-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 17:02:22 2024 UTC
-- File 20230721_170947_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.778195 to 28.300118
-each separately from 13s to 28s and from 299829us to 778484us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.521345 in 1 packets: 18
MIU packet time from 2023-07-21T17:10:03.778.195.000 to 2023-07-21T17:10:18.300.118.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.521345 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.0270376 to 30.0269884
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999508 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230721_170947_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.8052350 to 15.8055240
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5212950  -- OK for SL --
   MIU time from 2023-07-21T17:10:03.778.195.000 to 2023-07-21T17:10:03.778.484.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.521345

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.3268190 to 30.3271080
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-21T17:10:18.299.829.000 to 2023-07-21T17:10:18.300.118.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.805235000 corrected by 40bit 10MHz counter of 15.027037600 obtained at 1Hz counter of 13. Result: 13.778199968
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-07-21T17:10:03.778.054.725
   Max 1st sample MIU time among all SL packets: 2023-07-21T17:10:18.299.977.725
   Min 1st sample XO time among all SL packets: 13.778059693
   Max 1st sample XO time among all SL packets: 28.299980315
   TM values from -312 to 280 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-21T17:10:03.778.054.725 (XO: 13.778059693 - 13.778365694, t=45.2 degC)  Rank: 00079
   TM values: Max =   280  Min =  -312  Ave =    -4.36  Std =    27.41
   Min TM value:-312 at an offset of 123.963 usec -> 2023-07-21T17:10:03.778.178.687 (XO: 13.778183656)
   Max TM value:280 at an offset of 123.738 usec -> 2023-07-21T17:10:03.778.178.462 (XO: 13.778183431)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 126.238 usec -> 2023-07-21T17:10:03.778.180.962 (XO: 13.778185931)
   Min dTM/dt [1/us]:-2.94e+03 at an offset of 123.850 usec -> 2023-07-21T17:10:03.778.178.575 (XO: 13.778183543)

 - Block number: 2 starting at 2023-07-21T17:10:18.299.688.725 (XO: 28.299691314 - 28.299997315, t=45.2 degC)  Rank: 00084
   TM values: Max =   219  Min =  -234  Ave =    -4.41  Std =    29.07
   Min TM value:-234 at an offset of 178.776 usec -> 2023-07-21T17:10:18.299.867.500 (XO: 28.299870090)
   Max TM value:219 at an offset of 178.551 usec -> 2023-07-21T17:10:18.299.867.275 (XO: 28.299869865)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 121.938 usec -> 2023-07-21T17:10:18.299.810.662 (XO: 28.299813252)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 178.663 usec -> 2023-07-21T17:10:18.299.867.388 (XO: 28.299869977)
