-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 15:58:21 2024 UTC
-- File 20230721_121322_1229.dat, 1837  packets found.
APID:
      1229 (EW) in 1837 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 1837 packets (all packets)
Total number of packets: 
      36720 in 1837 packets (all packets)
Packet No from 1 to 1837
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 1.089971 to 4.068993
-each separately from 1s to 4s and from 68993us to 89971us.
   Differences in neigboring packets: 
      0.000000 in 1835 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -2.979022 in 1 packets: 1428
MIU packet time from 2023-07-21T12:13:26.089.971.000 to 2023-07-21T12:13:29.068.993.000
   Differences in neigboring packets: 
      0.000000 in 1835 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -2.979022 in 1 packets: 1428
MIU orbit number: 
      60000 in 1837 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 1837 packets (all packets)
MIU orbit information / number of events: 
      0 in 1837 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 1837 packets (all packets)
Instrument anomalies: 
      0 (OK) in 1837 packets (all packets)
FPGA configuration version: 
      211 in 1837 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7486108 to 5.7486010
   Differences in neigboring packets: 
      0.0000000 in 1835 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -2.9999902 in 1 packets: 1428
Temperature FPGA: 
      FF -> 56.3 degC in 1428 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 409 packets: 1429, 1430, 1431, 1432, 1433, 1434, 1435, 1436, 1437, 1438, 1439, 1440, 1441, 1442, 1443, 1444, 1445, 1446, ...
Temperature XO: 
      C6 -> 45.2 degC in 1837 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 1837 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 1837 packets (all packets)
Number of external events: 
      3 in 1428 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 409 packets: 1429, 1430, 1431, 1432, 1433, 1434, 1435, 1436, 1437, 1438, 1439, 1440, 1441, 1442, 1443, 1444, 1445, 1446, ...
Number of generated alerts: 
      3 in 1428 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 409 packets: 1429, 1430, 1431, 1432, 1433, 1434, 1435, 1436, 1437, 1438, 1439, 1440, 1441, 1442, 1443, 1444, 1445, 1446, ...
Survey memory error: 
      0 in 1837 packets (all packets)
Event memory error: 
      0 in 1837 packets (all packets)
Configuration error: 
      0 in 1837 packets (all packets)
SDRAM CRC error: 
      0 in 1837 packets (all packets)
Radio gain: 
      0 in 1837 packets (all packets)
Radio bandwidth: 
      0 in 1837 packets (all packets)
Digital gain: 
      0 in 1837 packets (all packets)
Analog gain: 
      0 in 1837 packets (all packets)

EVENT DATA READER
-- File 20230721_121322_1229.dat, 9 EW pages found. Extra packets: 1.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 1428 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 408 packets: 1429, 1430, 1431, 1432, 1433, 1434, 1435, 1436, 1437, 1438, 1439, 1440, 1441, 1442, 1443, 1444, 1445, 1446, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 2.7830949 to 5.8695979
      Difference between the neighboring packets: 
      0.0000170 in 1834 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -3.0865030 in 1 packets: 1428
   Memory page number from 0 to 8
      Difference between the neighboring packets: 
      0 in 1827 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 8 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632
   Event page number from 1 to 31
      Difference between the neighboring packets: 
      0 in 1827 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 7 packets: 204, 408, 612, 816, 1020, 1224, 1632
      -30 in 1 packets: 1428
   MIU event flag 
      1 in 1836 packets (all packets)
   Time triggered event flag 
      0 in 1836 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-07-21T12:13:26.034.482.825
   Min EW header abs. time found among all events            : 2023-07-21T12:13:26.089.971.000
   Max EW event (1st sample) abs. time found among all events: 2023-07-21T12:13:29.013.504.825
   Max EW header abs. time found among all events            : 2023-07-21T12:13:29.068.993.000
   TM values from -248 to 221 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 408
    Page no. 1:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 2  ordered packets: 408
   TM values from -206 to 172 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=1 C1MHz=0.089971
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 407 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 2.7830949
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-21T12:13:26.089.971.000
   Event 1st sample abs. time : 2023-07-21T12:13:26.034.482.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 1428
    Page no. 25:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 7  ordered packets: 1428
   TM values from -248 to 221 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=4 C1MHz=0.068993
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1427 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 5.8453389
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-07-21T12:13:29.068.993.000
   Event 1st sample abs. time : 2023-07-21T12:13:29.013.504.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 from 2023-07-21T12:13:26.034.482.825 to 2023-07-21T12:13:26.041.418.825 (XO: -333.000000175 - -332.993064175)
   Min TM value:-206 at an offset of 3577.150 usec -> 2023-07-21T12:13:26.038.059.975 (XO: -332.996423025)
   Max TM value:172 at an offset of 3394.538 usec -> 2023-07-21T12:13:26.037.877.362 (XO: -332.996605638)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 3561.400 usec -> 2023-07-21T12:13:26.038.044.225 (XO: -332.996438775)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 3577.012 usec -> 2023-07-21T12:13:26.038.059.837 (XO: -332.996423163)
   Med: -5.00000   Ave: -4.5214911   Std: 7.3611547

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2023-07-21T12:13:29.013.504.825 to 2023-07-21T12:13:29.037.780.825 (XO: -333.000000175 - -332.975724175)
   Min TM value:-248 at an offset of 1677.938 usec -> 2023-07-21T12:13:29.015.182.762 (XO: -332.998322238)
   Max TM value:221 at an offset of 11702.800 usec -> 2023-07-21T12:13:29.025.207.625 (XO: -332.988297375)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 1680.237 usec -> 2023-07-21T12:13:29.015.185.062 (XO: -332.998319938)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 11702.900 usec -> 2023-07-21T12:13:29.025.207.725 (XO: -332.988297275)
   Med: -5.00000   Ave: -4.5718956   Std: 6.6724773
