-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 17:46:43 2024 UTC
-- File 20230721_194902_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.218716 to 67.168001
-each separately from 12s to 67s and from 167712us to 898216us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.479487 in 1 packets: 18
      16.949095 in 1 packets: 36
      6.249762 in 1 packets: 54
      21.269496 in 1 packets: 72
MIU packet time from 2023-07-21T19:49:17.218.716.000 to 2023-07-21T19:50:12.168.001.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.479487 in 1 packets: 18
      16.949095 in 1 packets: 36
      6.249762 in 1 packets: 54
      21.269496 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.6962853 to 68.6961058
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999674 in 1 packets: 18
      16.9999445 in 1 packets: 36
      5.9999805 in 1 packets: 54
      21.9999281 in 1 packets: 72
Temperature FPGA: 
      FE -> 57.6 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20230721_194902_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9150029 to 13.9152919
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4794550  -- OK for SL --
   MIU time from 2023-07-21T19:49:17.218.716.000 to 2023-07-21T19:49:17.219.005.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.479487

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3947469 to 24.3950359
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9490390  -- OK for SL --
   MIU time from 2023-07-21T19:49:27.698.492.000 to 2023-07-21T19:49:27.698.781.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.949095

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      134 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.3440749 to 41.3443639
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.2497430  -- OK for SL --
   MIU time from 2023-07-21T19:49:44.647.876.000 to 2023-07-21T19:49:44.648.165.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.249762

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.5941069 to 47.5943959
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2694230  -- OK for SL --
   MIU time from 2023-07-21T19:49:50.897.927.000 to 2023-07-21T19:49:50.898.216.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.269496

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.8638189 to 68.8641079
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-21T19:50:12.167.712.000 to 2023-07-21T19:50:12.168.001.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.915002900 corrected by 40bit 10MHz counter of 13.696285300 obtained at 1Hz counter of 12. Result: 12.218718300
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2023-07-21T19:49:17.218.575.725
   Max 1st sample MIU time among all SL packets: 2023-07-21T19:50:12.167.860.725
   Min 1st sample XO time among all SL packets: 12.218578024
   Max 1st sample XO time among all SL packets: 67.167862379
   TM values from -991 to 1038 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-21T19:49:17.218.575.725 (XO: 12.218578024 - 12.218884025, t=45.2 degC)  Rank: 00077
   TM values: Max =   213  Min =  -293  Ave =    -4.45  Std =    29.61
   Min TM value:-293 at an offset of 113.950 usec -> 2023-07-21T19:49:17.218.689.675 (XO: 12.218691975)
   Max TM value:213 at an offset of 116.438 usec -> 2023-07-21T19:49:17.218.692.162 (XO: 12.218694462)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 116.300 usec -> 2023-07-21T19:49:17.218.692.025 (XO: 12.218694325)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 113.800 usec -> 2023-07-21T19:49:17.218.689.525 (XO: 12.218691825)

 - Block number: 2 starting at 2023-07-21T19:49:27.698.351.725 (XO: 22.698356230 - 22.698662231, t=45.2 degC)  Rank: 00080
   TM values: Max =   234  Min =  -227  Ave =    -4.50  Std =    26.30
   Min TM value:-227 at an offset of 112.938 usec -> 2023-07-21T19:49:27.698.464.662 (XO: 22.698469167)
   Max TM value:234 at an offset of 117.800 usec -> 2023-07-21T19:49:27.698.469.525 (XO: 22.698474030)
   Max dTM/dt [1/us]:1.28e+03 at an offset of 115.413 usec -> 2023-07-21T19:49:27.698.467.137 (XO: 22.698471642)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 117.913 usec -> 2023-07-21T19:49:27.698.469.637 (XO: 22.698474142)

 - Block number: 3 starting at 2023-07-21T19:49:44.647.735.725 (XO: 39.647739563 - 39.648045564, t=45.2 degC)  Rank: 00134
   TM values: Max =  1038  Min =  -991  Ave =    -4.46  Std =    39.85
   Min TM value:-991 at an offset of 98.725 usec -> 2023-07-21T19:49:44.647.834.450 (XO: 39.647838288)
   Max TM value:1038 at an offset of 98.613 usec -> 2023-07-21T19:49:44.647.834.337 (XO: 39.647838175)
   Max dTM/dt [1/us]:1.36e+04 at an offset of 98.763 usec -> 2023-07-21T19:49:44.647.834.487 (XO: 39.647838325)
   Min dTM/dt [1/us]:-1.27e+04 at an offset of 98.663 usec -> 2023-07-21T19:49:44.647.834.387 (XO: 39.647838225)

 - Block number: 4 starting at 2023-07-21T19:49:50.897.786.725 (XO: 45.897791888 - 45.898097889, t=45.2 degC)  Rank: 00072
   TM values: Max =   239  Min =  -320  Ave =    -4.47  Std =    35.83
   Min TM value:-320 at an offset of 104.113 usec -> 2023-07-21T19:49:50.897.890.837 (XO: 45.897896001)
   Max TM value:239 at an offset of 189.251 usec -> 2023-07-21T19:49:50.897.975.975 (XO: 45.897981138)
   Max dTM/dt [1/us]:2.92e+03 at an offset of 104.250 usec -> 2023-07-21T19:49:50.897.890.975 (XO: 45.897896138)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 104.063 usec -> 2023-07-21T19:49:50.897.890.787 (XO: 45.897895951)

 - Block number: 5 starting at 2023-07-21T19:50:12.167.571.725 (XO: 67.167573378 - 67.167879379, t=45.2 degC)  Rank: 00075
   TM values: Max =   242  Min =  -230  Ave =    -4.52  Std =    30.24
   Min TM value:-230 at an offset of 132.713 usec -> 2023-07-21T19:50:12.167.704.437 (XO: 67.167706091)
   Max TM value:242 at an offset of 120.875 usec -> 2023-07-21T19:50:12.167.692.600 (XO: 67.167694253)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 147.888 usec -> 2023-07-21T19:50:12.167.719.612 (XO: 67.167721266)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 132.575 usec -> 2023-07-21T19:50:12.167.704.300 (XO: 67.167705953)
