-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 18:43:50 2024 UTC
-- File 20230721_223450_1229.dat, 5446  packets found.
APID:
      1229 (EW) in 5446 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 5446 packets (all packets)
Total number of packets: 
      36720 in 5446 packets (all packets)
Packet No from 1 to 5446
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 22.276917 to 22.276917
-each separately from 22s to 22s and from 276917us to 276917us.
   Differences in neigboring packets: 
      0.000000 in 5445 packets (all packets)
MIU packet time from 2023-07-21T22:35:15.276.917.000 to 2023-07-21T22:35:15.276.917.000
   Differences in neigboring packets: 
      0.000000 in 5445 packets (all packets)
MIU orbit number: 
      60000 in 5446 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 5446 packets (all packets)
MIU orbit information / number of events: 
      0 in 5446 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 5446 packets (all packets)
Instrument anomalies: 
      0 (OK) in 5446 packets (all packets)
FPGA configuration version: 
      211 in 5446 packets (all packets)
10 MHz counter at the last 1Hz pulse from 23.7024785 to 23.7024785
   Differences in neigboring packets: 
      0.0000000 in 5445 packets (all packets)
Temperature FPGA: 
      FF -> 56.3 degC in 5446 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 5446 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 5446 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 5446 packets (all packets)
Number of external events: 
      3 in 5446 packets (all packets)
Number of generated alerts: 
      3 in 5446 packets (all packets)
Survey memory error: 
      0 in 5446 packets (all packets)
Event memory error: 
      0 in 5446 packets (all packets)
Configuration error: 
      0 in 5446 packets (all packets)
SDRAM CRC error: 
      0 in 5446 packets (all packets)
Radio gain: 
      0 in 5446 packets (all packets)
Radio bandwidth: 
      0 in 5446 packets (all packets)
Digital gain: 
      0 in 5446 packets (all packets)
Analog gain: 
      0 in 5446 packets (all packets)

EVENT DATA READER
-- File 20230721_223450_1229.dat, 26 EW pages found. Extra packets: 142.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 3 to 3
      3 in 5304 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 23.9239087 to 24.1319717
      Difference between the neighboring packets: 
      0.0000170 in 5302 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -0.2080630 in 1 packets: 204
   Memory page number from 0 to 83
      Difference between the neighboring packets: 
      0 in 5278 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      59 in 1 packets: 204
      1 in 24 packets: 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 5278 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -59 in 1 packets: 204
      1 in 24 packets: 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
   MIU event flag 
      1 in 5304 packets (all packets)
   Time triggered event flag 
      0 in 5304 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-21T22:35:15.221.428.648
   Min EW header abs. time found among all events            : 2023-07-21T22:35:15.276.917.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-21T22:35:15.221.428.648
   Max EW header abs. time found among all events            : 2023-07-21T22:35:15.276.917.000
   TM values from -326 to 246 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 5304
    Page no. 1:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 26  ordered packets: 5304
   TM values from -326 to 246 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=22 C1MHz=0.276917
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 5302 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      0.1179290 in 1 packets: 5100
      -- ERROR -- incorrect differences for packets of this event. 
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-21T22:35:15.276.917.000
   Event 1st sample abs. time : 2023-07-21T22:35:15.221.428.648)
   --- Rel. time correction for 1 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 22, 32bit 10MHz counter = 23.923908700 corrected by 40bit 10MHz counter of 14.702507900 obtained at 1Hz counter of 13. Result: 22.221430309
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              22.221430134
   Event data descriptor XO time (reconstructed seconds from XP start)    :              22.221430309
   Sampling frequency correction factor for this event:     0.999996800
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2023-07-21T22:35:15.221.428.648 to 2023-07-21T22:35:15.311.596.936 (XO: 22.221430134 - 22.311598422)
   Min TM value:-326 at an offset of 73005.796 usec -> 2023-07-21T22:35:15.294.434.444 (XO: 22.294435930)
   Max TM value:246 at an offset of 52971.332 usec -> 2023-07-21T22:35:15.274.399.980 (XO: 22.274401466)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 73005.921 usec -> 2023-07-21T22:35:15.294.434.569 (XO: 22.294436055)
   Min dTM/dt [1/us]:-2.83e+03 at an offset of 73003.471 usec -> 2023-07-21T22:35:15.294.432.119 (XO: 22.294433605)
   Med: -4.00000   Ave: -4.4317636   Std: 10.488283
