      -3.2 in 6 packets: 4, 12, 14, 15, 16, 17
Temperature FPGA: 
      FF -> 56.3 degC in 11 packets: 1, 4, 5, 6, 7, 8, 11, 12, 13, 14, 15
      FE -> 57.6 degC in 4 packets: 2, 10, 16, 17
      00 -> 55.0 degC in 3 packets: 3, 9, 18
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 10 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      2 in 2 packets: 17, 18
Number of generated alerts: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 10 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      2 in 2 packets: 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230724_020707_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5197323 to 15.6967123
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-24T02:07:10.000.000.000 to 2023-07-24T02:07:24.177.024.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 5, 8, 10, 11, 14, 16
      0.833944 in 6 packets: 3, 6, 9, 12, 15, 17
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.519732300 corrected by 40bit 10MHz counter of 1.519732200 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996700 - 0.999996900
   Min 1st sample MIU time among all SR packets: 2023-07-24T02:07:09.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-24T02:07:24.177.023.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177025391
   TM values from -154 to 164 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-24T02:07:09.999.999.925 (XO: 0.000000025 - 0.000017025, t=45.2 degC)
   TM values: Max =   164  Min =  -154  Ave =    -4.34  Std =    37.33

 - Packet number: 2 starting at 2023-07-24T02:07:10.833.937.925 (XO: 0.833942610 - 0.833959610, t=45.2 degC)
   TM values: Max =    13  Min =   -26  Ave =    -4.52  Std =     6.05

 - Packet number: 3 starting at 2023-07-24T02:07:11.667.880.925 (XO: 1.667885329 - 1.667902329, t=45.2 degC)
   TM values: Max =    19  Min =   -27  Ave =    -4.47  Std =     8.46

 - Packet number: 4 starting at 2023-07-24T02:07:12.501.824.925 (XO: 2.501828031 - 2.501845031, t=45.2 degC)
   TM values: Max =    16  Min =   -27  Ave =    -4.59  Std =     6.79

 - Packet number: 5 starting at 2023-07-24T02:07:13.335.767.925 (XO: 3.335770666 - 3.335787666, t=45.2 degC)
   TM values: Max =    11  Min =   -25  Ave =    -4.75  Std =     5.51

 - Packet number: 6 starting at 2023-07-24T02:07:14.169.710.925 (XO: 4.169713285 - 4.169730285, t=45.2 degC)
   TM values: Max =    11  Min =   -20  Ave =    -4.57  Std =     5.88

 - Packet number: 7 starting at 2023-07-24T02:07:15.003.654.925 (XO: 5.003656036 - 5.003673036, t=45.2 degC)
   TM values: Max =    13  Min =   -25  Ave =    -4.27  Std =     6.40

 - Packet number: 8 starting at 2023-07-24T02:07:15.837.593.925 (XO: 5.837598622 - 5.837615622, t=45.2 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.63  Std =     6.10

 - Packet number: 9 starting at 2023-07-24T02:07:16.671.536.925 (XO: 6.671541341 - 6.671558341, t=45.2 degC)
   TM values: Max =    17  Min =   -27  Ave =    -4.51  Std =     7.67

 - Packet number: 10 starting at 2023-07-24T02:07:17.505.480.925 (XO: 7.505483992 - 7.505500992, t=45.2 degC)
   TM values: Max =    14  Min =   -25  Ave =    -4.34  Std =     6.96

 - Packet number: 11 starting at 2023-07-24T02:07:18.339.423.925 (XO: 8.339426645 - 8.339443645, t=45.2 degC)
   TM values: Max =    20  Min =   -27  Ave =    -4.29  Std =     8.33

 - Packet number: 12 starting at 2023-07-24T02:07:19.173.366.925 (XO: 9.173369380 - 9.173386380, t=45.2 degC)
   TM values: Max =    17  Min =   -24  Ave =    -4.27  Std =     7.69

 - Packet number: 13 starting at 2023-07-24T02:07:20.007.310.925 (XO: 10.007312048 - 10.007329048, t=45.2 degC)
   TM values: Max =    16  Min =   -23  Ave =    -4.47  Std =     7.61

 - Packet number: 14 starting at 2023-07-24T02:07:20.841.249.925 (XO: 10.841254717 - 10.841271717, t=45.2 degC)
   TM values: Max =    17  Min =   -29  Ave =    -4.48  Std =     6.87

 - Packet number: 15 starting at 2023-07-24T02:07:21.675.192.925 (XO: 11.675197386 - 11.675214386, t=45.2 degC)
   TM values: Max =    17  Min =   -26  Ave =    -4.28  Std =     7.02

 - Packet number: 16 starting at 2023-07-24T02:07:22.509.136.925 (XO: 12.509140054 - 12.509157054, t=45.2 degC)
   TM values: Max =    11  Min =   -20  Ave =    -4.55  Std =     4.85

 - Packet number: 17 starting at 2023-07-24T02:07:23.343.079.925 (XO: 13.343082723 - 13.343099723, t=45.2 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.35  Std =     5.82

 - Packet number: 18 starting at 2023-07-24T02:07:24.177.023.925 (XO: 14.177025391 - 14.177042392, t=45.2 degC)
   TM values: Max =    14  Min =   -21  Ave =    -4.25  Std =     5.69
