-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 18:13:17 2024 UTC
-- File 20230721_213224_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177025
-each separately from 0s to 14s and from 0us to 841251us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 5, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 4, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU packet time from 2023-07-21T21:32:27.000.000.000 to 2023-07-21T21:32:41.177.025.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 5, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 4, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2478792 to 16.2478334
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999968 in 5 packets: 2, 8, 11, 14, 17
      0.9999967 in 8 packets: 3, 4, 5, 6, 9, 10, 15, 16
      0.9999966 in 1 packets: 12
   XO PPM: 
      -3.2 in 5 packets: 2, 8, 11, 14, 17
      -3.3 in 8 packets: 3, 4, 5, 6, 9, 10, 15, 16
      -3.4 in 1 packets: 12
Temperature FPGA: 
      FF -> 56.3 degC in 8 packets: 1, 4, 6, 7, 8, 9, 10, 17
      FE -> 57.6 degC in 9 packets: 2, 3, 11, 12, 13, 14, 15, 16, 18
      FD -> 58.9 degC in 1 packets: 5
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 12 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 12 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230721_213224_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2478793 to 16.4248593
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-21T21:32:27.000.000.000 to 2023-07-21T21:32:41.177.025.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 5, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 4, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.247879300 corrected by 40bit 10MHz counter of 2.247879200 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SR packets: 2023-07-21T21:32:26.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-21T21:32:41.177.024.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177026391
   TM values from -78 to 48 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-21T21:32:26.999.999.925 (XO: 0.000000025 - 0.000017025, t=45.2 degC)
   TM values: Max =    31  Min =   -38  Ave =    -4.35  Std =    11.68

 - Packet number: 2 starting at 2023-07-21T21:32:27.833.937.925 (XO: 0.833942694 - 0.833959694, t=45.2 degC)
   TM values: Max =    30  Min =   -47  Ave =    -4.51  Std =    11.13

 - Packet number: 3 starting at 2023-07-21T21:32:28.667.880.925 (XO: 1.667885429 - 1.667902429, t=45.2 degC)
   TM values: Max =    28  Min =   -46  Ave =    -4.38  Std =    11.19

 - Packet number: 4 starting at 2023-07-21T21:32:29.501.824.925 (XO: 2.501828181 - 2.501845181, t=45.2 degC)
   TM values: Max =    39  Min =   -50  Ave =    -4.31  Std =    15.12

 - Packet number: 5 starting at 2023-07-21T21:32:30.335.768.925 (XO: 3.335770933 - 3.335787933, t=45.2 degC)
   TM values: Max =    38  Min =   -48  Ave =    -4.56  Std =    16.02

 - Packet number: 6 starting at 2023-07-21T21:32:31.169.711.925 (XO: 4.169713685 - 4.169730685, t=45.2 degC)
   TM values: Max =    37  Min =   -43  Ave =    -4.17  Std =    15.68

 - Packet number: 7 starting at 2023-07-21T21:32:32.003.654.925 (XO: 5.003656437 - 5.003673437, t=45.2 degC)
   TM values: Max =    37  Min =   -42  Ave =    -4.46  Std =    13.39

 - Packet number: 8 starting at 2023-07-21T21:32:32.837.593.925 (XO: 5.837599105 - 5.837616105, t=45.2 degC)
   TM values: Max =    26  Min =   -38  Ave =    -4.63  Std =    11.41

 - Packet number: 9 starting at 2023-07-21T21:32:33.671.537.925 (XO: 6.671541841 - 6.671558841, t=45.2 degC)
   TM values: Max =    22  Min =   -32  Ave =    -4.43  Std =     9.91

 - Packet number: 10 starting at 2023-07-21T21:32:34.505.480.925 (XO: 7.505484593 - 7.505501593, t=45.2 degC)
   TM values: Max =    34  Min =   -39  Ave =    -4.50  Std =    12.88

 - Packet number: 11 starting at 2023-07-21T21:32:35.339.424.925 (XO: 8.339427311 - 8.339444311, t=45.2 degC)
   TM values: Max =    36  Min =   -43  Ave =    -4.48  Std =    11.98

 - Packet number: 12 starting at 2023-07-21T21:32:36.173.367.925 (XO: 9.173370014 - 9.173387015, t=45.2 degC)
   TM values: Max =    27  Min =   -33  Ave =    -4.27  Std =     9.97

 - Packet number: 13 starting at 2023-07-21T21:32:37.007.311.925 (XO: 10.007312848 - 10.007329848, t=45.2 degC)
   TM values: Max =    30  Min =   -47  Ave =    -4.56  Std =    11.15

 - Packet number: 14 starting at 2023-07-21T21:32:37.841.250.925 (XO: 10.841255517 - 10.841272517, t=45.2 degC)
   TM values: Max =    48  Min =   -78  Ave =    -4.57  Std =    16.11

 - Packet number: 15 starting at 2023-07-21T21:32:38.675.193.925 (XO: 11.675198253 - 11.675215253, t=45.2 degC)
   TM values: Max =    31  Min =   -33  Ave =    -4.30  Std =    11.46

 - Packet number: 16 starting at 2023-07-21T21:32:39.509.137.925 (XO: 12.509141005 - 12.509158005, t=45.2 degC)
   TM values: Max =    38  Min =   -50  Ave =    -4.40  Std =    15.38

 - Packet number: 17 starting at 2023-07-21T21:32:40.343.080.925 (XO: 13.343083723 - 13.343100723, t=45.2 degC)
   TM values: Max =    31  Min =   -45  Ave =    -4.41  Std =    12.92

 - Packet number: 18 starting at 2023-07-21T21:32:41.177.024.925 (XO: 14.177026391 - 14.177043392, t=45.2 degC)
   TM values: Max =    35  Min =   -43  Ave =    -4.30  Std =    15.68
