-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 13:58:33 2024 UTC
-- File 20230721_002445_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177040
-each separately from 0s to 14s and from 0us to 841263us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833945 in 4 packets: 2, 6, 9, 16
      0.833944 in 10 packets: 3, 4, 5, 8, 10, 11, 12, 14, 15, 17
      0.833940 in 1 packets: 7
      0.833941 in 1 packets: 13
MIU packet time from 2023-07-21T00:24:48.000.000.000 to 2023-07-21T00:25:02.177.040.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833945 in 4 packets: 2, 6, 9, 16
      0.833944 in 10 packets: 3, 4, 5, 8, 10, 11, 12, 14, 15, 17
      0.833940 in 1 packets: 7
      0.833941 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.2794861 to 15.2794253
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999957 in 8 packets: 2, 4, 5, 8, 10, 14, 16, 17
      0.9999956 in 6 packets: 3, 6, 9, 11, 12, 15
   XO PPM: 
      -4.3 in 8 packets: 2, 4, 5, 8, 10, 14, 16, 17
      -4.4 in 6 packets: 3, 6, 9, 11, 12, 15
Temperature FPGA: 
      06 -> 47.2 degC in 2 packets: 1, 18
      04 -> 49.8 degC in 6 packets: 2, 9, 10, 13, 14, 15
      05 -> 48.5 degC in 10 packets: 3, 4, 5, 6, 7, 8, 11, 12, 16, 17
Temperature XO: 
      C4 -> 42.0 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      1 in 7 packets: 1, 2, 3, 4, 5, 6, 7
      2 in 2 packets: 8, 9
      3 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 7 packets: 1, 2, 3, 4, 5, 6, 7
      1 in 2 packets: 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230721_002445_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.2794862 to 15.4564662
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-21T00:24:48.000.000.000 to 2023-07-21T00:25:02.177.040.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833945 in 4 packets: 2, 6, 9, 16
      0.833944 in 10 packets: 3, 4, 5, 8, 10, 11, 12, 14, 15, 17
      0.833940 in 1 packets: 7
      0.833941 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.279486200 corrected by 40bit 10MHz counter of 1.279486100 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999995600 - 0.999995700
   Min 1st sample MIU time among all SR packets: 2023-07-21T00:24:47.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-21T00:25:02.177.039.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177041586
   TM values from -27 to 26 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-21T00:24:47.999.999.925 (XO: 0.000000025 - 0.000017025, t=42.0 degC)
   TM values: Max =    13  Min =   -24  Ave =    -4.46  Std =     6.29

 - Packet number: 2 starting at 2023-07-21T00:24:48.833.938.925 (XO: 0.833943611 - 0.833960611, t=42.0 degC)
   TM values: Max =    18  Min =   -22  Ave =    -3.89  Std =     5.93

 - Packet number: 3 starting at 2023-07-21T00:24:49.667.883.925 (XO: 1.667887264 - 1.667904264, t=42.0 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.39  Std =     6.22

 - Packet number: 4 starting at 2023-07-21T00:24:50.501.827.925 (XO: 2.501830883 - 2.501847883, t=42.0 degC)
   TM values: Max =    17  Min =   -24  Ave =    -4.38  Std =     6.32

 - Packet number: 5 starting at 2023-07-21T00:24:51.335.771.925 (XO: 3.335774469 - 3.335791469, t=42.0 degC)
   TM values: Max =    16  Min =   -23  Ave =    -4.22  Std =     6.00

 - Packet number: 6 starting at 2023-07-21T00:24:52.169.715.925 (XO: 4.169718072 - 4.169735072, t=42.0 degC)
   TM values: Max =    16  Min =   -22  Ave =    -4.37  Std =     5.78

 - Packet number: 7 starting at 2023-07-21T00:24:53.003.660.925 (XO: 5.003661741 - 5.003678741, t=42.0 degC)
   TM values: Max =    13  Min =   -23  Ave =    -4.26  Std =     6.23

 - Packet number: 8 starting at 2023-07-21T00:24:53.837.600.925 (XO: 5.837605327 - 5.837622327, t=42.0 degC)
   TM values: Max =    17  Min =   -23  Ave =    -4.25  Std =     6.34

 - Packet number: 9 starting at 2023-07-21T00:24:54.671.544.925 (XO: 6.671548980 - 6.671565980, t=42.0 degC)
   TM values: Max =    16  Min =   -27  Ave =    -4.36  Std =     6.87

 - Packet number: 10 starting at 2023-07-21T00:24:55.505.489.925 (XO: 7.505492599 - 7.505509599, t=42.0 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.33  Std =     5.73

 - Packet number: 11 starting at 2023-07-21T00:24:56.339.433.925 (XO: 8.339436219 - 8.339453219, t=42.0 degC)
   TM values: Max =    15  Min =   -27  Ave =    -4.20  Std =     6.69

 - Packet number: 12 starting at 2023-07-21T00:24:57.173.377.925 (XO: 9.173379888 - 9.173396888, t=42.0 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.28  Std =     6.20

 - Packet number: 13 starting at 2023-07-21T00:24:58.007.321.925 (XO: 10.007323556 - 10.007340557, t=42.0 degC)
   TM values: Max =    12  Min =   -19  Ave =    -4.43  Std =     5.41

 - Packet number: 14 starting at 2023-07-21T00:24:58.841.262.925 (XO: 10.841267142 - 10.841284143, t=42.0 degC)
   TM values: Max =    14  Min =   -24  Ave =    -4.27  Std =     6.35

 - Packet number: 15 starting at 2023-07-21T00:24:59.675.206.925 (XO: 11.675210796 - 11.675227796, t=42.0 degC)
   TM values: Max =    15  Min =   -25  Ave =    -4.41  Std =     5.98

 - Packet number: 16 starting at 2023-07-21T00:25:00.509.150.925 (XO: 12.509154414 - 12.509171414, t=42.0 degC)
   TM values: Max =    26  Min =   -22  Ave =    -4.30  Std =     6.07

 - Packet number: 17 starting at 2023-07-21T00:25:01.343.095.925 (XO: 13.343098000 - 13.343115000, t=42.0 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.39  Std =     5.43

 - Packet number: 18 starting at 2023-07-21T00:25:02.177.039.925 (XO: 14.177041586 - 14.177058586, t=42.0 degC)
   TM values: Max =    16  Min =   -27  Ave =    -4.27  Std =     6.48
